8.2 Common Forms of Noise

 Leona Williams
 5 months ago
 Views:
Transcription
1 8.2 Common Forms of Noise Johnson or thermal noise shot or Poisson noise 1/f noise or drift interference noise impulse noise real noise 8.2 : 1/19
2 Johnson Noise Johnson noise characteristics produced by the thermal motion of electrons in resistors the amplitude has a normal pdf with a mean of zero the noise magnitude is the standard deviation, σ, of the pdf the phase has a uniform pdf over π/2 to +π/2 radians noise can be averaged to zero The noise voltage is given by, ej = 4kTRΔf, where k is Boltzmann's constant, T is the temperature in Kelvin, R is the resistance across which the voltage is measured, and Δf is the measurement bandwidth. The graph shows Johnson noise for measurements taken one per second: T = 298 K, R = 1 kω, and Δf = 1 khz. The blue lines are ±2.5σ. voltage Johnson Noise Voltage 8.2 : 2/ time
3 Johnson Noise Spectrum Noise amplitude versus frequency (the noise spectrum) looks much the same as noise amplitude versus time. Noise phase is also random in time and frequency Noise Amplitude vs. Frequency Noise Phase vs. Frequenc y voltage phase frequency frequency 8.2 : 3/19
4 Johnson Noise Power In electronics, power can be computed as: P = ei = e 2 /R. Using the equation for Johnson noise voltage, the Johnson noise power is given by P J = 4kTΔf. Note that resistance has dropped out of the equation. Below is the noise power for T = 298 K and Δf = 1 khz. The solid blue line in time is theory from the above equation. The area under the blue line in frequency equals the temporal amplitude theory value. Johnson Noise Power Noise Power vs. Frequency power (W) power per bandwidth (W/Delta f) time frequency 8.2 : 4/19
5 Reducing Johnson Noise Johnson noise power can be reduced by changing the temperature. Not much is gained unless liquid helium is used as the coolant. That is, for liquid nitrogen, 77 K/298 K =.26; for liquid helium, 4 K/298 K =.13. Johnson noise power can be reduced by using a smaller bandwidth. Values down to.1 Hz are practical. However, Fourier transforms tell you that a.1 Hz bandwidth will require at least a 1 s measurement time! It doesn't matter where in frequency the bandwidth is located. A 1 khz bandwidth from to 1 khz has the same noise power as that from 1. MHz to 1.1 MHz. Johnson noise is simulated by adding normally distributed random voltages to the true value. Set the mean of the normal pdf to zero and the standard deviation to the noise voltage, e J. 8.2 : 5/19
6 Electronic Shot Noise Electronic shot noise characteristics produced by Poisson fluctuations in the flow of electrons when discrete charges move across a junction such as those found in semiconductors or a cathode and anode in vacuum tubes the noise is proportional to the square root of the average current for average currents above ~1 pa the noise spectrum looks like Johnson noise, except for a spike at f = due to the average current the phase has a uniform pdf over π/2 to π/2 radians The noise current is given by, ishot = 2qiavgΔf, where q is the charge on an electron, i avg is the average current, Δf is the measurement bandwidth where 1/(2Δf ) = t, the measurement time. (The factor of two comes from the Nyquist theorem which will be explained when analogtodigital converters are covered.) For currents down to ~1 pa, shot noise is less than 1% of the average. Circuits composed of nonsemiconductor components have far less shot noise than predicted by the above equation. 8.2 : 6/19
7 Photon and Ion Shot Noise When detecting individual photons or ions, the current output comes in discrete, countable packets of charge, Q. The average current depends upon the count rate, N R, i avg = N R Q The noise follows Poisson statistics for the counted particle, thus depends upon the square root of the total counts, N 1/2 = (N R t) 1/2. The shot noise charge is given by the noise counts times Q. Q shot = (N 1/2 )Q =(N R t) 1/2 Q = (Q 2 N R t) 1/2 = (Qi avg t) 1/2 Finally, Q shot is converted into a noise current by dividing by time. i shot = Q shot /t = (Qi avg /t) 1/2 = (2Qi avg Δf) 1/2 Note that this is the same equation as electronic shot noise, except for the use of a charge packet, Q, instead of the charge on one electron, q. 8.2 : 7/19
8 Example of Photon Detection A typical photon counting photomultiplier has a gain of 1 8, that is, each photon that ejects an electron from the photocathode produces a pulse containing 1 8 electrons. A typical impulse response of such a detector would be ~5 ns FWHM Gaussian pulse. Consider an optical signal having 1, photons per second. photon counting noise Each charge packet has a peak current given by gain, charge on the electron, and impulse FWHM: /5 19 = ~3.2 ma. Such a pulse is very easy to detect, thus the counts in one second would be 1 3. The Poisson noise would be (1 3 ) 1/2 = ~32. current noise If a picoammeter is used to measure the detector output, it will see an average current: /1 = A. The noise current can be computed using the equation on the last slide. 8.2 : 8/19 i shot Qiavg = = = A t 1
9 Shot Noise Spectrum Noise Amplitude vs. Time Shot noise differs from Johnson noise by having a nonzero mean. This is shown at the left. Except for f = (the mean), the amplitude and phase spectra look like Johnson noise. In the left graph below the amplitude at f = is A. current time Noise Amplitude vs. Frequency Noise Phase vs. Frequenc y current phase : 9/19 frequency frequency
10 Reducing Shot Noise the reduction of shot noise is generally not of interest, this is because a reduction in noise implies a reduction in the mean signaltonoise enhancement is the goal with shot noise iavg NQ t SNR = = = N = NRt i NQ t shot for a fixed counting period, the count rate can be increased by improving the measurement efficiency, e.g. lens f/# with fluorescence the counting time can be increased (the bandwidth decreased) With small counts shot noise is simulated by using a Poisson random number generator. With large counts shot noise can be simulated with a normal random number generator having the standard deviation set equal to the square root of the mean. 8.2 : 1/19
11 Oneoverf Noise (1/f) 1/f noise characteristics appears as drift in a measurement it can be introduced by long term power supply fluctuations, changes in component values, temperature drifts, etc. the longer the time required for a measurement, the more effort needs to expended to keep everything under control. the presence of 1/f noise makes measurements near zero frequency very difficult this noise is generally unimportant above ~1 khz Measured 1/f noise can be loosely approximated by P 1/f a = b + f where a and b are adjustable parameters. Note that noise amplitude will follow the square root of the power. 8.2 : 11/19
12 Example 1/f Noise Spectrum At the right is the 1/f power spectrum with a = 1 and b =.1. Below left is the temporal noise corresponding to the noise spectrum. The low noise frequencies cause drift. By comparing 1/f noise to temporal Johnson noise (at the right) it is possible to see that the high frequencies are attenuated. power /f Noise vs. Frequency frequency /f Amplitude Noise Comparison of 1/f and JohnsonNoise voltage voltage : 12/19 time time
13 Reducing 1/f Noise 1/f noise can be reduced if the measurement can be divided by a reference signal. For example, laser excited fluorescence intensity can be divided by the average laser power. the effect of 1/f noise on a measurement can be reduced by interspersing working standards with the sample the effect of 1/f noise can be reduced or eliminated by modulating the signal to frequencies above ~1 khz 1/f noise is simulated by the following procedure generate a vector of Johnson noise in the time domain use Mathcad's CFFT( ) function to generate a Johnson noise spectrum multiply the Johnson noise spectrum by (a/(f+b)) 1/2, where a/b controls the noise power at f =, and b controls the relative contributions of low to high frequencies (small b favors low frequencies) use Mathcad's ICFFT( ) function to generate 1/f voltage noise in the time domain 8.2 : 13/19
14 Interference Noise Interference noise characteristics interference noise appears within a very narrow range of frequencies often appears as a temporal harmonic wave caused by bad shielding of electrical cables, dc power supply ripple, noise on the 11 V power, 12 Hz frequency of fluorescent lights, etc. Shown below is a temporal Gaussian signal with superimposed interference noise at.5 Hz. Temporal Signal Plus Interfering Cosine Spectrum of Signal Plus Interfering Cosine 1.2 voltage.5 voltage : 14/ time frequency
15 Reducing Interference Noise Reducing interference noise if interference noise is wellseparated spectrally from the signal, it can usually be reduced or eliminated by electronic band pass (signal) and/or band reject filters (noise). if interference noise is not separated spectrally, it can often be reduced or eliminated by using modulation to shift the signal frequencies cable pickup and/or power supply ripple can be eliminated by proper choice of electronic components pickup off the 11 V power line can be reduced or eliminated by the use of 6 Hz notchpass filters (from a company called CorCom) digital filters can easily postprocess your data and remove sinusoidal interference 8.2 : 15/19
16 Impulse Noise Impulse noise characteristics impulse noise appears as a very sharp spike in the time domain the spectrum of impulse noise is very broad sources of impulse noise are rapid electrical discharges, such as lightening and pulsed laser power supplies, or rapid discharge of capacitors Shown below is a temporal Gaussian with an impulse interference at 125 s. Temporal Signal Plus Impulse Noise Spectrum of Signal Plus Impulse Noise 1 voltage.5 voltage : 16/19 time frequency
17 Reducing Impulse Noise Reducing impulse noise frequencybased techniques cannot be used to reduce or eliminate temporal impulse noise If the noise and signal are temporally separated, the measurement can be terminated during the noise spike (for example, an impulse followed by an exponentially decaying signal). This is accomplished with an electronic device called a boxcar integrator. if the noise and signal are temporally overlapped, the data can be postprocessed with an algorithm that only permits small changes in signal amplitude (this would be premised on the anticipated maximum rate of change for the signal) if the signal can be repeated and the impulse is random in time, the impulses can be recognized by an algorithm and removed 8.2 : 17/19
18 Noise Summary type time domain frequency domain phase Johnson random, uniform mean of zero random, uniform mean of zero random shot random, uniform nonzero mean random uniform except impulse at f = random except at f = 1/ f random drift 1/ f random interference cosine impulse fixed impulse impulse all frequencies nonrandom 8.2 : 18/19
19 Real Noise 8.2 : 19/19 Measured noise can consist of all mentioned types. The total noise is determined by adding together noise power (noise amplitude is a standard deviation, thus noise power is a variance). Johnson + 1/f (very common): modulate the signal to frequencies greater than ~1 khz to avoid 1/f noise and use a narrow band pass filter to reduce Johnson noise Shot + 1/f (common in counting experiments): ratio the measurement to reduce 1/f and count for longer periods of time Johnson + interference: modulate the signal to a clean part of the spectrum and use a narrow band pass filter impulse noise + Johnson noise: use a boxcar integrator to eliminate the impulse noise and average the boxcar output to reduce the Johnson noise
Homework Set 3.5 Sensitive optoelectronic detectors: seeing single photons
Homework Set 3.5 Sensitive optoelectronic detectors: seeing single photons Due by 12:00 noon (in class) on Tuesday, Nov. 7, 2006. This is another hybrid lab/homework; please see Section 3.4 for what you
More information8.5 Modulation of Signals
8.5 Modulation of Signals basic idea and goals measuring atomic absorption without modulation measuring atomic absorption with modulation the tuned amplifier, diode rectifier and low pass the lockin amplifier
More informationNoise Lecture 1. EEL6935 Chris Dougherty (TA)
Noise Lecture 1 EEL6935 Chris Dougherty (TA) An IEEE Definition of Noise The IEEE Standard Dictionary of Electrical and Electronics Terms defines noise (as a general term) as: unwanted disturbances superposed
More informationModule 10 : Receiver Noise and Bit Error Ratio
Module 10 : Receiver Noise and Bit Error Ratio Lecture : Receiver Noise and Bit Error Ratio Objectives In this lecture you will learn the following Receiver Noise and Bit Error Ratio Shot Noise Thermal
More informationEENG473 Mobile Communications Module 3 : Week # (12) Mobile Radio Propagation: SmallScale Path Loss
EENG473 Mobile Communications Module 3 : Week # (12) Mobile Radio Propagation: SmallScale Path Loss Introduction Smallscale fading is used to describe the rapid fluctuation of the amplitude of a radio
More information2013 LMIC Imaging Workshop. Sidney L. Shaw Technical Director.  Light and the Image  Detectors  Signal and Noise
2013 LMIC Imaging Workshop Sidney L. Shaw Technical Director  Light and the Image  Detectors  Signal and Noise The Anatomy of a Digital Image Representative Intensities Specimen: (molecular distribution)
More informationME scope Application Note 01 The FFT, Leakage, and Windowing
INTRODUCTION ME scope Application Note 01 The FFT, Leakage, and Windowing NOTE: The steps in this Application Note can be duplicated using any Package that includes the VES3600 Advanced Signal Processing
More informationRead pp Impossible to detect a signal when the S/N becomes less than about 2. mean Std. Deviation. X s S N 1 RSD
Chapter 5 Signals and Noise Read pp. 110123 Impossible to detect a signal when the S/N becomes less than about 2. S N = mean Std. Deviation = X s = 1 RSD Signal Transduction Internet: Wikipedia (Signals
More informationSpectrum Analysis: The FFT Display
Spectrum Analysis: The FFT Display Equipment: Capstone, voltage sensor 1 Introduction It is often useful to represent a function by a series expansion, such as a Taylor series. There are other series representations
More informationFrequency Domain Representation of Signals
Frequency Domain Representation of Signals The Discrete Fourier Transform (DFT) of a sampled time domain waveform x n x 0, x 1,..., x 1 is a set of Fourier Coefficients whose samples are 1 n0 X k X0, X
More informationComponents of Optical Instruments
Components of Optical Instruments General Design of Optical Instruments Sources of Radiation Wavelength Selectors (Filters, Monochromators, Interferometers) Sample Containers Radiation Transducers (Detectors)
More informationProblems from the 3 rd edition
(2.11) Find the energies of the signals: a) sin t, 0 t π b) sin t, 0 t π c) 2 sin t, 0 t π d) sin (t2π), 2π t 4π Problems from the 3 rd edition Comment on the effect on energy of sign change, time shifting
More informationInvestigate the characteristics of PIN Photodiodes and understand the usage of the Lightwave Analyzer component.
PIN Photodiode 1 OBJECTIVE Investigate the characteristics of PIN Photodiodes and understand the usage of the Lightwave Analyzer component. 2 PRELAB In a similar way photons can be generated in a semiconductor,
More informationWorking in Visible NHMFL
Working in Visible Optics @ NHMFL NHMFL Summer School 05192016 Stephen McGill Optical Energy Range Energy of Optical Spectroscopy Range SCM3 Optics Facility Energy Range of Optical Spectroscopy SCM3
More informationFigure Figure E E09. Dark Current (A) 1.
OSI Optoelectronics, is a leading manufacturer of fiber optic components for communication systems. The products offer range for Silicon, GaAs and InGaAs to full turnkey solutions. Photodiodes are semiconductor
More informationNONAMPLIFIED PHOTODETECTOR USER S GUIDE
NONAMPLIFIED PHOTODETECTOR USER S GUIDE Thank you for purchasing your Nonamplified Photodetector. This user s guide will help answer any questions you may have regarding the safe use and optimal operation
More informationLaboratory Assignment 5 Amplitude Modulation
Laboratory Assignment 5 Amplitude Modulation PURPOSE In this assignment, you will explore the use of digital computers for the analysis, design, synthesis, and simulation of an amplitude modulation (AM)
More informationModerne Teilchendetektoren  Theorie und Praxis 2. Dr. Bernhard Ketzer Technische Universität München SS 2013
Moderne Teilchendetektoren  Theorie und Praxis 2 Dr. Bernhard Ketzer Technische Universität München SS 2013 7 Signal Processing and Acquisition 7.1 Signals 7.2 Amplifier 7.3 Electronic Noise 7.4 AnalogtoDigital
More informationUniversity Tunku Abdul Rahman LABORATORY REPORT 1
University Tunku Abdul Rahman FACULTY OF ENGINEERING AND GREEN TECHNOLOGY UGEA2523 COMMUNICATION SYSTEMS LABORATORY REPORT 1 Signal Transmission & Distortion Student Name Student ID 1. Low Hui Tyen 14AGB06230
More informationFilters And Waveform Shaping
Physics 3330 Experiment #3 Fall 2001 Purpose Filters And Waveform Shaping The aim of this experiment is to study the frequency filtering properties of passive (R, C, and L) circuits for sine waves, and
More informationAcoustics and Fourier Transform Physics Advanced Physics Lab  Summer 2018 Don Heiman, Northeastern University, 1/12/2018
1 Acoustics and Fourier Transform Physics 3600  Advanced Physics Lab  Summer 2018 Don Heiman, Northeastern University, 1/12/2018 I. INTRODUCTION Time is fundamental in our everyday life in the 4dimensional
More informationTiming Noise Measurement of HighRepetitionRate Optical Pulses
564 Timing Noise Measurement of HighRepetitionRate Optical Pulses Hidemi Tsuchida National Institute of Advanced Industrial Science and Technology 111 Umezono, Tsukuba, 3058568 JAPAN Tel: 81298615342;
More informationBiomedical Signals. Signals and Images in Medicine Dr Nabeel Anwar
Biomedical Signals Signals and Images in Medicine Dr Nabeel Anwar Noise Removal: Time Domain Techniques 1. Synchronized Averaging (covered in lecture 1) 2. Moving Average Filters (today s topic) 3. Derivative
More informationThe quality of the transmission signal The characteristics of the transmission medium. Some type of transmission medium is required for transmission:
Data Transmission The successful transmission of data depends upon two factors: The quality of the transmission signal The characteristics of the transmission medium Some type of transmission medium is
More informationPART I: The questions in Part I refer to the aliasing portion of the procedure as outlined in the lab manual.
Lab. #1 Signal Processing & Spectral Analysis Name: Date: Section / Group: NOTE: To help you correctly answer many of the following questions, it may be useful to actually run the cases outlined in the
More informationAnnouncements : Wireless Networks Lecture 3: Physical Layer. Bird s Eye View. Outline. Page 1
Announcements 18759: Wireless Networks Lecture 3: Physical Layer Please start to form project teams» Updated project handout is available on the web site Also start to form teams for surveys» Send mail
More informationPreliminary simulation study of the frontend electronics for the central detector PMTs
Angra Neutrino Project AngraNote 127 (Draft) Preliminary simulation study of the frontend electronics for the central detector PMTs A. F. Barbosa Centro Brasileiro de Pesquisas Fsicas  CBPF, email:
More informationAPPLICATION NOTE. Making Accurate Voltage Noise and Current Noise Measurements on Operational Amplifiers Down to 0.1Hz. Abstract
APPLICATION NOTE Making Accurate Voltage Noise and Current Noise Measurements on Operational Amplifiers Down to 0.1Hz AN1560 Rev.1.00 Abstract Making accurate voltage and current noise measurements on
More informationWIRELESS COMMUNICATION TECHNOLOGIES (16:332:546) LECTURE 5 SMALL SCALE FADING
WIRELESS COMMUNICATION TECHNOLOGIES (16:332:546) LECTURE 5 SMALL SCALE FADING Instructor: Dr. Narayan Mandayam Slides: SabarishVivek Sarathy A QUICK RECAP Why is there poor signal reception in urban clutters?
More informationLecture 2 Physical Layer  Data Transmission
DATA AND COMPUTER COMMUNICATIONS Lecture 2 Physical Layer  Data Transmission Mei Yang Based on Lecture slides by William Stallings 1 DATA TRANSMISSION The successful transmission of data depends on two
More informationMassachusetts Institute of Technology Dept. of Electrical Engineering and Computer Science Fall Semester, Introduction to EECS 2
Massachusetts Institute of Technology Dept. of Electrical Engineering and Computer Science Fall Semester, 2006 6.082 Introduction to EECS 2 Modulation and Demodulation Introduction A communication system
More information1/14. Signal. Surasak Sanguanpong Last updated: 11 July Signal 1/14
1/14 Signal Surasak Sanguanpong nguan@ku.ac.th http://www.cpe.ku.ac.th/~nguan Last updated: 11 July 2000 Signal 1/14 Transmission structure 2/14 Transmitter/ Receiver Medium Amplifier/ Repeater Medium
More informationShielding. Fig. 6.1: Using a Steel Paint Can
Analysis and Measurement of Intrinsic Noise in Op Amp Circuits Part VI: Noise Measurement Examples by Art Kay, Senior Applications Engineer, Texas Instruments Incorporated In Part IV we introduced the
More informationThe Fundamentals of FFTBased Signal Analysis and Measurement Michael Cerna and Audrey F. Harvey
Application ote 041 The Fundamentals of FFTBased Signal Analysis and Measurement Michael Cerna and Audrey F. Harvey Introduction The Fast Fourier Transform (FFT) and the power spectrum are powerful tools
More informationChemistry Instrumental Analysis Lecture 10. Chem 4631
Chemistry 4631 Instrumental Analysis Lecture 10 Types of Instrumentation Single beam Double beam in space Double beam in time Multichannel Speciality Types of Instrumentation Single beam Requires stable
More informationLecture 3: Data Transmission
Lecture 3: Data Transmission 1 st semester 14392017 1 By: Elham Sunbu OUTLINE Data Transmission DATA RATE LIMITS Transmission Impairments Examples DATA TRANSMISSION The successful transmission of data
More informationEqualization. Isolated Pulse Responses
Isolated pulse responses Pulse spreading Group delay variation Equalization Equalization Magnitude equalization Phase equalization The Comlinear CLC014 Equalizer Equalizer bandwidth and noise Bit error
More informationAntenna Measurements using Modulated Signals
Antenna Measurements using Modulated Signals Roger Dygert MI Technologies, 1125 Satellite Boulevard, Suite 100 Suwanee, GA 300244629 Abstract Antenna test engineers are faced with testing increasingly
More informationFrequency Domain Analysis
1 Frequency Domain Analysis Concerned with analysing the frequency (wavelength) content of a process Application example: Electromagnetic Radiation: Represented by a Frequency Spectrum: plot of intensity
More informationIntroduction to Telecommunications and Computer Engineering Unit 3: Communications Systems & Signals
Introduction to Telecommunications and Computer Engineering Unit 3: Communications Systems & Signals Syedur Rahman Lecturer, CSE Department North South University syedur.rahman@wolfson.oxon.org Acknowledgements
More informationAdvancement in development of photomultipliers dedicated to new scintillators studies.
Advancement in development of photomultipliers dedicated to new scintillators studies. Maciej Kapusta, Pascal Lavoutea, Florence Lherbet, Cyril Moussant, Paul Hink INTRODUCTION AND OUTLINE In the validation
More informationOutline / Wireless Networks and Applications Lecture 3: Physical Layer Signals, Modulation, Multiplexing. Cartoon View 1 A Wave of Energy
Outline 18452/18750 Wireless Networks and Applications Lecture 3: Physical Layer Signals, Modulation, Multiplexing Peter Steenkiste Carnegie Mellon University Spring Semester 2017 http://www.cs.cmu.edu/~prs/wirelesss17/
More informationORTEC. TimetoAmplitude Converters and Time Calibrator. Choosing the Right TAC. Timing with TACs
ORTEC TimetoAmplitude Converters Choosing the Right TAC The following topics provide the information needed for selecting the right timetoamplitude converter (TAC) for the task. The basic principles
More informationMedia. Twisted pair db/km at 1MHz 2 km. Coaxial cable 7 db/km at 10 MHz 1 9 km. Optical fibre 0.2 db/km 100 km
Media Attenuation Repeater spacing Twisted pair 1012 db/km at 1MHz 2 km Coaxial cable 7 db/km at 10 MHz 1 9 km Optical fibre 0.2 db/km 100 km conniq.com provides an excellent tutorial on physical media.
More informationModule 12 : System Degradation and Power Penalty
Module 12 : System Degradation and Power Penalty Lecture : System Degradation and Power Penalty Objectives In this lecture you will learn the following Degradation during Propagation Modal Noise Dispersion
More information6.S02 MRI Lab Acquire MR signals. 2.1 Free Induction decay (FID)
6.S02 MRI Lab 1 2. Acquire MR signals Connecting to the scanner Connect to VMware on the Lab Macs. Download and extract the following zip file in the MRI Lab dropbox folder: https://www.dropbox.com/s/ga8ga4a0sxwe62e/mit_download.zip
More informationInfrared Detectors an overview
Infrared Detectors an overview Mariangela Cestelli Guidi Sinbad IR beamline @ DaFne EDIT 2015, October 22 Frederick William Herschel (1738 1822) was born in Hanover, Germany but emigrated to Britain at
More informationPHYSICS 330 LAB Operational Amplifier Frequency Response
PHYSICS 330 LAB Operational Amplifier Frequency Response Objectives: To measure and plot the frequency response of an operational amplifier circuit. History: Operational amplifiers are among the most widely
More informationMuhammad Ali Jinnah University, Islamabad Campus, Pakistan. Fading Channel. Base Station
Fading Lecturer: Assoc. Prof. Dr. Noor M Khan Department of Electronic Engineering, Muhammad Ali Jinnah University, Islamabad Campus, Islamabad, PAKISTAN Ph: +9 (51) 111878787, Ext. 19 (Office), 186 (ARWiC
More informationPart A: Spread Spectrum Systems
1 Telecommunication Systems and Applications (TL  424) Part A: Spread Spectrum Systems Dr. ir. Muhammad Nasir KHAN Department of Electrical Engineering Swedish College of Engineering and Technology February
More informationBiomedical Instrumentation B2. Dealing with noise
Biomedical Instrumentation B2. Dealing with noise B18/BME2 Dr Gari Clifford Noise & artifact in biomedical signals Ambient / power line interference: 50 ±0.2 Hz mains noise (or 60 Hz in many data sets)
More information6.02 Practice Problems: Modulation & Demodulation
1 of 12 6.02 Practice Problems: Modulation & Demodulation Problem 1. Here's our "standard" modulationdemodulation system diagram: at the transmitter, signal x[n] is modulated by signal mod[n] and the
More informationSignal Characteristics
Data Transmission The successful transmission of data depends upon two factors:» The quality of the transmission signal» The characteristics of the transmission medium Some type of transmission medium
More informationCourse 2: Channels 1 1
Course 2: Channels 1 1 "You see, wire telegraph is a kind of a very, very long cat. You pull his tail in New York and his head is meowing in Los Angeles. Do you understand this? And radio operates exactly
More informationEIE339 Digital Transmission and Switching Systems
EIE339 Digital Transmission and Switching Systems Lecturer: Dr. W.Y.Tam Office: DE604 Telephone no.: 666665 email address: enwytam@polyu.edu.hk Continuous Assessment Tests 5% Assignments and quizzes 5%
More informationLocal Oscillator Phase Noise and its effect on Receiver Performance C. John Grebenkemper
WatkinsJohnson Company Technotes Copyright 1981 WatkinsJohnson Company Vol. 8 No. 6 November/December 1981 Local Oscillator Phase Noise and its effect on Receiver Performance C. John Grebenkemper All
More informationEE 791 EEG5 Measures of EEG Dynamic Properties
EE 791 EEG5 Measures of EEG Dynamic Properties Computer analysis of EEG EEG scientists must be especially wary of mathematics in search of applications after all the number of ways to transform data is
More informationData Communications & Computer Networks
Data Communications & Computer Networks Chapter 3 Data Transmission Fall 2008 Agenda Terminology and basic concepts Analog and Digital Data Transmission Transmission impairments Channel capacity Home Exercises
More informationEXAMINATION FOR THE DEGREE OF B.E. and M.E. Semester
EXAMINATION FOR THE DEGREE OF B.E. and M.E. Semester 2 2009 101908 OPTICAL COMMUNICATION ENGINEERING (Elec Eng 4041) 105302 SPECIAL STUDIES IN MARINE ENGINEERING (Elec Eng 7072) Official Reading Time:
More informationThe Fundamentals of Mixed Signal Testing
The Fundamentals of Mixed Signal Testing Course Information The Fundamentals of Mixed Signal Testing course is designed to provide the foundation of knowledge that is required for testing modern mixed
More informationFundamental Noise and Fundamental Constants
Fundamental Noise and Fundamental Constants In this experiment a series of measurements of the Johnson noise produced by several resistors is used to derive Boltzmann s constant, and the shot noise produced
More informationWireless Channel Propagation Model Smallscale Fading
Wireless Channel Propagation Model Smallscale Fading Basic Questions T x What will happen if the transmitter  changes transmit power?  changes frequency?  operates at higher speed? Transmit power,
More informationElectronic Instrumentation. Experiment 8: Diodes (continued) Project 4: Optical Communications Link
Electronic Instrumentation Experiment 8: Diodes (continued) Project 4: Optical Communications Link Agenda Brief Review: Diodes Zener Diodes Project 4: Optical Communication Link Why optics? Understanding
More informationBasic Concepts in Data Transmission
Basic Concepts in Data Transmission EE450: Introduction to Computer Networks Professor A. Zahid A.ZahidEE450 1 Data and Signals Data is an entity that convey information Analog Continuous values within
More informationElements of Communication System Channel Fig: 1: Block Diagram of Communication System Terminology in Communication System
Content: Fundamentals of Communication Engineering : Elements of a Communication System, Need of modulation, electromagnetic spectrum and typical applications, Unit V (Communication terminologies in communication
More informationUnderstanding Noise Figure
Understanding Noise Figure Iulian Rosu, YO3DAC / VA3IUL, http://www.qsl.net/va3iul One of the most frequently discussed forms of noise is known as Thermal Noise. Thermal noise is a random fluctuation in
More informationAC Theory and Electronics
AC Theory and Electronics An Alternating Current (AC) or Voltage is one whose amplitude is not constant, but varies with time about some mean position (value). Some examples of AC variation are shown below:
More informationRemoval of Line Noise Component from EEG Signal
1 Removal of Line Noise Component from EEG Signal Removal of Line Noise Component from EEG Signal When carrying out timefrequency analysis, if one is interested in analysing frequencies above 30Hz (i.e.
More informationSAMPLING THEORY. Representing continuous signals with discrete numbers
SAMPLING THEORY Representing continuous signals with discrete numbers Roger B. Dannenberg Professor of Computer Science, Art, and Music Carnegie Mellon University ICM Week 3 Copyright 20022013 by Roger
More informationFrequency Multiplier (using PLL 565)
Frequency Multiplier (using PLL 565) In electronics, a frequency multiplier is an electronic circuit that generates an output signal whose output frequency is a harmonic (multiple) of its input frequency.
More informationNew Features of IEEE Std Digitizing Waveform Recorders
New Features of IEEE Std 10572007 Digitizing Waveform Recorders William B. Boyer 1, Thomas E. Linnenbrink 2, Jerome Blair 3, 1 Chair, Subcommittee on Digital Waveform Recorders Sandia National Laboratories
More informationNoise and Interference Limited Systems
Chapter 3 Noise and Interference Limited Systems 47 Basics of link budgets Link budgets show how different components and propagation processes influence the available SNR Link budgets can be used to compute
More informationSampling and Signal Processing
Sampling and Signal Processing Sampling Methods Sampling is most commonly done with two devices, the sampleandhold (S/H) and the analogtodigitalconverter (ADC) The S/H acquires a continuoustime signal
More informationUnraveling Zero Crossing and Full Spectrum What does it all mean?
Unraveling Zero Crossing and Full Spectrum What does it all mean? Ian Agranat Wildlife Acoustics, Inc. 2 nd Symposium on Bat Echolocation Research, Tucson AZ March 29, 2017 Let s start with a sound wave
More informationJohnson Noise and the Boltzmann Constant
Johnson Noise and the Boltzmann Constant 1 Introduction The purpose of this laboratory is to study Johnson Noise and to measure the Boltzmann constant k. You will also get use a lownoise preamplifier,
More informationMercury Cadmium Telluride Detectors
Mercury Cadmium Telluride Detectors ISO 9001 Certified J15 Mercury Cadmium Telluride Detectors (2 to 26 µm) General HgCdTe is a ternary semiconductor compound which exhibits a wavelength cutoff proportional
More informationLecture 3 Complex Exponential Signals
Lecture 3 Complex Exponential Signals Fundamentals of Digital Signal Processing Spring, 2012 WeiTa Chu 2012/3/1 1 Review of Complex Numbers Using Euler s famous formula for the complex exponential The
More informationScintillators as an external trigger for cathode strip chambers
Scintillators as an external trigger for cathode strip chambers J. A. Muñoz Department of Physics, Princeton University, Princeton, NJ 08544 An external trigger was set up to test cathode strip chambers
More informationChapter 2: Digitization of Sound
Chapter 2: Digitization of Sound Acoustics pressure waves are converted to electrical signals by use of a microphone. The output signal from the microphone is an analog signal, i.e., a continuousvalued
More informationAudio Noise Figure Meter
Audio Noise Figure Meter Abstract Low noise amplifiers in the audio range are used in many applications. The definition of 'lownoise' is very flexible and poorly defined so any experimenter in this field
More informationKeyser, Ronald M., Twomey, Timothy R., and Bingham, Russell D. ORTEC, 801 South Illinois Avenue, Oak Ridge, TN 37831s
Improved Performance in Germanium Detector Gamma Spectrometers based on Digital Signal Processing Keyser, Ronald M., Twomey, Timothy R., and Bingham, Russell D. ORTEC, 801 South Illinois Avenue, Oak Ridge,
More information781/ /
781/32947 781/4613113 SPECIFICATIONS DC SPECIFICATIONS J Parameter Min Typ Max Units SAMPLING CHARACTERISTICS Acquisition Time 5 V Step to.1% 25 375 ns 5 V Step to.1% 2 35 ns Small Signal Bandwidth 15
More informationMeasure the rolloff frequency of an acoustooptic modulator
Slide 1 Goals of the Lab: Get to know some of the properties of pin photodiodes Measure the rolloff frequency of an acoustooptic modulator Measure the cutoff frequency of a pin photodiode as a function
More informationPhysical Layer: Outline
18345: Introduction to Telecommunication Networks Lectures 3: Physical Layer Peter Steenkiste Spring 2015 www.cs.cmu.edu/~prs/netsece Physical Layer: Outline Digital networking Modulation Characterization
More informationComponents of Optical Instruments. Chapter 7_III UV, Visible and IR Instruments
Components of Optical Instruments Chapter 7_III UV, Visible and IR Instruments 1 Grating Monochromators Principle of operation: Diffraction Diffraction sources: grooves on a reflecting surface Fabrication:
More informationCHAPTER 9 POSITION SENSITIVE PHOTOMULTIPLIER TUBES
CHAPTER 9 POSITION SENSITIVE PHOTOMULTIPLIER TUBES The current multiplication mechanism offered by dynodes makes photomultiplier tubes ideal for lowlightlevel measurement. As explained earlier, there
More informationLecture 8 Fiber Optical Communication Lecture 8, Slide 1
Lecture 8 Bit error rate The Q value Receiver sensitivity Sensitivity degradation Extinction ratio RIN Timing jitter Chirp Forward error correction Fiber Optical Communication Lecture 8, Slide Bit error
More informationEE 318 Electronic Design Lab. Hifi Audio Transmitter from first principles
EE 318 Electronic Design Lab Hifi Audio Transmitter from first principles Supervised by Prof. Jayanta Mukherjee Prof. Dipankar Prof. L. Subramaniam By Group9 Vipul Chaudhary (08d07039) Vineet Raj (08d07040)
More informationFundamentals of Microelectronics
Fundamentals of Microelectronics CH1 Why Microelectronics? CH2 Basic Physics of Semiconductors CH3 Diode Circuits CH4 Physics of Bipolar Transistors CH5 Bipolar Amplifiers CH6 Physics of MOS Transistors
More information5. Scintillation counters
5. Scintillation counters to detect radiation by means of scintillation is among oldest methods of particle detection historical example: particle impinging on ZnS screen > emission of light flash principle
More informationExperiment 2 Effects of Filtering
Experiment 2 Effects of Filtering INTRODUCTION This experiment demonstrates the relationship between the time and frequency domains. A basic rule of thumb is that the wider the bandwidth allowed for the
More informationTone decoder/phaselocked loop
NE/SE DESCRIPTION The NE/SE tone and frequency decoder is a highly stable phaselocked loop with synchronous AM lock detection and power output circuitry. Its primary function is to drive a load whenever
More informationMultipath can be described in two domains: time and frequency
Multipath can be described in two domains: and frequency Time domain: Impulse response Impulse response Frequency domain: Frequency response f Sinusoidal signal as input Frequency response Sinusoidal signal
More informationSignals. Periodic vs. Aperiodic. Signals
Signals 1 Periodic vs. Aperiodic Signals periodic signal completes a pattern within some measurable time frame, called a period (), and then repeats that pattern over subsequent identical periods R s.
More informationFigure 1: The Penobscot Narrows Bridge in Maine, U.S.A. Figure 2: Arrangement of stay cables tested
Figure 1: The Penobscot Narrows Bridge in Maine, U.S.A. Figure 2: Arrangement of stay cables tested EXPERIMENTAL SETUP AND PROCEDURES Dynamic testing was performed in two phases. The first phase took place
More informationLow Cost, General Purpose High Speed JFET Amplifier AD825
a FEATURES High Speed 41 MHz, 3 db Bandwidth 125 V/ s Slew Rate 8 ns Settling Time Input Bias Current of 2 pa and Noise Current of 1 fa/ Hz Input Voltage Noise of 12 nv/ Hz Fully Specified Power Supplies:
More informationPractical issue: Group definition. TSTE17 System Design, CDIO. Quadrature Amplitude Modulation (QAM) Components of a digital communication system
1 2 TSTE17 System Design, CDIO Introduction telecommunication OFDM principle How to combat ISI How to reduce out of band signaling Practical issue: Group definition Project group sign up list will be put
More informationSpectrum Analysis  Elektronikpraktikum
Spectrum Analysis Introduction Why measure a spectra? In electrical engineering we are most often interested how a signal develops over time. For this timedomain measurement we use the Oscilloscope. Like
More informationLecture Outline. ESE 531: Digital Signal Processing. AntiAliasing Filter with ADC ADC. Oversampled ADC. Oversampled ADC
Lecture Outline ESE 531: Digital Signal Processing Lec 12: February 21st, 2017 Data Converters, Noise Shaping (con t)! Data Converters " Antialiasing " ADC " Quantization "! Noise Shaping 2 AntiAliasing
More informationSampling and Reconstruction of Analog Signals
Sampling and Reconstruction of Analog Signals Chapter Intended Learning Outcomes: (i) Ability to convert an analog signal to a discretetime sequence via sampling (ii) Ability to construct an analog signal
More information