Lab E5: Filters and Complex Impedance


 Clare Jackson
 1 years ago
 Views:
Transcription
1 E5.1 Lab E5: Filters and Complex Impedance Note: It is strongly recommended that you complete lab E4: Capacitors and the RC Circuit before performing this experiment. Introduction Ohm s law, a well known experimental fact, states that the current through a circuit is related to the overall voltage drop and the resistance of the components by the relation (1) I = This can be further generalized to become (2) I = where Z represents the complex impedance, which has both real and imaginary components. Impedance is analogous to direct current (DC) resistance, but in addition to the real resistance also includes an imaginary reactance term due to the oscillatory effects of, for instance, charging and discharging a capacitor that come into play when we switch to using an oscillating alternating current (AC) source to power our circuit. (Recall that the simplest form of a capacitor is two parallel plates with a gap inbetween them, creating time dependence as observed in the E4: Capacitors lab. In that lab we looked at a single discharging cycle of the capacitor to determine the time constant of the circuit. Here we apply an oscillating voltage that causes the capacitor to constantly charge and discharge.) The imaginary nature of the reactance gives phase to the impedance, indicating that the current is out of phase with the voltage across that component. Another common passive component, the inductor, creates a time dependence of its own. An inductor is essentially just a coil of wire; when current flows through, a magnetic field is created in the coil. If the input voltage changes, the inductor creates a voltage across itself opposing that change (you may remember Faraday s Law of Inductance and Lenz Law from Physics 1120). The voltage drop across the inductor is given by (3) V = L( "() " ) where L, known as the inductance, is determined by the geometry and number of coils in the loop, and i(t) is the time dependent current through the inductor. We have switched to the lower case i in order to indicate time dependence, a common convention. However, from this point forward in the lab i will denote the imaginary number, i = 1. Notice that there must be a time dependence in the current for any voltage drop to occur for this reason inductors don t factor into DC circuit analysis, where the input voltage is constant in time. The same is true for capacitors. The complex impedances of the capacitor and inductor are given by (4) Z = = "# " (impedance of a capacitor)
2 E5.2 and (5) Z = iωl. (impedance of an inductor) where (6) ω = 2πf The impedances are purely imaginary and depend on frequency, indicating that they are only relevant when the voltage changes with time. In contrast, the impedance of a resistor is purely real, where (7) Z = R. (impedance of a resistor) This is to be expected, since the resistor creates a voltage drop even in a DC circuit with a time independent voltage. These complex impedances add in the same way the resistances add; recall that in series this means that (8) Z = Z + Z + Z + while in parallel: (9) Z = A common method of decreasing the voltage sent to a given part of your circuit is to create a voltage divider (pictured below). We can analyze the divider by using Ohm s law: Figure 1: Voltage Divider (10) I = " "#$ (Think about it: where is I measured? Does it matter?) (11) Z "#$ = Z + Z
3 E5.3 V out, the voltage we re trying to find, is the voltage dropped across the second impedance, so we use Ohm s law once again to find that (12) V "# = IZ = V ". When we use resistors for Z 1 and Z 2, the divider simply reduces the voltage output independent of the signal frequency, even if we are using an alternating current (AC) input. Figure 2: CR highpass filter When we use an inductor or a capacitor, however, we find (through a little algebra) interesting frequency dependence. Given the circuit above, we see that: (13) I = " = " "#$ = " " " (14) V "# = IZ = IR = " " R (15) "# " = "# " "# " = Note: the * indicates the complex conjugate. (16) V "# = "#$ "#$ V " This circuit is called a CR highpass filter. You can see that, for large frequencies f, Equation (15) approaches unity, whereas for small f it approaches 0. If we switch the positions of the resistor and capacitor above we get a similar result, (17) V "# = "#$ V " which is known as an RC lowpass filter. For these circuits we define the cutoff frequency to be the point at which the ratio
4 E5.4 (18) "# " =.707 The cutoff frequency in either case is therefore (19) f c = If we were to construct a loglog scale plot for this ratio as a function of frequency (using R = 4700 Ω and C = 33 nf for example) we would get " for the first circuit and for the second. The cutoff frequency (~1026 Hz) is marked for each case. As we can see the two circuits block out low and high frequency inputs respectively.
5 E5.5 Similar filters can be constructed using resistors and inductors, although with slightly different time dependence (see prelab questions). Procedure Part 1. CR High Pass Filter In this part we will analyze a simple high pass filter and plot attenuation (the ratio of V in to V out ) vs. frequency over a broad range. The circuit is built for you at the lab station and the components are labeled with their actual values. Determine the cutoff frequency f c for the CR circuit (figure 2) at your table. You will use this calculated cutoff frequency in your measurements. The values should be about 33 nf and 4.7 kω, but the actual values can vary so use the values labeled on the circuit box. To connect the signal generator to your circuit you will use a BNC cable. These cables carry your signal internally but also have a grounded shell (not connected to the inner wire). Connect a BNC cable to the Input terminal on the circuit box. Connect the other end to the function generator using a BNC Tsplitter. Use another BNC cable to connect the other end of the Tsplitter and Channel 1 on the Oscilloscope. Take another BNC cable and attach it from the output TTL on the function generator to the EXT Trig slot on your scope. This will be used to trigger the measurement. Triggering tells the scope when to take measurements in order to get a consistent signal, as opposed to taking a different signal with each sweep which shows up as a signal that appears to move across your screen. Turn on the function generator making sure channel 1 is onscreen on your oscilloscope and that the oscilloscope is set to trigger off of the channel connected to Output TTL on the function generator.
6 E5.6 Figure 3: Wiring Setup (Figure 2 gives the circuit diagram) Connect another BNC to the circuit Output and into channel 2 on the oscilloscope. Write down the input voltage, with error, as measured on your scope, using the vertical scale divisions knob to make sure the entire signal is on screen while also filling it as much as possible. This allows you to take the most accurate measurements possible when collecting data for the remainder of the lab always be sure to use this technique. Any time you adjust the input frequency you will need to check your waveforms so that they fill the screen on the oscilloscope again. Make voltage measurements at values of.001*(f c ),.01*(f c ),.1*(f c ),.5*(f c ), f c, 2*(f c ), 10*(f c ), 100*(f c ), and 1000*(f c ). Where f c is the cutoff frequency that you calculated earlier. Record the exact frequency off of your function generator. For very low frequencies, such as.001 and.01 f c, you may not be able to trigger your signal. If this is the case, use the Run/Stop button on your oscilloscope, allowing you to look at just a single measurement. Divide your measurements by V in, and create a data table including error for each result. Plot your data and the theory curve on a loglog scale plot in Mathematica, and comment on any discrepancy. Part 2. Integrator and Differentiator Circuits Using the high pass filter you constructed in the previous part of this lab, look at the input and output signals for various input waveforms (i.e. square wave, triangle wave, sine wave, etc.) found on the right side of the function generator. Keep the frequency of the signals well below your calculated f c (in the 0.01 f c f c range). What do you notice about the waveform of the output signal as compared to the input (Hint: see section title)? Switch the resistor and capacitor to create a low pass filter, and change your input frequency to well above the circuit s f c. What do you notice now? Which of these two circuits
7 E5.7 would you call an integrator? Which a differentiator? Explain. State what the integral and derivative waveforms of each of the three inputs mentioned above are. Do the circuits make a good approximation of differentiation and integration? Hint: think carefully about what the integral of a triangle waveform should look like. It is NOT a sine wave. Part 3. Signal Processing and the ECG One of the most important applications of filters is to reduce electronic noise in signals from measurement devices. When measuring a signal of a certain frequency you are nearly certain to have your measurement distorted by a multitude of other signals over a range of other frequencies. If your signal amplitude is low enough your measurement can be drowned out completely The goal of signal processing is to reduce this noise as much as possible so that you only look at the data you are interested in. By attenuating signals far from the cutoff frequency in one direction while allowing signals on the opposite side to pass through, RC and RL low and high pass filters used in combination are good candidates for this task. The electrocardiogram (ECG) is commonly used in hospitals to monitor patients heart beats and detect heart problems as a diagnostic tool. It works by amplifying tiny voltages present in the skin caused by polarization and depolarization of the heart. The human heart has four chambers: two atria, where blood enters the heart, and two ventricles, which pump the blood out. During the heart s cycle the atria and then the ventricles contract then relax in succession, pushing blood through the body. These contractions are caused by an electrical signal proliferated by the sinoatrial node, otherwise known as the pacemaker due to the fact that the cells spontaneously and rhythmically depolarize of their own accord. This electrical signal creates a very small net voltage difference across the body due to the asymmetric positioning of the transduction cells on the heart. This tiny signal can be read off the skin on opposite sides of the body and amplified to give doctors an idea of how a heart is functioning. The largest peak in an ECG readout, known as the QRS complex, represents ventricular contraction. Depending on the heart of the person in question, the conductivity of their skin, and the positions at which the signals are measured, the voltage difference across the body at this point will be around 13 mv. This presents a challenge, since the largest source of noise in most of these circuits, which comes from power lines, is large enough to completely obscure the signal and will be amplified during the amplification phase of the circuits. Luckily, power lines operate at 60 Hz, whereas the human heart (which beats only around 100 times per minute) operates at a frequency between 1 and 2 Hz. Almost all of the circuitry that goes into an ECG is aimed at eliminating this noise by applying various filters to allow visualization of the heart cycle. In this part of the lab you will compute the actual maximum voltage difference created at your hands by your heart. ECG circuits first amplify the signal before filtering it, since otherwise peaks would be too small to resolve. To determine your heart voltage you will thus need to know the degree to which the ECG you are using amplifies the signal. Make sure the power cord on the ECG box is plugged in, then connect the OUTPUT jack to channel 1 of your oscilloscope using a BNC cable. Turn on the circuit, set the mode to CALIBRATE, and switch three to Vin. Using a DMM (digital multimeter) with a BNC cable attachment, measure the input voltage to the calibration pulse and it s error by connecting the DMM to the LEFT/Vin socket. Pressing the CALIBRATING PULSE button will send a signal into the circuit at this voltage. Remove the
8 E5.8 DMM and flip switch three to OPERATE. On the oscilloscope, adjust the time scale to around 250 ms/division and, using the calibrate button, adjust the voltage divisions until the pulses take up most of the oscilloscope screen. Measure the output calibration voltage on the oscilloscope with error (the error on any oscilloscope measurement is + one small division note that making the large divisions as small as possible minimizes error, which is why it is best to take up as much of the screen with the pulses as you can without clipping the signal). Use the input and output calibration voltages to determine the amplification factor of the circuit and its error. Next attach the handle bar BNC cables to the LEFT/Vin and RIGHT sockets on the box, and switch the mode to OPERATE. In this mode the box will pick up signals from the handlebars, amplify and filter the signal, then output what remains. To get an accurate reading, rest your hands gently on the bars and stay as still and relaxed as you can muscle contractions also create transient voltages that will distort your signal. If you don t see your heartbeat right off the bat don t worry sometimes it can take seconds to resolve to a steady pattern. When you see your heartbeat, pick a typical cycle and measure the peak to peak voltage of the QRS complex with error. Using the amplification factor and the measured output voltage for your heartbeat, calculate the voltage running across your hands whenever your ventricles contract. Neat Guiding Questions 1) What is the function in Mathematica used to create a loglog scale plot of discrete data points? (1 point) 2) When plotting the theory curve on a loglog scale, be careful not to begin your plot range at 0. Why is this necessary? (1 point) 3) For an EKG monitor, one problem in signal amplification is power line noise which comes in at around 60 Hz (the human heart beats at around 1 Hz). If you wanted to eliminate this noise using an RC filter with a 1 µf capacitor, what valued resistor would you choose? Draw a schematic of the filter. (2 points) 4) Derive the ratio "# " for the RL filter shown below, and determine the cutoff frequency. Hint: see equations Your final answer should be "# " = /"#. (3 points) Figure 4: RL lowpass filter 5) Show that the units on the right hand side of the ratio above cancel to give a unitless quantity. (1 point) 6) Filters have many real world applications. Write down a couple ideas you have for possible uses. (1 point) 7) By combining a resistor, an inductor, and a capacitor all into one RLC circuit, as shown below, it is possible to filter out signals above and below a certain frequency. This is
9 E5.9 known as a bandpass filter, since only a narrow band of signals is allowed through. The width of the band is determined by the Q (for quality) factor of the circuit, which we won t discuss in depth. Give the values for Z 1 and Z 2 in the circuit below. (1 point) Figure 5: RLC bandpass filter
10 E5.10 Prelab assignment for E5: Filters and Complex Impedance. This prelab is designed to focus your attention on what is important in the lab before you start the experiment and give you a leg up on writing your report. Use this template when you write your report. 1. Carefully read the lab instructions for E5 2. Using Mathematica, set up a template for your lab report. This will include: I. Header information such as title, author, lab partner, and lab section: (to do this, In Mathematica, go to File, Print Settings, Header and Footer from this dialog box you can enter all of the required information.) Mathematica automatically enters the file name in the right side of the header. Leave this alone. II. Section Headings including Title, Summary of Experiment, Data and Calculations, Discussion of Uncertainty, and specific section headings for each part of the experiment. (All of the Physics 1140 lab manuals include multiple parts.) 3. Write a Summary of the experiment. What are you going to measure and what data will you collect to make the measurement? (For example, in part one of M1, you will measure the length of a pendulum along with the period of the pendulum to determine the acceleration due to gravity.) 4. For each part of the lab do the following: A.) For part one: Suppose you place a 3600 Ohm resistor and a 23nF capacitor in the High Pass circuit as described above. Calculate the cutoff frequency and create a theory plot of the output voltage (V out ) versus the frequency. Use equation (16) for inspiration and assume a 5V input voltage. B.) For Part two: you are asked to determine whether the high pass filter is an integrator or differentiator, and then again determine the same thing for the low pass filter. For this part, draw what you would expect to see as a graph for the differentiator of the triangle wave. Draw the same for the square wave. Don t worry about doing this in Mathematica, just draw it by hand. C.) For Part three: our ECG uses electrodes that you place your hands on. In two or three sentences describe how the ECG can detect your heartbeat. Also, how do high and low pass filters hone in on your heart beat?
11 E5.11 Remember, all plots you make in a prelab or lab report should include the following: I. Plot your theory or expectation as a line. (In Mathematica, you should define a function and plot it using the command Plot ) Reference the Mathematica tutorial to do this. II. Label your axes, in English (not just symbols) and with units. III. Include a brief caption (namely, a text statement of what the plot shows.) IV. Set the x and y range of the plot to be close to what you expect for your data. For example: in M1 your longest length pendulum should be no more than 130 cm in length. 5. Write a Discussion of Uncertainty. What are the major sources of uncertainty in the experiment and how will you account for them? 6. Turn in a printout of your Mathematica document that includes 25 above. This document should be no more than 2 pages long.
Lab E5: Filters and Complex Impedance
E5.1 Lab E5: Filters and Complex Impedance Note: It is strongly recommended that you complete lab E4: Capacitors and the RC Circuit before performing this experiment. Introduction Ohm s law, a well known
More informationOPERATIONAL AMPLIFIERS (OPAMPS) II
OPERATIONAL AMPLIFIERS (OPAMPS) II LAB 5 INTRO: INTRODUCTION TO INVERTING AMPLIFIERS AND OTHER OPAMP CIRCUITS GOALS In this lab, you will characterize the gain and frequency dependence of inverting opamp
More informationEK307 Active Filters and Steady State Frequency Response
EK307 Active Filters and Steady State Frequency Response Laboratory Goal: To explore the properties of active signalprocessing filters Learning Objectives: Active Filters, OpAmp Filters, Bode plots Suggested
More informationLAB 8: Activity P52: LRC Circuit
LAB 8: Activity P52: LRC Circuit Equipment: Voltage Sensor 1 Multimeter 1 Patch Cords 2 AC/DC Electronics Lab (100 μf capacitor; 10 Ω resistor; Inductor Coil; Iron core; 5 inch wire lead) The purpose of
More informationLab E2: Bfield of a Solenoid. In the case that the Bfield is uniform and perpendicular to the area, (1) reduces to
E2.1 Lab E2: Bfield of a Solenoid In this lab, we will explore the magnetic field created by a solenoid. First, we must review some basic electromagnetic theory. The magnetic flux over some area A is
More informationEK307 Passive Filters and Steady State Frequency Response
EK307 Passive Filters and Steady State Frequency Response Laboratory Goal: To explore the properties of passive signalprocessing filters Learning Objectives: Passive filters, Frequency domain, Bode plots
More informationUniversity of Jordan School of Engineering Electrical Engineering Department. EE 219 Electrical Circuits Lab
University of Jordan School of Engineering Electrical Engineering Department EE 219 Electrical Circuits Lab EXPERIMENT 4 TRANSIENT ANALYSIS Prepared by: Dr. Mohammed Hawa EXPERIMENT 4 TRANSIENT ANALYSIS
More informationET1210: Module 5 Inductance and Resonance
Part 1 Inductors Theory: When current flows through a coil of wire, a magnetic field is created around the wire. This electromagnetic field accompanies any moving electric charge and is proportional to
More informationLab 2: Linear and Nonlinear Circuit Elements and Networks
OPTI 380B Intermediate Optics Laboratory Lab 2: Linear and Nonlinear Circuit Elements and Networks Objectives: Lean how to use: Function of an oscilloscope probe. Characterization of capacitors and inductors
More informationStudy of Inductive and Capacitive Reactance and RLC Resonance
Objective Study of Inductive and Capacitive Reactance and RLC Resonance To understand how the reactance of inductors and capacitors change with frequency, and how the two can cancel each other to leave
More informationIntegrators, differentiators, and simple filters
BEE 233 Laboratory4 Integrators, differentiators, and simple filters 1. Objectives Analyze and measure characteristics of circuits built with opamps. Design and test circuits with opamps. Plot gain vs.
More informationFlorida Atlantic University Biomedical Signal Processing Lab Experiment 2 Signal Transduction: Building an analog Electrocardiogram (ECG)
Florida Atlantic University Biomedical Signal Processing Lab Experiment 2 Signal Transduction: Building an analog Electrocardiogram (ECG) 1. Introduction: The Electrocardiogram (ECG) is a technique of
More informationECE3204 D2015 Lab 1. See suggested breadboard configuration on following page!
ECE3204 D2015 Lab 1 The Operational Amplifier: Inverting and Noninverting Gain Configurations GainBandwidth Product Relationship Frequency Response Limitation Transfer Function Measurement DC Errors
More informationEE 2274 RC and Op Amp Circuit Completed Prior to Coming to Lab. Prelab Part I: RC Circuit
EE 2274 RC and Op Amp Circuit Completed Prior to Coming to Lab Prelab Part I: RC Circuit 1. Design a high pass filter (Fig. 1) which has a break point f b = 1 khz at 3dB below the midband level (the 3dB
More informationFilters And Waveform Shaping
Physics 3330 Experiment #3 Fall 2001 Purpose Filters And Waveform Shaping The aim of this experiment is to study the frequency filtering properties of passive (R, C, and L) circuits for sine waves, and
More informationDC and AC Circuits. Objective. Theory. 1. Direct Current (DC) RC Circuit
[International Campus Lab] Objective Determine the behavior of resistors, capacitors, and inductors in DC and AC circuits. Theory  Reference  Young
More informationDepartment of Electrical & Computer Engineering Technology. EET 3086C Circuit Analysis Laboratory Experiments. Masood Ejaz
Department of Electrical & Computer Engineering Technology EET 3086C Circuit Analysis Laboratory Experiments Masood Ejaz Experiment # 1 DC Measurements of a Resistive Circuit and Proof of Thevenin Theorem
More informationExperiment Guide: RC/RLC Filters and LabVIEW
Description and ackground Experiment Guide: RC/RLC Filters and LabIEW In this lab you will (a) manipulate instruments manually to determine the inputoutput characteristics of an RC filter, and then (b)
More informationLABORATORY 7 v2 BOOST CONVERTER
University of California Berkeley Department of Electrical Engineering and Computer Sciences EECS 100, Professor Bernhard Boser LABORATORY 7 v2 BOOST CONVERTER In many situations circuits require a different
More informationLab 2: Capacitors. Integrator and Differentiator Circuits
Lab 2: Capacitors Topics: Differentiator Integrator LowPass Filter HighPass Filter BandPass Filter Integrator and Differentiator Circuits The simple RC circuits that you built in a previous section
More informationAC Circuits. "Look for knowledge not in books but in things themselves." W. Gilbert ( )
AC Circuits "Look for knowledge not in books but in things themselves." W. Gilbert (15401603) OBJECTIVES To study some circuit elements and a simple AC circuit. THEORY All useful circuits use varying
More informationDigital Applications of the Operational Amplifier
Lab Procedure 1. Objective This project will show the versatile operation of an operational amplifier in a voltage comparator (Schmitt Trigger) circuit and a sample and hold circuit. 2. Components Qty
More informationResonant Frequency of the LRC Circuit (Power Output, Voltage Sensor)
72 Resonant Frequency of the LRC Circuit (Power Output, Voltage Sensor) Equipment List Qty Items Part Numbers 1 PASCO 750 Interface 1 Voltage Sensor CI6503 1 AC/DC Electronics Laboratory EM8656 2 Banana
More informationChapter 2. The Fundamentals of Electronics: A Review
Chapter 2 The Fundamentals of Electronics: A Review Topics Covered 21: Gain, Attenuation, and Decibels 22: Tuned Circuits 23: Filters 24: Fourier Theory 21: Gain, Attenuation, and Decibels Most circuits
More informationPHYSICS 221 LAB #6: CAPACITORS AND AC CIRCUITS
Name: Partners: PHYSICS 221 LAB #6: CAPACITORS AND AC CIRCUITS The electricity produced for use in homes and industry is made by rotating coils of wire in a magnetic field, which results in alternating
More informationExperiment #6: Biasing an NPN BJT Introduction to CE, CC, and CB Amplifiers
SCHOOL OF ENGINEERING AND APPLIED SCIENCE DEPARTMENT OF ELECTRICAL AND COMPUTER ENGINEERING ECE 2115: ENGINEERING ELECTRONICS LABORATORY Experiment #6: Biasing an NPN BJT Introduction to CE, CC, and CB
More informationUncovering a Hidden RCL Series Circuit
Purpose Uncovering a Hidden RCL Series Circuit a. To use the equipment and techniques developed in the previous experiment to uncover a hidden series RCL circuit in a box and b. To measure the values of
More informationLaboratory 4. Bandwidth, Filters, and Diodes
Laboratory 4 Bandwidth, Filters, and Diodes Required Components: k resistor 0. F capacitor N94 smallsignal diode LED 4. Objectives In the previous laboratory exercise you examined the effects of input
More informationSTUDY OF RC AND RL CIRCUITS Venue: Microelectronics Laboratory in E2 L2
EXPERIMENT #1 STUDY OF RC AND RL CIRCUITS Venue: Microelectronics Laboratory in E2 L2 I. INTRODUCTION This laboratory is about verifying the transient behavior of RC and RL circuits. You need to revise
More informationSampling and Reconstruction
Experiment 10 Sampling and Reconstruction In this experiment we shall learn how an analog signal can be sampled in the time domain and then how the same samples can be used to reconstruct the original
More informationRLC Frequency Response
1. Introduction RLC Frequency Response The student will analyze the frequency response of an RLC circuit excited by a sinusoid. Amplitude and phase shift of circuit components will be analyzed at different
More informationVoltage Current and Resistance II
Voltage Current and Resistance II Equipment: Capstone with 850 interface, analog DC voltmeter, analog DC ammeter, voltage sensor, RLC circuit board, 8 male to male banana leads 1 Purpose This is a continuation
More informationLab 4: Transmission Line
1 Introduction Lab 4: Transmission Line In this experiment we will study the properties of a wave propagating in a periodic medium. Usually this takes the form of an array of masses and springs of the
More informationReactance and Impedance
eactance and Impedance Theory esistors, inductors, and capacitors all have the effect of modifying the size of the current in an AC circuit and the time at which the current reaches its maximum value (in
More informationEE 3305 Lab I Revised July 18, 2003
Operational Amplifiers Operational amplifiers are highgain amplifiers with a similar general description typified by the most famous example, the LM741. The LM741 is used for many amplifier varieties
More informationLABORATORY 3 v3 CIRCUIT ELEMENTS
University of California Berkeley Department of Electrical Engineering and Computer Sciences EECS 100, Professor Leon Chua LABORATORY 3 v3 CIRCUIT ELEMENTS The purpose of this laboratory is to familiarize
More informationExperiment 5 The Oscilloscope
Experiment 5 The Oscilloscope Vision is the art of seeing things invisible. J. Swift (16671745) OBJECTIVE To learn to operate a cathode ray oscilloscope. THEORY The oscilloscope, or scope for short, is
More informationPOLYTECHNIC UNIVERSITY Electrical Engineering Department. EE SOPHOMORE LABORATORY Experiment 3 The Oscilloscope
POLYTECHNIC UNIVERSITY Electrical Engineering Department EE SOPHOMORE LABORATORY Experiment 3 The Oscilloscope Modified for Physics 18, Brooklyn College I. Overview of the Experiment The main objective
More informationEquipment: You will use the bench power supply, function generator and oscilloscope.
EE203 Lab #0 Laboratory Equipment and Measurement Techniques Purpose Your objective in this lab is to gain familiarity with the properties and effective use of the lab power supply, function generator
More informationEC5 MAGNETIC INDUCTION
EC5 MAGNETIC INDUCTION If an object is placed in a changing magnetic field, or if an object is moving in a nonuniform magnetic field in such a way that it experiences a changing magnetic field, a voltage
More informationOperational Amplifiers: Part II
1. Introduction Operational Amplifiers: Part II The name "operational amplifier" comes from this amplifier's ability to perform mathematical operations. Three good examples of this are the summing amplifier,
More informationME 365 EXPERIMENT 7 SIGNAL CONDITIONING AND LOADING
ME 365 EXPERIMENT 7 SIGNAL CONDITIONING AND LOADING Objectives: To familiarize the student with the concepts of signal conditioning. At the end of the lab, the student should be able to: Understand the
More informationLab 13 AC Circuit Measurements
Lab 13 AC Circuit Measurements Objectives concepts 1. what is impedance, really? 2. function generator and oscilloscope 3. RMS vs magnitude vs PeaktoPeak voltage 4. phase between sinusoids skills 1.
More informationClass #9: Experiment Diodes Part II: LEDs
Class #9: Experiment Diodes Part II: LEDs Purpose: The objective of this experiment is to become familiar with the properties and uses of LEDs, particularly as a communication device. This is a continuation
More informationMechatronics. Analog and Digital Electronics: Studio Exercises 1 & 2
Mechatronics Analog and Digital Electronics: Studio Exercises 1 & 2 There is an electronics revolution taking place in the industrialized world. Electronics pervades all activities. Perhaps the most important
More informationLaboratory Project 1: Design of a Myogram Circuit
1270 Laboratory Project 1: Design of a Myogram Circuit AbstractYou will design and build a circuit to measure the small voltages generated by your biceps muscle. Using your circuit and an oscilloscope,
More informationCHAPTER 6. Motor Driver
CHAPTER 6 Motor Driver In this lab, we will construct the circuitry that your robot uses to drive its motors. However, before testing the motor circuit we will begin by making sure that you are able to
More informationEE320L Electronics I. Laboratory. Laboratory Exercise #2. Basic OpAmp Circuits. Angsuman Roy. Department of Electrical and Computer Engineering
EE320L Electronics I Laboratory Laboratory Exercise #2 Basic OpAmp Circuits By Angsuman Roy Department of Electrical and Computer Engineering University of Nevada, Las Vegas Objective: The purpose of
More informationAPPENDIX D DISCUSSION OF ELECTRONIC INSTRUMENTS
APPENDIX D DISCUSSION OF ELECTRONIC INSTRUMENTS DC POWER SUPPLIES We will discuss these instruments one at a time, starting with the DC power supply. The simplest DC power supplies are batteries which
More informationExperiment VI: The LRC Circuit and Resonance
Experiment VI: The ircuit and esonance I. eferences Halliday, esnick and Krane, Physics, Vol., 4th Ed., hapters 38,39 Purcell, Electricity and Magnetism, hapter 7,8 II. Equipment Digital Oscilloscope Digital
More information#8A RLC Circuits: Free Oscillations
#8A RL ircuits: Free Oscillations Goals In this lab we investigate the properties of a series RL circuit. Such circuits are interesting, not only for there widespread application in electrical devices,
More informationUNIVERSITY OF CALIFORNIA, BERKELEY. EE40: Introduction to Microelectronic Circuits Lab 1. Introduction to Circuits and Instruments Guide
UNERSTY OF CALFORNA, BERKELEY EE40: ntroduction to Microelectronic Circuits Lab 1 ntroduction to Circuits and nstruments Guide 1. Objectives The electronic circuit is the basis for all branches of electrical
More informationGoals. Introduction. To understand the use of root mean square (rms) voltages and currents.
Lab 10. AC Circuits Goals To show that AC voltages cannot generally be added without accounting for their phase relationships. That is, one must account for how they vary in time with respect to one another.
More informationThe Tuned Circuit. Aim of the experiment. Circuit. Equipment and components. Display of a decaying oscillation. Dependence of L, C and R.
The Tuned Circuit Aim of the experiment Display of a decaying oscillation. Dependence of L, C and R. Circuit Equipment and components 1 Rastered socket panel 1 Resistor R 1 = 10 Ω, 1 Resistor R 2 = 1 kω
More informationLab 6  Inductors and LR Circuits
Lab 6 Inductors and LR Circuits L61 Name Date Partners Lab 6  Inductors and LR Circuits The power which electricity of tension possesses of causing an opposite electrical state in its vicinity has been
More informationET 304A Laboratory TutorialCircuitmaker For Transient and Frequency Analysis
ET 304A Laboratory TutorialCircuitmaker For Transient and Frequency Analysis All circuit simulation packages that use the Pspice engine allow users to do complex analysis that were once impossible to
More informationKanchan S. Shrikhande. Department of Instrumentation Engineering, Vivekanand Education Society s Institute of.
ISOLATED ECG AMPLIFIER WITH RIGHT LEG DRIVE Kanchan S. Shrikhande Department of Instrumentation Engineering, Vivekanand Education Society s Institute of Technology(VESIT),kanchans90@gmail.com Abstract
More informationTransformer Waveforms
OBJECTIVE EXPERIMENT Transformer Waveforms SteadyState Testing and Performance of SinglePhase Transformers Waveforms The voltage regulation and efficiency of a distribution system are affected by the
More informationTTL LOGIC and RING OSCILLATOR TTL
ECE 2274 TTL LOGIC and RING OSCILLATOR TTL We will examine two digital logic inverters. The first will have a passive resistor pullup output stage. The second will have an active transistor and current
More informationUniversity of North Carolina, Charlotte Department of Electrical and Computer Engineering ECGR 3157 EE Design II Fall 2009
University of North Carolina, Charlotte Department of Electrical and Computer Engineering ECGR 3157 EE Design II Fall 2009 Lab 1 Power Amplifier Circuits Issued August 25, 2009 Due: September 11, 2009
More informationExercise 1: Series RLC Circuits
RLC Circuits AC 2 Fundamentals Exercise 1: Series RLC Circuits EXERCISE OBJECTIVE When you have completed this exercise, you will be able to analyze series RLC circuits by using calculations and measurements.
More informationEE 201 Function / Arbitrary Waveform Generator and Oscilloscope Tutorial
EE 201 Function / Arbitrary Waveform Generator and Oscilloscope Tutorial 1 This is a programmed learning instruction manual. It is written for the Agilent DSO3202A Digital Storage Oscilloscope. The prerequisite
More informationExperiment 6: Biasing Circuitry
1 Objective UNIVERSITY OF CALIFORNIA AT BERKELEY College of Engineering Department of Electrical Engineering and Computer Sciences EE105 Lab Experiments Experiment 6: Biasing Circuitry Setting up a biasing
More informationPhysics 120 Lab 1 (2018)  Instruments and DC Circuits
Physics 120 Lab 1 (2018)  Instruments and DC Circuits Welcome to the first laboratory exercise in Physics 120. Your stateofthe art equipment includes: Digital oscilloscope w/usb output for SCREENSHOTS.
More informationBasic Analog Circuits
Basic Analog Circuits Overview This tutorial is part of the National Instruments Measurement Fundamentals series. Each tutorial in this series, will teach you a specific topic of common measurement applications,
More informationTeacher s Guide  Activity P51: LR Circuit (Power Output, Voltage Sensor)
Teacher s Guide  Activity P51: LR Circuit (Power Output, Voltage Sensor) Concept DataStudio ScienceWorkshop (Mac) ScienceWorkshop (Win) Circuits P51 LR Circuit.DS (See end of activity) (See end of activity)
More informationLab 9 AC FILTERS AND RESONANCE
151 Name Date Partners ab 9 A FITES AND ESONANE OBJETIES OEIEW To understand the design of capacitive and inductive filters To understand resonance in circuits driven by A signals In a previous lab, you
More informationExperiment P50: Transistor Lab 3 CommonEmitter Amplifier (Power Amplifier, Voltage Sensor)
PASCO scientific Vol. 2 Physics Lab Manual: P501 Experiment P50: Transistor Lab 3 CommonEmitter Amplifier (Power Amplifier, Voltage Sensor) Concept Time SW Interface Macintosh file Windows file semiconductors
More informationTest No. 2. Advanced Scope Measurements. History. University of Applied Sciences Hamburg. Last chance!! EEL2 No 2
University of Applied Sciences Hamburg Group No : DEPARTMENT OF INFORMATION ENGINEERING Laboratory for Instrumentation and Measurement L1: in charge of the report Test No. 2 Date: Assistant A2: Professor:
More informationThe Series RLC Circuit and Resonance
Purpose Theory The Series RLC Circuit and Resonance a. To study the behavior of a series RLC circuit in an AC current. b. To measure the values of the L and C using the impedance method. c. To study the
More informationEXPERIMENT 2.2 NONLINEAR OPAMP CIRCUITS
2.16 EXPERIMENT 2.2 NONLINEAR OPAMP CIRCUITS 2.2.1 OBJECTIVE a. To study the operation of 741 opamp as comparator. b. To study the operation of active diode circuits (precisions circuits) using opamps,
More informationLABORATORY 5 v3 OPERATIONAL AMPLIFIER
University of California Berkeley Department of Electrical Engineering and Computer Sciences EECS 100, Professor Bernhard Boser LABORATORY 5 v3 OPERATIONAL AMPLIFIER Integrated operational amplifiers opamps
More informationLaboratory Project 1: AC Circuit Measurements and Simulation
Objectives The purpose of this laboratory project is to introduce to equipment, measurement techniques, and simulations commonly used in C circuit analysis. In this laboratory session, each student will:
More informationLRC Circuit PHYS 296 Your name Lab section
LRC Circuit PHYS 296 Your name Lab section PRELAB QUIZZES 1. What will we investigate in this lab? 2. Figure 1 on the following page shows an LRC circuit with the resistor of 1 Ω, the capacitor of 33
More informationTheory: The idea of this oscillator comes from the idea of positive feedback, which is described by Figure 6.1. Figure 6.1: Positive Feedback
Name1 Name2 12/2/10 ESE 319 Lab 6: Colpitts Oscillator Introduction: This lab introduced the concept of feedback in combination with bipolar junction transistors. The goal of this lab was to first create
More informationFREQUENCY RESPONSE AND PASSIVE FILTERS LABORATORY
FREQUENCY RESPONSE AND PASSIVE FILTERS LABORATORY In this experiment we will analytically determine and measure the frequency response of networks containing resistors, AC source/sources, and energy storage
More informationGoals. Introduction. To understand the use of root mean square (rms) voltages and currents.
Lab 10. AC Circuits Goals To show that AC voltages cannot generally be added without accounting for their phase relationships. That is, one must account for how they vary in time with respect to one another.
More informationME 365 EXPERIMENT 1 FAMILIARIZATION WITH COMMONLY USED INSTRUMENTATION
Objectives: ME 365 EXPERIMENT 1 FAMILIARIZATION WITH COMMONLY USED INSTRUMENTATION The primary goal of this laboratory is to study the operation and limitations of several commonly used pieces of instrumentation:
More informationI = I 0 cos 2 θ (1.1)
Chapter 1 Faraday Rotation Experiment objectives: Observe the Faraday Effect, the rotation of a light wave s polarization vector in a material with a magnetic field directed along the wave s direction.
More informationENG 100 Lab #2 Passive FirstOrder Filter Circuits
ENG 100 Lab #2 Passive FirstOrder Filter Circuits In Lab #2, you will construct simple 1 st order RL and RC filter circuits and investigate their frequency responses (amplitude and phase responses).
More informationEE 221 L CIRCUIT II. by Ming Zhu
EE 22 L CIRCUIT II LABORATORY 9: RC CIRCUITS, FREQUENCY RESPONSE & FILTER DESIGNS by Ming Zhu DEPARTMENT OF ELECTRICAL AND COMPUTER ENGINEERING UNIVERSITY OF NEVADA, LAS VEGAS OBJECTIVE Enhance the knowledge
More informationExperiment 7: Undriven & Driven RLC Circuits
MASSACHUSETTS INSTITUTE OF TECHNOLOGY Department of Physics 8.02 Spring 2006 OBJECTIVES Experiment 7: Undriven & Driven RLC Circuits 1. To explore the time dependent behavior of RLC Circuits, both driven
More informationECE 3274 CommonEmitter Amplifier Project
ECE 3274 CommonEmitter Amplifier Project 1. Objective The objective of this lab is to design and build three variations of the common emitter amplifier. 2. Components Qty Device 1 2N2222 BJT Transistor
More informationOpAmp Simulation Part II
OpAmp Simulation Part II EE/CS 5720/6720 This assignment continues the simulation and characterization of a simple operational amplifier. Turn in a copy of this assignment with answers in the appropriate
More informationLab 5: EC3, Capacitors and RCDecay Lab Worksheet
, Capacitors and RCDecay Lab Worksheet Name Your TA will use this sheet to score your lab. It is to be turned in at the end of lab. You must use complete sentences and clearly explain your reasoning to
More informationLAB I. INTRODUCTION TO LAB EQUIPMENT
1. OBJECTIVE LAB I. INTRODUCTION TO LAB EQUIPMENT In this lab you will learn how to properly operate the oscilloscope Agilent MSO6032A, the Keithley Source Measure Unit (SMU) 2430, the function generator
More informationNavy Electricity and Electronics Training Series
NONRESIDENT TRAINING COURSE SEPTEMBER 1998 Navy Electricity and Electronics Training Series Module 9 Introduction to Wave Generation and WaveShaping NAVEDTRA 14181 DISTRIBUTION STATEMENT A: Approved
More informationPower Electronics Laboratory2 Uncontrolled Rectifiers
Roll. No: Checked By: Date: Grade: Power Electronics Laboratory2 and Uncontrolled Rectifiers Objectives: 1. To analyze the working and performance of a and half wave uncontrolled rectifier. 2. To analyze
More informationC and solving for C gives 1 C
Physics 241 Lab RLC Radios http://bohr.physics.arizona.edu/~leone/ua/ua_spring_2010/phys241lab.html Name: Section 1: 1. Begin today by reviewing the experimental procedure for finding C, L and resonance.
More informationLab 5 Second Order Transient Response of Circuits
Lab 5 Second Order Transient Response of Circuits Lab Performed on November 5, 2008 by Nicole Kato, Ryan Carmichael, and Ti Wu Report by Ryan Carmichael and Nicole Kato E11 Laboratory Report Submitted
More informationEXPERIMENT 10: SINGLETRANSISTOR AMPLIFIERS 11/11/10
EXPERIMENT 10: SINGLETRANSISTOR AMPLIFIERS 11/11/10 In this experiment we will measure the characteristics of the standard common emitter amplifier. We will use the 2N3904 npn transistor. If you have
More informationEENG201 Experiment # 4: Function Generator, Oscilloscope
EENG201 Experiment # 4: Function Generator, Oscilloscope I. Objectives Upon completion of this experiment, the student should be able to 1. To become familiar with the use of a function generator. 2.
More informationUniversity Tunku Abdul Rahman LABORATORY REPORT 1
University Tunku Abdul Rahman FACULTY OF ENGINEERING AND GREEN TECHNOLOGY UGEA2523 COMMUNICATION SYSTEMS LABORATORY REPORT 1 Signal Transmission & Distortion Student Name Student ID 1. Low Hui Tyen 14AGB06230
More informationI1 19u 5V R11 1MEG IDC Q7 Q2N3904 Q2N3904. Figure 3.1 A scaled down 741 op amp used in this lab
Lab 3: 74 Op amp Purpose: The purpose of this laboratory is to become familiar with a two stage operational amplifier (op amp). Students will analyze the circuit manually and compare the results with SPICE.
More informationECE 3274 CommonEmitter Amplifier Project
ECE 3274 CommonEmitter Amplifier Project 1. Objective The objective of this lab is to design and build the commonemitter amplifier with partial bypass of the emitter resistor to control the AC voltage
More informationUNIVERSITY OF NORTH CAROLINA AT CHARLOTTE Department of Electrical and Computer Engineering
UNIVERSITY OF NORTH CAROLINA AT CHARLOTTE Department of Electrical and Computer Engineering EXPERIMENT 5 GAINBANDWIDTH PRODUCT AND SLEW RATE OBJECTIVES In this experiment the student will explore two
More informationAC Magnitude and Phase
AC Magnitude and Phase Objectives: oday's experiment provides practical experience with the meaning of magnitude and phase in a linear circuits and the use of phasor algebra to predict the response of
More informationB. Equipment. Advanced Lab
Advanced Lab Measuring Periodic Signals Using a Digital Oscilloscope A. Introduction and Background We will use a digital oscilloscope to characterize several different periodic voltage signals. We will
More informationWaveform Generators and Oscilloscopes. Lab 6
Waveform Generators and Oscilloscopes Lab 6 1 Equipment List WFG TEK DPO 4032A (or MDO3012) Resistors: 10kΩ, 1kΩ Capacitors: 0.01uF 2 Waveform Generators (WFG) The WFG supplies a variety of timevarying
More informationNonideal Behavior of Electronic Components at High Frequencies and Associated Measurement Problems
Nonideal Behavior of Electronic Components at High Frequencies and Associated Measurement Problems Matthew Beckler beck0778@umn.edu EE30 Lab Section 008 October 27, 2006 Abstract In the world of electronics,
More information