1 Kent Academic Repository Full text document (pdf) Citation for published version Toadere, Florin and Stancu, Radu.-F. and Poon, Wallace and Schultz, David and Podoleanu, Adrian G.H. (2017) 1 MHz Akinetic Dispersive Ring Cavity Swept Source at 850 nm. IEEE Photonics Technology Letters, 29 (11). pp ISSN DOI Link to record in KAR Document Version Author's Accepted Manuscript Copyright & reuse Content in the Kent Academic Repository is made available for research purposes. Unless otherwise stated all content is protected by copyright and in the absence of an open licence (eg Creative Commons), permissions for further reuse of content should be sought from the publisher, author or other copyright holder. Versions of research The version in the Kent Academic Repository may differ from the final published version. Users are advised to check for the status of the paper. Users should always cite the published version of record. Enquiries For any further enquiries regarding the licence status of this document, please contact: If you believe this document infringes copyright then please contact the KAR admin team with the take-down information provided at
2 PTL MHz akinetic dispersive ring cavity swept source at 850 nm Florin Toadere 1, Radu-F. Stancu 1, Wallace Poon 2, David Schultz 2, Adrian Podoleanu 1 Abstract - A fast dual mode-locked akinetic optical swept source at 850 nm central wavelength is presented using a dispersive cavity. We demonstrate that single mode fiber can be successfully used as dispersive element to induce mode locking. A first locking condition is imposed by driving the optical gain at a high frequency, to induce mode locking, while a second locking condition involves sweeping at a rate close to resonance value. In this regime, using the same fiber length in the loop, sweeping rates of 0.5 MHz and 1 MHz are demonstrated with proportional reduction in the tuning bandwidth. The axial range of the swept source was evaluated by scanning through the channeled spectrum of a Michelson interferometer. Index Terms Semiconductor optical amplifier, laser mode locking, ring laser, optical fiber dispersion, radio frequency. T I. INTRODUCTION he majority of commercial OCT systems for retina imaging operate in the nm band using spectrometer based OCT principle , . There is a trend in equipping commercial systems for ophthalmology with swept sources, due to longer axial range and better sensitivity, however fast sweeping rates are available at longer wavelengths only. At shorter wavelength, transversal resolution is better and the 800 nm band is also suitable for oxygenation measurements. Therefore, a continuous interest exists in developing swept sources at a 850 nm central wavelength. Development of fast swept sources (SS) focused on optomechanical principles of tuning applied to a closed loop laser cavity. To exemplify, the following devices were employed: a Fabry-Perot filter , a polygon filter , or a micro electrical mechanical scanning (MEMS) device [6,7]. These optomechanical elements can exhibit hysteresis and are strongly influenced by electrical instabilities. An alternative in the technology of swept sources for OCT consists in the elimination of the mechanical tuning elements from the laser cavity , to achieve akinetic operation. Another akinetic principle is based on dispersion tuning [9-12]. We have demonstrated recently an improvement to this concept, represented by dual resonance excitation . A first resonance condition is imposed by driving the optical gain at a multiple of the cavity resonance frequency, f R, to induce mode-locking. The second mechanism Paper submitted on: 01/11/2016. Radu-F. Stancu, Florin Toadere, Adrian Gh. Podoleanu are with the Applied Optics Group, School of Physical Sciences, University of Kent, Canterbury, United Kingdom, CT2 7NH ( Wallace Poon and David Schultz are with the East Kent Hospital University NHS Foundation Trust. A. P. was is responsible for generating sweeping at a rate close to f R, more exactly at values detuned from f R . So far, this dual resonance mechanism was proven at wavelengths longer than 1 m. In this paper, the concept of dual mode-locking and dispersion tuning is extended to the 850 nm band. In comparison with previous implementations, we demonstrate that the dispersion of single mode fiber at 850 nm, used as the necessary delay element, is sufficient to act as dispersive element for the principle illustrated. At longer wavelengths, other dispersive elements have been reported, such as dispersive compensating fiber [9,10,13] or chirped fiber Bragg gratings . MHz sweeping is reported with sufficient narrow linewidth (90 pm) and tuning bandwidth over 10 nm. II. DUAL MODE-LOCKING The operation of mode-locking dispersion ring cavity akinetic swept source (AKSS), sweeping at hundreds of khz was presented in several reports [9-13]. The resonance frequency of the ring laser cavity is defined as: c fr (1) nl where: c represents the speed of light in vacuum, n is the refractive index of the cavity and L is the laser cavity length. f R denotes the spacing between adjacent modes in a resonant cavity. Mode locking is induced by modulating the semiconductor optical amplifier (SOA) at a frequency f m0, that is a multiple, N, of the frequency f r, e.g. Nf R. Mode-locking allows obtaining ultrashort laser pulses at the SOA output. Thus, SOA operates at a wavelength that fulfills the modelooking condition. The wavelength tuning range is given by: n 0 fm (2) cdfm 0 where f m0 is the central mode-locking frequency, n 0 is the refractive index at central optical frequency, D is the dispersion parameter of the ring, and f m represents the change in modulation frequency. When f m reaches the resonance frequency, the maximum tuning range max is achieved: 1 max (3) D Lf m 0 The frequency of the RF signal is changed within the bandwidth f m using a voltage controlled oscillator (VCO). When the frequency of the signal driving the VCO is swept at f s = f R, the VCO s spectrum is described by a comb of frequencies f m0 ± f R, supported by the Royal Society Wolfson Research Merit Award and by the BRC-UCL at Inst. of Ophthalmology London; F.T was supported by Univ. of Kent; R.-F. S. was supported by the Brilliant Club Researchers in Schools Programme; F.T. and R-F. S. have equally contributed to the paper.
3 PTL f m0 ± 2f R, f m0 ± 3f R, and so on. Under these conditions, each component of the comb differs from f m0 by a multiple of f R and therefore all comb components determine the same laser optical frequency, repeated at f R. Therefore, to obtain optical tuning, the frequency of the signal driving the laser has to be detuned from resonance. For a detuning f R, with f s = f R ± f R, the spectrum of the VCO is a comb spectrum consisting in f m0, f m0 ± f R ± f R, f m0 ± 2f R ± 2 f R, f m0 ± 3f R ± 3 f R and so on. The step f r of the radiofrequency determines a change in wavelength. In such case, the output bandwidth is given by: (4) where C m is a coefficient and F represents the frequency tuning range of the VCO. The step change in wavelength can be assimilated to the linewidth of the optical emission. The optical configuration of the AKSS experimented is shown in Fig. 1. It uses a SOA model SOA-372, operating at 850 nm, maximum current 200 ma and 3 db bandwidth up to 40 nm. The SOA is placed in a laser system mount model HS501, and it is driven by a Thorlabs LDC 210C power supply. Temperature stability is ensured through thermo-electric cooling provided by a Thorlabs TEC 200C driver (sensor TH 20kΩ). The SOA is delimited by two polarization insensitive isolators IO-F-SLD which ensure unidirectional lasing. Output power is drawn out via a 50/50 FOC Photonics coupler with the central wavelength 850 nm, with 50% of power reinjected in the cavity. Fig. 1. Akinetic swept source schematic diagram The output spectra of the SS configuration are measured using a Yokogawa AQ6375 optical spectrum analyzer (OSA). The coherence length of the source is evaluated by feeding a balanced interferometer, equipped with a spectral delay line for dispersion adjustment, terminated on a 50/50 SIFAM coupler with the central wavelength 850 nm, ending on a New Focus balanced detector model 1807, 80 MHz, nm bandwidth, whose output is displayed with a LeCroy Wave Runner 104 MXi-A 1GHz oscilloscope, at a sampling rate of 10 GS/s (Gyga Sample points/ second). In dynamic regime, a ZX CA-S+ Mini Circuits low noise VCO and a ZHL-42W Mini Circuits radio frequency amplifier are used to drive the SOA. The VCO is driven by a ramp signal generated by an Agilent 81160A radio frequency signal generator (RFSG), with adjustable amplitude, frequency and offset. The offset of the ramp signal determines the central mode-locking frequency f mo generated by the VCO and its amplitude sets the tuning bandwidth F of the VCO. The repetition frequency of the ramp signal determines the tuning rate f s. Then the VCO s signal is amplified by the radio frequency amplifier before it is delivered to SOA. By changing the frequency of the VCO signal, the SOA optical frequency is swept. Empirically it was noted that ramps driving the optical frequency from high to low leads to a larger tuning bandwidth and a better signal to noise ratio than ramps driving the optical frequency from low to high. This is in line with previous observations in FDML lasers . III. EXPERIMENTAL RESULTS A Hewlett Packard 8648C RFSG, operating in the 9 khz MHz frequency band, was employed in the measurements performed in the static regime (manually adjusting the modelocking frequency). The amplitude of signal at the SOA driver input is 15 V at 250 MHz and 12 V at 1 GHz. At larger frequencies, although the generator output was increased, no change was noticed in the spectrum, suggesting that the power amplifier worked in saturation regime. However, we cannot measure the exact current swing through the SOA due to limitation of our electronic bandwidth (photodetector and oscilloscope). Inside the driver, there is a serial resistance of 33 in series with the SOA, determining a load of approximately 50. The cavity dispersion was adjusted by modifying the length of the resonant cavity. Thus, four types of single mode fibers are tested as listed in Table 1. For all four configurations the resonance frequencies are listed in the 3 rd column of Table 1. As expected, the larger the length of the cavity, less is the resonance frequency. The fourth and fifth columns list the optical bandwidth and respectively, the tuning of the f m0 range to generate the output bandwidth, for a mode-locking frequency f mo = 1 GHz. The widest tuning bandwidth was achieved when using fiber F2. Table 1. Static regime of the resonant cavity Fiber type Cavity length [m] Resonance fr [khz] max [nm] f s [khz] F1: SM 780/125 A F2: Thorlabs FS SN-4224 F3: Lightwave Tech Inc., F1506C F4: Ensign Bickford SMC A0820B Fig. 2 (left) illustrates the static bandwidth for two different values of the RF frequency, f m0. By varying the RF to either extremity of the RF bandwidth, two optical frequency components are seen. This proves that f m reaches the resonance frequency condition, that lead to Eq (3). Fig. 2. Left: Optical spectrum for f m0 = 250MHz ± 115 khz (solid line) and f m0 = 2GHz ± 125 khz for (dashed line); Right: Maximum tuning bandwidth (blue solid line) and the radio frequency tuning range F (dashed red line) necessary to achieve = 4 nm versus the frequency of the mode-locking frequency, f m0.
4 PTL Table 2 presents several measurements of the wavelengths and linewidths obtained in static regime employing a 400 m ring cavity. The central mode-locking frequency f m0 inducing the mode locking is varied from 250 MHz to 2 GHz. For each value of f m0, its frequency was statically modified and the optical frequency determined by the change was followed in its variation using the optical spectrum analyzer. The maximum bandwidth values obtained, max, are listed in the second column, while the linewidths measured are shown in the third column. The number of points in the spectrum are calculated in the fourth column, by dividing the maximum optical tuning bandwidth to the value of linewidth. Because this is measured for a static condition, where the frequency of the VCO is held fixed, we refer to such linewidth as static, st This is the minimum achievable linewidth and should be distinguished from the step used in Eq. (4) in dynamic regime. In the fifth column, the values of the dispersion parameter D are calculated using Eq. (3). These give an average value D = 199 [ps/nm/km]. The RF deviations that created these optical tuning bandwidths are mentioned in the sixth column. Table 2. Measured parameters of the AKSS employing a 400 m Lightwave Technologies, F1506C fiber ring cavity f m0 [MHz] max [nm] st [nm) max/ st Then we maintained the maximum frequency of 2 GHz of the signal applied to the SOA and changed the fiber. The narrowest static linewidths st of 70 pm are achieved using F3 and F4. Larger static linewidths of 130 pm and 120 pm are obtained using the fiber F1 and, respectively, F2. For the linewidth measurements, the OSA was adjusted on a resolution of 20 pm. These measurements refer to the width of envelope emission over many optical modes, impacting both the sensitivity decay with the path difference in the interferometer and what can be effectively measured with an OSA. Fig. 3. Amplitude versus radio frequency applied to VCO D [ps/nm/km] f m [khz] Understanding the RF spectrum is important, as each RF component in the spectrum determines an optical frequency, as described by equation (4). At 1500 khz tuning rate, only a single peak is left in the RF spectrum in Fig. 3. According to the theory of frequency modulation, the index of modulation is the ratio of F/f m. For the data in Fig. 2 (left), using 60 MHz/f m, suggests that even at 1 MHz, the index is larger than 1. This means that the radio frequency spectrum consists of more components than J1 (integer number Bessel function first order). At 1.2 MHz these multiple components can be resolved by the RFSA. According to Table 2, when using the VCO close to 1 GHz, if the RF is varied by 230 khz, then a tuning bandwidth of 13 nm should be achievable. Next, the dynamic regime operation of the source is characterized. A sinusoidal wave produced by the RSFG is applied to the VCO and the RF spectrum is measured using an Agilent RF Spectrum analyzer. The VCO operation is tested with signals of repetition frequency between 100 khz to 1500 khz and amplitude kept constant at 5 V. According to the VCO data sheet, this amplitude should produce a tuning of 100 MHz. Dynamically, the laser optical output is tuned according to two interleaved mechanisms, as previously communicated [8, 9]. A first mechanism is that of dispersive tuning. The second tuning mechanism involves resonance tuning using the cavity roundtrip; the tuning must be slightly detuned from the cavity resonance. We show the effects obtained for seven such steps, where each step is f R = 500 Hz. For f S = f R 500 Hz, the VCO frequency contains the following components: f m0 to f m0 f R 500 Hz, f m0 2f R 1 khz, f m0 3f R 1.5 khz and so on, in the MHz range, in steps of khz. The emitted spectrum consists of a comb having a linewidth of 80 pm and a repetition rate of khz. For a 3.8 Vpp signal driving the VCO and a laser driving current of 50 ma the output power of the AKSS is 1.6 mw. The 3.8Vpp determines a VCO tuning bandwidth of 60 MHz. Fig. 4. Optical spectra at the SS output measured for seven values of detuning of the sweeping frequency starting from resonance By modulating the VCO at sweeping rates of 999 khz, khz, 998 khz, and so on, down to 996 khz, in steps of 0.5 khz, the measured output bandwidths are: 1.3 nm, 2.5 nm, 4.5 nm, 7.5 nm, 10 nm, 11 nm and, respectively 13.5 nm. This is illustrated in Fig. 4. For an OPD = 50 µm, for each step of detuning the frequency of the VCO by 500 Hz, a new peak is observed in each channeled spectrum at the interferometer output (seen on the oscilloscope screen, not shown here) in addition to the previously existing peaks. While the tuning bandwidth increases, the optical power decreases, due to driving the laser away from the round trip resonance. The effects of tuning on the output optical power are better documented in Fig. 5. Fig. 5 (left) illustrates the output power versus amplitude of the ramp signal applied to the VCO s input for a detuning of f R = 3 khz when operating the AKSS dynamically at a 996 khz sweep rate. The larger the
5 PTL amplitude of the ramp driving the VCO, the larger the tuning bandwidth. On the other hand, a decrease in power is noticeable. Fig. 5 (right) depicts the optical tuning bandwidth and the output power versus the repetition rates around the central resonance of 999 khz. The larger the detuning from the resonance frequency, the wider the bandwidth and the lower the optical power. A maximum power of 1.6 mw is achieved at the output, without using a booster. Furthermore, the AKSS output was connected to a Michelson interferometer and used the following formulae to determine the useful power of the swept source ( P SS ) and the ASE power ( P ASE ), based on the optical power measured in the reference and, respectively, object arm: Pobj O PSS PASE, Pref R PSS PASE, where: O represents the object arm coefficient, R represents the reference arm coefficient. From the two equations above, the interference power is obtained as: int ORPSS. Using the method described above, it was concluded that the useful swept source output power represents 20% of the total output power. Fig. 5. (left) Output power (dashed blue line) and tuning bandwidth (red solid line) versus the amplitude of the ramp signal applied to the VCO input for f R = 3 khz; (right) tuning bandwidth (red solid line) and optical power (dashed blue line) versus the AKSS repetition rate Fig. 6. Amplitude decay of the photo-detected signal with OPD (roll off) for seven detuning values of the sweeping frequency f S detuned from f R = khz. The sensitivity S of the interferometer, in case is used for reflectivity measurement, is calculated as, where A OPD is amplitude of the FFT peak for an OPD close to 0, while A noise represents the amplitude of the noise floor measured outside the peak . S = 88.5 db is obtained for an OPD value of 0.1 mm. The shape for the spectrum emission is assumed Gaussian and using the OPD 6dB worked out from Fig. 6 of 1 mm (obtained for 2 khz detuning, i.e. the 998 khz curve), a dynamic linewidth of = pm results. This is slightly larger than the static linewidth measured in Table 2, of 80 pm. As the sweeping speed increases, the A-scans showed a decrease in the axial range (OPD). This can be explained by a decrease in the swept source coherence length with the sweeping speed, due to the decrease of roundtrip in the cavity, as it was previously shown in . However sufficient amplitude is obtained up to 6 mm for less detuning, as shown by the curves for and 999 khz in Fig. 4 and Fig. 6. IV. CONCLUSION In this paper, an 850 nm central wavelength AKSS swept at 1 MHz, with over 1 mw power is demonstrated. At a sweeping rate of khz, determining a 4 nm tuning bandwidth, the OPD for 6 db decay in the interferometer signal is 1 mm. The versatility of this source comes from the possibility of dynamically adjusting their tuning parameters. There is a large ASE power that limits the effectiveness of employing a booster. With more work on the electronic circuitry to secure enhanced mode locking, that couples more modes in the cavity, it is expected to shift more power from the ASE towards the swept frequency components. REFERENCES  M. Wojtkowski, R. Leitgeb, A. Kowalczyk, T. Bajraszewski and A. F. Fercher, In vivo human retinal imaging by Fourier domain optical coherence tomography, J. Biomed. Opt., 7(3), , (2002).    I. Trifanov, A. Bradu, L. Neagu, P. Guerreiro, A. B. Lobo Ribeiro and A. Gh. Podoleanu., Experimental method to find the optimum excitation waveform to quench mechanical resonances of Fabry Pérot tunable filters used in swept sources, IEEE Phot. Tech. Lett. 23(12), (2011).  S. H. Yun, G. Boudoux, G. J. Tearney, B. E. Bouma, High-speed wavelength-swept semiconductor laser with a polygon-scanner-based wavelength filter, Optics Letters 28, (2003)  J. Masson, R. St-Gelais, A. Poulin, Y.-A. Peter, Tunable Fiber Laser Using a MEMS-Based In Plane Fabry-Pérot Filter, IEEE Journal of Quantum Electronics 46, , (2010).  B. Potsaid et al., "MEMS tunable VCSEL light source for ultrahigh speed 60 khz-1 MHz axial scan rate and long range centimeter class OCT imaging," Proc. SPIE, 8213, 82130M-8 (2012).  T-H Tsai, B. Potsaid, Y. K. Tao, V. Jayaraman, J. Jiang, P. J. S. Heim, M. F. Kraus, C. Zhou, J. Hornegger, H. Mashimo, A. E. Cable and J. G. Fujimoto, "Ultrahigh speed endoscopic optical coherence tomography using micro-motor imaging catheter and VCSEL technology" Proc. SPIE, 8571, 85710N (2013).  Y. Nakazaki and S. Yamashita, Fast and wide tuning range wavelength-swept fiber laser based on dispersion tuning and its application to dyamic FBG sensing, Opt. Exp. 17(10), (2009).  S. Yamashita, Y. Nakazaki, R. Konishi, O. Kusakari, Wide and fast wavelength-swept fiber laser based on dispersion tuning for dynamic sensing, Journal of Sensors 2009, A Special Issue on Fiber and Integrated Waveguide-Based Opt. Sens., (2008).  Y. Takubo and S. Yamashita, High-speed dispersion-tuned wavelength-swept fiber laser using a reflective SOA and a chirped FBG, Optics Express, vol.21, no.4, pp , (2013).  R. F. Stancu, D. A. Jackson and A. G. Podoleanu, Versatile Swept Source with Adjustable Coherence Length, IEEE Photonics Technology Letters 26 (16), (2014).  R. F. Stancu and A. G. Podoleanu, Dual-mode-locking mechanism for an akinetic dispersive ring cavity swept source, Opt. Lett. 40(7), (2015).  R. Huber, M. Wojtkowski and J.G. Fujimoto, Fourier Domain Modelocking (FDML): A new laser operating regime and applications for optical coherence tomography, Optics Express 14(8), (2006).  A. Gh. Podoleanu and A. Bradu, Master-Slave interferometry for parallel spectral domain interferometry sensing and versatile 3D optical coherence tomography, Optics Express, Vol. 21, No. 16, (2013).  A. Takada, M. Fujino, S. Nagano, Dispersion dependence of linewidth in actively mode-locked ring lasers, Opt. Exp. 20, 4753 (2012).
PHOTONIC SENSORS / Vol. 5, No. 3, 215: 251 256 Short Ring Cavity Swept Source Based on a Highly Reflective Chirped FBG Radu-Florin STANCU * and Adrian PODOLEANU Applied Optics Group, School of Physical
Frequency comb swept lasers Tsung-Han Tsai 1, Chao Zhou 1, Desmond C. Adler 1, and James G. Fujimoto 1* 1 Department of Electrical Engineering and Computer Science and Research Laboratory of Electronics,
Frequency comb swept lasers for optical coherence tomography The MIT Faculty has made this article openly available. Please share how this access benefits you. Your story matters. Citation As Published
Frequency comb swept lasers The MIT Faculty has made this article openly available. Please share how this access benefits you. Your story matters. Citation As Published Publisher Tsai, Tsung-Han et al.
VI Temporal coherence characteristics of a superluminescent diode system with an optical feedback mechanism Fang-Wen Sheu and Pei-Ling Luo Department of Applied Physics, National Chiayi University, Chiayi
Optical coherence tomography Peter E. Andersen Optics and Plasma Research Department Risø National Laboratory E-mail firstname.lastname@example.org Outline Part I: Introduction to optical coherence tomography
Testing with Femtosecond Pulses White Paper PN 200-0200-00 Revision 1.3 January 2009 Calmar Laser, Inc www.calmarlaser.com Overview Calmar s femtosecond laser sources are passively mode-locked fiber lasers.
Wavelength switching using multicavity semiconductor laser diodes A. P. Kanjamala and A. F. J. Levi Department of Electrical Engineering University of Southern California Los Angeles, California 989-1111
IEEE TRANSACTIONS ON MICROWAVE THEORY AND TECHNIQUES, VOL. 54, NO. 2, FEBRUARY 2006 763 Photonic Generation of Microwave Signal Using a Rational Harmonic Mode-Locked Fiber Ring Laser Zhichao Deng and Jianping
564 Timing Noise Measurement of High-Repetition-Rate Optical Pulses Hidemi Tsuchida National Institute of Advanced Industrial Science and Technology 1-1-1 Umezono, Tsukuba, 305-8568 JAPAN Tel: 81-29-861-5342;
> 5' O ft I o Vi LASER DIODE MODULATION AND NOISE K. Petermann lnstitutfiir Hochfrequenztechnik, Technische Universitdt Berlin Kluwer Academic Publishers i Dordrecht / Boston / London KTK Scientific Publishers
High-speed imaging of human retina in vivo with swept-source optical coherence tomography H. Lim, M. Mujat, C. Kerbage, E. C. W. Lee, and Y. Chen Harvard Medical School and Wellman Center for Photomedicine,
Application Note 13 Swept Wavelength Testing: Characterizing the Tuning Linearity of Tunable Laser Sources In a swept-wavelength measurement system, the wavelength of a tunable laser source (TLS) is swept
60 MHz Aline rate ultrahigh speed Fourierdomain optical coherence tomography K. Ohbayashi a,b), D. Choi b), H. HiroOka b), H. Furukawa b), R. Yoshimura b), M. Nakanishi c), and K. Shimizu c) a Graduate
314 JOURNAL OF LIGHTWAVE TECHNOLOGY, VOL. 27, NO. 3, FEBRUARY 1, 2009 Microwave Photonics Jianping Yao, Senior Member, IEEE, Member, OSA (Invited Tutorial) Abstract Broadband and low loss capability of
International Conference on Logistics Engineering, Management and Computer Science (LEMCS 2014) All-Optical Clock Division Using Period-one Oscillation of Optically Injected Semiconductor Laser Shengxiao
Pulsed-source and swept-source spectral-domain optical coherence tomography with reduced motion artifacts S. H. Yun, G. J. Tearney, J. F. de Boer, and B. E. Bouma Harvard Medical School and Wellman Center
PHOTONIC INTEGRATED CIRCUITS FOR PHASED-ARRAY BEAMFORMING F.E. VAN VLIET J. STULEMEIJER # K.W.BENOIST D.P.H. MAAT # M.K.SMIT # R. VAN DIJK * * TNO Physics and Electronics Laboratory P.O. Box 96864 2509
High stability multiplexed fibre interferometer and its application on absolute displacement measurement and on-line surface metrology Dejiao Lin, Xiangqian Jiang and Fang Xie Centre for Precision Technologies,
Mode-locking and frequency beating in Michael J. Strain Institute of Photonics Dept. of Physics University of Strathclyde compact semiconductor lasers Outline Pulsed lasers Mode-locking basics Semiconductor
Multi-wavelength laser generation with Bismuthbased Erbium-doped fiber H. Ahmad 1, S. Shahi 1 and S. W. Harun 1,2* 1 Photonics Research Center, University of Malaya, 50603 Kuala Lumpur, Malaysia 2 Department
328 JOURNAL OF LIGHTWAVE TECHNOLOGY, VOL. 17, NO. 2, FEBRUARY 1999 High-Performance Phase Locking of Wide Linewidth Semiconductor Lasers by Combined Use of Optical Injection Locking and Optical Phase-Lock
Optical Sources (a) Optical Sources (b) The main light sources used with fibre optic systems are: Light-emitting diodes (LEDs) Semiconductor lasers (diode lasers) Fibre laser and other compact solid-state
1/36 All-Optical Signal Processing and Optical Regeneration Govind P. Agrawal Institute of Optics University of Rochester Rochester, NY 14627 c 2007 G. P. Agrawal Outline Introduction Major Nonlinear Effects
838 IEEE JOURNAL OF QUANTUM ELECTRONICS, VOL. 43, NO. 9, SEPTEMBER 2007 Design of a Fiber Bragg Grating External Cavity Diode Laser to Realize Mode-Hop Isolation Toshiya Sato Abstract Recently, a unique
Laser Diode Light emitters are a key element in any fiber optic system. This component converts the electrical signal into a corresponding light signal that can be injected into the fiber. The light emitter
All optical wavelength converter based on fiber cross-phase modulation and fiber Bragg grating Pavel Honzatko a, a Institute of Photonics and Electronics, Academy of Sciences of the Czech Republic, v.v.i.,
FFP-C Fiber Fabry-Perot Controller OPERATING INSTRUCTIONS Version 1.0 MICRON OPTICS, INC. 1852 Century Place NE Atlanta, GA 30345 USA Tel (404) 325-0005 Fax (404) 325-4082 www.micronoptics.com Page 2 Table
I. Introduction II. III. IV. SLED Fundamentals SLED Temperature Performance SLED and Optical Feedback V. Operation Stability, Reliability and Life VI. Summary InPhenix, Inc., 25 N. Mines Road, Livermore,
756 IEEE JOURNAL OF SELECTED TOPICS IN QUANTUM ELECTRONICS, VOL. 6, NO. 5, SEPTEMBER/OCTOBER 2000 FMCW Multiplexing of Fiber Bragg Grating Sensors Peter K. C. Chan, Wei Jin, Senior Member, IEEE, and M.
JOURNAL OF LIGHTWAVE TECHNOLOGY, VOL. 29, NO. 19, OCTOBER 1, 2011 2927 Ultrafast and Ultrahigh-Resolution Interrogation of a Fiber Bragg Grating Sensor Based on Interferometric Temporal Spectroscopy Chao
Optical Coherence Tomography Systems and signal processing in SD-OCT Chandan S.Rawat 1, Vishal S.Gaikwad 2 1 Associate Professor V.E.S.I.T., Mumbai email@example.com 2 P.G.Student, V.E.S.I.T., Mumbai
University of Lübeck, Medical Laser Center Lübeck GmbH Optical Coherence Tomography 3. The Art of OCT Dr. Gereon Hüttmann / 2009 System perspective (links clickable) Light sources Superluminescent diodes
Holography Transmitter Design Bill Shillue 2000-Oct-03 Planned Photonic Reference Distribution for Test Interferometer The transmitter for the holography receiver is made up mostly of parts that are already
Ring cavity tunable fiber laser with external transversely chirped Bragg grating A. Ryasnyanskiy, V. Smirnov, L. Glebova, O. Mokhun, E. Rotari, A. Glebov and L. Glebov 2 OptiGrate, 562 South Econ Circle,
Pulse shortening of gain switched single mode semiconductor lasers using a variable delay interferometer Antonio Consoli* and Ignacio Esquivias Departamento de Tecnología Fotónica y Bioingeniería - CEMDATIC,
Edith Cowan University Research Online ECU Publications 2011 2011 Stabilisation of Linear-cavity Fibre Laser Using a Saturable Absorber David Michel Edith Cowan University Feng Xiao Edith Cowan University
High-power semiconductor lasers for applications requiring GHz linewidth source Ivan Divliansky* a, Vadim Smirnov b, George Venus a, Alex Gourevitch a, Leonid Glebov a a CREOL/The College of Optics and
Mitnick et al. Vol. 15, No. 9/September 1998/J. Opt. Soc. Am. B 2433 Coupling effects of signal and pump beams in three-level saturable-gain media Yuri Mitnick, Moshe Horowitz, and Baruch Fischer Department
EXERCISES OF OPTICAL MEASUREMENTS BY ENRICO RANDONE AND CESARE SVELTO EXERCISE 1 A CW laser radiation (λ=2.1 µm) is delivered to a Fabry-Pérot interferometer made of 2 identical plane and parallel mirrors
Periodic Error Correction in Heterodyne Interferometry Tony L. Schmitz, Vasishta Ganguly, Janet Yun, and Russell Loughridge Abstract This paper describes periodic error in differentialpath interferometry
High-speed spectral-domain optical coherence tomography at 1.3 µm wavelength S. H. Yun, G. J. Tearney, B. E. Bouma, B. H. Park, and J. F. de Boer Harvard Medical School and Wellman Center of Photomedicine,
846 PIERS Proceedings, Cambridge, USA, July 6, 8 Design and Analysis of Resonant Leaky-mode Broadband Reflectors M. Shokooh-Saremi and R. Magnusson Department of Electrical and Computer Engineering, University
Wavelength Control and Locking with Sub-MHz Precision A PZT actuator on one of the resonator mirrors enables the Verdi output wavelength to be rapidly tuned over a range of several GHz or tightly locked
JOURNAL OF L A TEX CLASS FILES, VOL. X, NO. XX, XXXX XXX 1 Downstream Transmission in a WDM-PON System Using a Multiwavelength SOA-Based Fiber Ring Laser Source Jérôme Vasseur, Jianjun Yu Senior Member,
Spectral domain optical coherence tomography with balanced detection using single line-scan camera and optical delay line Min Gyu Hyeon, 1 Hyung-Jin Kim, 2 Beop-Min Kim, 1,2,4 and Tae Joong Eom 3,5 1 Department
Q-switched resonantly diode-pumped Er:YAG laser Igor Kudryashov a) and Alexei Katsnelson Princeton Lightwave Inc., 2555 US Route 130, Cranbury, New Jersey, 08512 ABSTRACT In this work, resonant diode pumping
A WDM passive optical network enabling multicasting with color-free ONUs Yue Tian, Qingjiang Chang, and Yikai Su * State Key Laboratory of Advanced Optical Communication Systems and Networks, Department
Second International Symposium on Space Terahertz Technology Page 523 MILLIMETER WAVE RADIATION GENERATED BY OPTICAL MIXING IN FETs INTEGRATED WITH PRINTED CIRCUIT ANTENNAS by D.V. Plant, H.R. Fetterman,
Experts in next generation test equipment 56:/)'2 :+9: 3+'9;8+3+4: Optical Spectrum Analyzer Optical Complex Spectrum Analyzer Optical MultiTest Platform & Modules AP2040 series - OSA 4 AP2050 series -
Experiment No. 3 Pre-Lab Phase Locked Loops and Frequency Modulation The Pre-Labs are informational and although they follow the procedures in the experiment, they are to be completed outside of the laboratory.
Phase Noise Modeling of Opto-Mechanical Oscillators Siddharth Tallur, Suresh Sridaran, Sunil A. Bhave OxideMEMS Lab, School of Electrical and Computer Engineering Cornell University Ithaca, New York 14853
University of Central Florida Electronic Theses and Dissertations Doctoral Dissertation (Open Access) A Laser Radar Employing Linearly Chirped Pulses From A Mode-locked Laser For Long Range, Unambiguous,
Tunable single frequency fiber laser based on FP-LD injection locking Aiqin Zhang, Xinhuan Feng, * Minggui Wan, Zhaohui Li, and Bai-ou Guan Institute of Photonics Technology, Jinan University, Guangzhou,
Suppression of Rayleigh-scattering-induced noise in OEOs Olukayode Okusaga, 1,* James P. Cahill, 1,2 Andrew Docherty, 2 Curtis R. Menyuk, 2 Weimin Zhou, 1 and Gary M. Carter, 2 1 Sensors and Electronic
Fusiek, Grzegorz and Niewczas, Pawel (215) Laboratory investigation of an intensiometric dual FBG-based hybrid voltage sensor. In: Proceedings of SPIE - The International Society for Optical Engineering.
Journal of Physics: Conference Series Realization of 16-channel digital PGC demodulator for fiber laser sensor array To cite this article: Lin Wang et al 2011 J. Phys.: Conf. Ser. 276 012134 View the article
DESIGN AND CHARACTERIZATION OF HIGH PERFORMANCE C AND L BAND ERBIUM DOPED FIBER AMPLIFIERS (C,L-EDFAs) Ahmet Altuncu Arif Başgümüş Burçin Uzunca Ekim Haznedaroğlu e-mail: firstname.lastname@example.org e-mail:
Dr. Rüdiger Paschotta RP Photonics Consulting GmbH Competence Area: Fiber Devices Topics in this Area Fiber lasers, including exotic types Fiber amplifiers, including telecom-type devices and high power
Proc. of Int. Conf. on Recent Trends in Information, Telecommunication and Computing, ITC Performance Evaluation using M-QAM Modulated Optical OFDM Signals Harsimran Jit Kaur 1 and Dr.M. L. Singh 2 1 Chitkara
Amplitude independent RF instantaneous frequency measurement system using photonic Hilbert transform H. Emami, N. Sarkhosh, L. A. Bui, and A. Mitchell Microelectronics and Material Technology Center School
Phase-Lock Techniques for Phase and Frequency Control of Semiconductor Lasers Lee Center Workshop 05/22/2009 Amnon Yariv California Institute of Technology Naresh Satyan, Wei Liang, Arseny Vasilyev Caltech
Testing with 40 GHz Laser Sources White Paper PN 200-0500-00 Revision 1.1 January 2009 Calmar Laser, Inc www.calmarlaser.com Overview Calmar s 40 GHz fiber lasers are actively mode-locked fiber lasers.
Science in China Ser. G Physics, Mechanics & Astronomy 2004 Vol.47 No.6 767 772 767 A CW seeded femtosecond optical parametric amplifier ZHU Heyuan, XU Guang, WANG Tao, QIAN Liejia & FAN Dianyuan State
Fabry Perot Resonator (CA-1140) The open frame Fabry Perot kit CA-1140 was designed for demonstration and investigation of characteristics like resonance, free spectral range and finesse of a resonator.
The Virgo detector The Virgo detector L. Rolland LAPP-Annecy GraSPA summer school 2013 1 Table of contents Principles Effect of GW on free fall masses Basic detection principle overview Are the Virgo mirrors
ECEN689: Special Topics in Optical Interconnects Circuits and Systems Spring 016 Lecture 7: Transmitter Analysis Sam Palermo Analog & Mixed-Signal Center Texas A&M University Optical Modulation Techniques
Agilent 81662A DFB Laser Agilent 81663A DFB Laser Agilent Fabry-Perot Lasers Technical Specifications May 2003 The Agilent 81662A low power and 81663A high power DFB Laser Source modules are best suited
Supplementary Methods and Data 1. Apparatus Design The time-of-flight measurement apparatus built in this study is shown in Supplementary Figure 1. An erbium-doped femtosecond fibre oscillator (C-Fiber,
Optically switched erbium fibre laser using a tunable fibre-bragg grating Robert J. Williams, * Nemanja Jovanovic, Graham D. Marshall and Michael J. Withford. Centre for Ultrahigh bandwidth Devices for
LASER INTERFEROMETER GRAVITATIONAL WAVE OBSERVATORY - LIGO - CALIFORNIA INSTITUTE OF TECHNOLOGY MASSACHUSETTS INSTITUTE OF TECHNOLOGY LIGO-XXXXXXX-XX-X Date: 2009/09/25 Multiply Resonant EOM for the LIGO
Lu et al. Vol. 20, No. 7/July 2003 / J. Opt. Soc. Am. B 1473 Soliton stability conditions in actively modelocked inhomogeneously broadened lasers Wei Lu,* Li Yan, and Curtis R. Menyuk Department of Computer
Communication using Synchronization of Chaos in Semiconductor Lasers with optoelectronic feedback S. Tang, L. Illing, J. M. Liu, H. D. I. barbanel and M. B. Kennel Department of Electrical Engineering,
Photonics and Optical Communication (Course Number 300352) Spring 2007 Dr. Dietmar Knipp Assistant Professor of Electrical Engineering http://www.faculty.iu-bremen.de/dknipp/ 1 Photonics and Optical Communication
CREOL The College of Optics & Photonics Single-mode lasing in PT-symmetric microring resonators Matthias Heinrich 1, Hossein Hodaei 2, Mohammad-Ali Miri 2, Demetrios N. Christodoulides 2 & Mercedeh Khajavikhan
ModBox Pulse Generation Unit The ModBox Family The ModBox systems are a family of turnkey optical transmitters and external modulation benchtop units for digital and analog transmission, pulsed and other
Optimisation of DSF and SOA based Phase Conjugators by Incorporating Noise-Suppressing Fibre Gratings Paper no: 1471 S. Y. Set, H. Geiger, R. I. Laming, M. J. Cole and L. Reekie Optoelectronics Research