SPECIFICATION Item no.: T60404N4646X100


 Charlene Johnson
 1 years ago
 Views:
Transcription
1 VACUUSCHELZE SPECFCATON tem no.: T60404N4646X100 For the electronic measurement of currents: solation between the primary circuit (high power) and the secondary circuit (electronic circuit) Customer: Standard type Customers Part no.: Page 1 of 2 Description Closed loop (compensation) Current Sensor with magnetic field probe Printed circuit board mounting Casing and materials ULlisted Electrical data Ratings Characteristics Excellent accuracy Very low offset current Very low temperature dependency and offset current drift Very low hysteresis of offset current Short response time Wide frequency bandwidth Compact design Reduced offset ripple Applications ainly used for stationary operation in industrial applications: AC variabel speed drives and servo motor drives Static converters for for DC motor drives Battery supplied applications Switched ode Power Supplies (SPS) Power Supplies for welding applications Uninterruptable Power Supllies (UPS) PN Primary nominal r.m.s. current 100 A R easuring resistance V C=± 12V Ω V C=± 15V Ω Secondary nominal r.m.s. current 100 ma K N Turns ratio 1: 1000 Accuracy Dynamic performance data min. typ. max. Unit P,max ax. measuring V C = ±12V, R = 10 Ω (t max = 10sec) ±230 V C = ±15V, R = 40 Ω (t max = 10sec) ±180 A X PN, T A= 25 C % ε L Linearity 0.1 % 0 Offset P=0, T A= 25 C ma t r Response time 1 µs Δt ( P,max) Delay time at di/dt = 100 A/µs 200 ns f Frequency bandwidth DC khz General data min. typ. max. Unit T A Ambient operating temperature C T S Ambient storage temperature C m ass 14 g V C Supply voltage ±11.4 ±12 or ±15 ±15.75 V C Current consumption 18 ma Constructed and manufactored and tested in accordance with EN (primary vs. secondary) Reinforced insulation, nsulation material group 1, Pollution degree 2 S clear Clearance (component without solder pad) 12 mm S creep Creepage (component without solder pad) 12 mm V sys System voltage overvoltage category 3 RS 600 V V work Working voltage (table 7 acc. to EN ) over voltage category 2 RS 1000 V U PD Rated discharge voltage peak value 1225 V aximal continuous and peak currents at defined temperatures Supply voltage ±12 V: Supply voltage ±15V: T A 85 C 85 C 70 C 55 C P 60 A 100 A 80 A 100 A P,max 235 A 149 A 241 A 246 A R 10 Ω 36 Ω 10 Ω 10 Ω T A 85 C 85 C 70 C 55 C P 50 A 75 A 70 A 100 A P,max 182 A 130 A 184 A 186 A R 40 Ω 70 Ω 40 Ω 40 Ω Date Name suue Amendment 81 KBP A: Sn. purposes without express written authorization by use illegally forbidden
2 VACUUSCHELZE SPECFCATON tem no.: T60404N4646X100 For the electronic measurement of currents: solation between the primary circuit (high power) and the secondary circuit (electronic circuit) Customer: Standard type Customers Part no.: Page 2 of 2 echanical outline (mm): General tolerances DN SO 2768c 14,3 6, Toleranz der Stiftabstände ±0,2mm (Tolerances grid distance) Connections: 1...3: 0,7 x 0,7 mm 22,86 36,5 5,08 DC = Date Code F = Factory *= vorläufig (preliminary) arking: 4646X100 F DC 27,6 F DC 8,5 12,35 4,5±0,5* 0,4 12,7 0,7x0,7 Schematic diagram i 1 1 V c 3 s  R Ca 2  V c Temperature of the primary conductor should not exceed 110 C Additional indications are obtainable on request. This specification is no declaration of warranty acc. BGB 443 dar. KBP A: Sn. purposes without express written authorization by use illegally forbidden
3 Additional nformation tem No.: T60404N4646X100 VACUUSCHELZE Customer: Customers Part No.: Page 1 of 3 Electrical Data (investigate by a type ing) min. typ. max. Unit V Ctot aximum supply voltage (without function) ±18 V ±15.75 to ±18 V: for 1s per hour R S Secondary coil T A=85 C 38.5 Ω X Ti Temperature drift of T A = C 0.1 % 0ges Offset current (including 0, 0t, 0T) 0.14 ma 0t Long term drift Offset current ma 0T Offset current temperature drift T A = C 0.05 ma 0H Hyteresis P=0 (caused by primary current 10 x PN) ma 0/ V C Supply voltage rejection ratio 0.01 ma/v i oss Offset ripple (with1 Hz filter first order) 0.2 ma i oss Offset ripple (with 100 khz filter first order) ma i oss Offset ripple (with 20 khz filter first order) ma C k aximum possible coupling capacity (primary secondary) 6 pf nspection (easurement after temperature balance of the samples at room temperature) K N(N 1/N 2) (V) 3011/6 Transformation ratio ( P=100A, Hz) 1 : 1000 ± 0,5 % 0 (V) 3226 Offset current < 0.1 ma V d (V) 3014: Test voltage, rms, 1 s 1.8 kv pin 1 3 vs. hole V e (AQL 1/S4) Partial discharge voltage acc.3024 (RS) 1300 V with V vor (RS) 1625 V Type Testing (Pin 13 to hole) V W HV transient test according to 3064 (1,2 µs / 50 µswave form) 8 kv V d Testing voltage to 3014 (5 s) 3,6 kv V e Partial discharge voltage acc.3024 (RS) 1300 V with V vor (RS) 1625 V Datum Name ndex Änderung 81 Weitergabe sowie Vervielfältigung dieser Unterlage, Verwertung und itteilung ihres nhalts Schadenersatz. Alle Rechte für den Fall der Patenterteilung oder GEintragung vorbehalten KBP A: Sn.
4 Additional nformation tem No.: T60404N4646X100 VACUUSCHELZE Customer: Customers Part No.: Page 2 of 3 Limit curve of measurable current Î P(R ambient temperature 85 C aximum measuring range (µsrange) Output current behaviour of a 3kA current V C = ±15V und R = 100Ω N P * Î P [A] V C = /15V  5% [A] V C = /12V  5% R [Ω] 200 p [A] 0 a / KN [A] t [µs] Fast increasing currents (higher than the specified p,max), e.g. in case of a short circuit, can be transmitted because the currents are transformed directly and be limited by diodes only. The offset ripple can be reduced by an external low pass. Simplest solution is a passive low pass filter of 1st order with f g 1 = 2π R C a n this case the response time is enlarged. t is calculated from: t t 2, 5R r r C a Applicable documents Current direction: A positive output current appears at point S, by primary current in direction of the arrow. Housing and bobbin material ULlisted: Flammability class 94V0. Weitergabe sowie Vervielfältigung dieser Unterlage, Verwertung und itteilung ihres nhalts Schadenersatz. Alle Rechte für den Fall der Patenterteilung oder GEintragung vorbehalten KBP A: Sn.
5 Additional nformation tem No.: T60404N4646X100 VACUUSCHELZE Customer: Customers Part No.: Page 3 of 3 Explanation of several of the terms used in the tablets (in alphabetical order) 0H: 0t: Zero variation after overloading with a DC of tenfold the rated value (R = R N) Long term drift of o after 100 temperature cycles in the range 40 bis 85 C. t r: Response time, measured as delay time at P = 0,8. Pmax between a rectangular current and the output current. t ( Pmax): Delay time between Pmax and the output current i a with a primary current rise of di 1/dt = 100 A/µs. U PD Rated discharge voltage (recurring peak voltage separated by the insulation) proved with a sinusoidal voltage V e U PD = 2 * V e / 1,5 V vor Defined voltage is the RS valve of a sinusoidal voltage with peak value of 1,875 * U PD required for partial discharge test in EC V vor = 1,875 *U PD / 2 V sys System voltage RS value of rated voltage according to EC V work Working voltage voltage according to EC which occurs by design in a circuit or across insulation X ges( PN): The sum of all possible errors over the temperature range by measuring a current PN: X ges = 100 S K ( ) N PN 1 X: Permissible measurement error in the final inspection at RT, defined by X = where is the output DC value of an input DC current of the same magnitude as the (positive) rated current ( o = 0 ) X Ti: ε L: Temperature drift of the rated value orientated output term. (cf. Notes on F i) in a specified temperature range, obtained by: X Ti = 100 ( T ) ( T ) A2 Linearity fault defined by ε L = A1 P 100 PN Where P is any input DC and Sx the corresponding output term. : see notes of F i ( o = 0). Sx This Additional information is no declaration of warranty according BGB $443. Weitergabe sowie Vervielfältigung dieser Unterlage, Verwertung und itteilung ihres nhalts Schadenersatz. Alle Rechte für den Fall der Patenterteilung oder GEintragung vorbehalten KBP A: Sn.
SPECIFICATION Item no.: T60404N4646X201
SPECFCATON tem no.: T60404N4646X201 For the electronic measurement of currents: isolation between the primary circuit (high power) and the secondary circuit (electronic circuit) Customer: Standard type
More informationSPECIFICATION Item no.: T60404N4646X200
SPECFCATON tem no.: T60404N4646X200 For the electronic measurement of currents: isolation between the primary circuit (high power) and the secondary circuit (electronic circuit) Customer: Standard type
More informationSPECIFICATION Item no.: T60404N4646X100
VACUUMSCHMELZE SPECFCATON tem no.: T60404N4646X100 KNo.: 25033 100 A Current Sensor Date: 06.11.2007 For the electronic measurement of currents: solation between the primary circuit (high power) and
More informationSPECIFICATION Item no.: T60404N4646X300
SPECFCATON tem no: T60404N4646X300 KNo: 24577 Date: 16042014 For the electronic measurement of currents: solation between the primary circuit (high power and the secondary circuit (electronic circuit
More informationSPECIFICATION Item no.: T60404N4646X410
V A C U U S C H E L Z E SPECFCATON tem no: T60404N4646410 KNo: 24616 Date: 18042013 For the electronic measurement of currents: solation between the primary circuit (high power) and the secondary circuit
More informationSPECIFICATION Item no.: T60404N4646X400
SPECFCATON tem no: T60404N4646X400 Date: 17042013 For the electronic measurement of currents: solation between the primary circuit (high power) and the secondary circuit (electronic circuit) Customer:
More informationSPECIFICATION Item no.: T60404N4646X410
VACUUMSCHMELZE SPECFCATON tem no.: T60404N4646X410 KNo.: 24616 50 A Current Sensor Date: 28.01.2008 For the electronic measurement of currents: solation between the primary circuit (high power) and
More informationSPECIFICATION Item no.: T60404N4646X300
VACUUMSCHMELZE SPECFCATON tem no.: T60404N4646X300 KNo.: 24577 1 5 8 12 25 A Current Sensor Date: 15.07.2010 For the electronic measurement of currents: solation between the primary circuit (high power)
More informationSPECIFICATION Item no.: T60404N4646X400
Customer: Standard type Customers Part no.: Page 1 of 5 Description Closed loop (compensation) Current Sensor with magnetic field probe Printed circuit board mounting Casing and materials ULlisted Electrical
More informationSPECIFICATION Item no.: T60404N4644X101
SPECFCATON tem no.: T60404N4644X0 Kno.: 50 A Current SensorModule Date: 24.02.204 solation between the primary circuit (electronic circuit) Customer: Standard type Customers Part no.: Page of 2 Description
More informationSPECIFICATION Item no.: T60404N4646X651
VACUUMSCHMELZE SPECIFICATION Item no: T60404N4646X651 Kno: 24507 25 A Current Sensor modul for 5Vsupply voltage Date: 10122010 For electronic current measurement: DC, AC, pulsed, mixed, with a galvanic
More informationSPECIFICATION Item no.: T60404N4646X762
V A C U U M S C H M E L Z E SPECIFICATION Item no: T60404N4646X762 Kno: 26020 15 A Current Sensor for 5V Supply Voltage Date: 03072012 For electronic current measurement: DC, AC, pulsed, mixed, with
More informationSPECIFICATION Item no.: T60404N4646X663
SPECIFICATION Item no: T60404N4646X663 Kno: 24513 6 A Current Sensor for 5V Supply Voltage Date: 182014 For electronic current measurement: DC, AC, pulsed, mixed, with a galvanic isolation between
More informationSPECIFICATION Item no.: T60404N4646X662
SPECIFICATION Item no: T60404N4646X662 Kno: 24512 15 A Current Sensor for 5V Supply Voltage Date: 182014 For electronic current measurement: DC, AC, pulsed, mixed, with a galvanic isolation between
More informationSPECIFICATION Item no.: T60404N4646X764
Customer: Standard type Customers Part no: Page 1 of 4 Description Closed loop (compensation) Current Sensor with magnetic field probe Printed circuit board mounting Casing and materials ULlisted Electrical
More informationSPECIFICATION Item No.: T60404P4640X102
SPECFCATON tem No.: T60404P4640X102 Customer: Standard Type Customer part no..: Page 1 of 6 Electrical data PN Primary nominal r.m.s. current 1000 A R 1 M Measuring resistance 0... 100 Ω SN Secondary
More informationSPECIFICATION Item no.: T60404N4646X975
Customer: Standard type Customers Part no.: Page 1 of 2 8Description Characteristics Applications Closed loop (compensation) Excellent accuracy Mainly used for stationary operation in industrial Current
More informationSPECIFICATION Item No.: T60404P4640X100
Customer: Standard Type Customer part no..: Page 1 of 6 Electrical data PN Primary nominal r.m.s. current 1000 A R 1 M Measuring resistance 0... 100 Ω SN Secondary nominal r.m.s. current 200 ma K N Turns
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current transducer LF 510S I PN = 500 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Bipolar and insulated
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary circuit and the secondary circuit.
Current Transducer CTSR 0.3TP/SP4 I PRN = 300 ma For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary circuit and the secondary circuit. Features
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current transducer LF 2010S I PN = 2000 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Bipolar and insulated
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary circuit and the secondary circuit.
Current Transducer CTSR 0.6TP/SP2 I PRN = 600 ma For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary circuit and the secondary circuit. Features
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and secondary circuit.
Current Transducer LF 510S I P N = 500 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and secondary circuit. Features Bipolar and insulated
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic isolation between the primary and the secondary circuit.
Current Transducer CASR series I PN = 6, 5, 25, 5 A Ref: CASR 6NP, CASR 5NP, CASR 25NP, CASR 5NP For the electronic measurement of current: DC, AC, pulsed..., with galvanic isolation between the primary
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current transducer LF 210S/SP3 I PN = 100 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Bipolar and insulated
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current Transducer CAS 25NP/SP2 N = 25 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Closed loop (compensated)
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current Transducer LF 2010S/SPA0 I P N = 2000 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Bipolar and
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current transducer LF 510S/SP13 I PN = 500 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Bipolar and
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current Transducer LF 2010S/SPA7 I P N = 2000 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Bipolar and
More informationCurrent Transducer CTSR 1P = 1A
Current Transducer CTSR 1P I PRN = 1A For the electronic measurement of current: DC, AC, pulsed..., with galvanic isolation between the primary (high power) and the secondary circuit (electronic circuit).
More informationFor ultrahigh precision measurement of current: DC, AC, pulsed..., with galvanic separation between primary and secondary. Applications.
Current Transducer IT 700S ULTRASTAB I PM = 700 A For ultrahigh precision measurement of current: DC, AC, pulsed..., with galvanic separation between primary and secondary. Features Closed loop (compensated)
More informationHigh Performance Current Transducer ITL 900T = A
High Performance Current Transducer ITL 900T For the electronic measurement of currents: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. I PM = 0... 900 A Electrical
More informationHigh Performance Current Transducer IT 200S ULTRASTAB = A. ε L
High Performance Current Transducer IT 200S ULTRASTAB For the electronic measurement of currents: DC, AC, pulsed..., with galvanic isolation between the primary circuit and the secondary circuit. I PM
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current Transducer IN 1000S I P N = 1000 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Closed loop (compensated)
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current Transducer IN 1000S N = 1000 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Closed loop (compensated)
More informationFor ultrahigh precision measurement of current: DC, AC, pulsed..., with galvanic separation between primary and secondary. Applications.
Current Transducer IT 605S ULTRASTAB I PN = 600 A For ultrahigh precision measurement of current: DC, AC, pulsed..., with galvanic separation between primary and secondary. Features Wide operating temperature
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current transducer CKSR series N = 6, 5, 25, 5 A Ref: CKSR 6NP, CKSR 5NP, CKSR 25NP, CKSR 5NP For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary
More informationRef: HLSR 10P/SP3, HLSR 20P/SP3, HLSR 40P/SP3, HLSR 50P/SP3
Current Transducer HLSRP/SP3 series N = 10... 50 A Ref: HLSR 10P/SP3, HLSR 20P/SP3, HLSR 40P/SP3, HLSR 50P/SP3 For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation
More informationFor ultrahigh precision measurement of current: DC, AC, pulsed..., with galvanic separation between primary and secondary. Applications.
Current Transducer ITN 1000S ULTRASTAB I PM = 1000 A For ultrahigh precision measurement of current: DC, AC, pulsed..., with galvanic separation between primary and secondary. Features Closed loop (compensated)
More informationFor ultrahigh precision measurement of current: DC, AC, pulsed..., with galvanic separation between primary and secondary. Applications.
Current Transducer ITN 900S ULTRASTAB I PM = 900 A For ultrahigh precision measurement of current: DC, AC, pulsed..., with galvanic separation between primary and secondary. Features Closed loop (compensated)
More informationFor ultrahigh precision measurement of current: DC, AC, pulsed..., with galvanic separation between primary and secondary. Applications.
Current Transducer IT 700SB ULTRASTAB I PM = 700 A For ultrahigh precision measurement of current: DC, AC, pulsed..., with galvanic separation between primary and secondary. Features ± 10 V voltage output
More informationFor ultrahigh precision measurement of current: DC, AC, pulsed..., with galvanic separation between primary and secondary. Applications.
Current Transducer IT 205S ULTRASTAB I PN = 200 A For ultrahigh precision measurement of current: DC, AC, pulsed..., with galvanic separation between primary and secondary. Features Wide operating temperature
More informationFor the electronic measurement of voltage: DC, AC, pulsed..., with galvanic isolation between the primary and the secondary circuit.
Voltage transducer V PN = 4 V Ref: DV 4/SP4 For the electronic measurement of voltage: DC, AC, pulsed..., with galvanic isolation between the primary and the secondary circuit. Features Bipolar and isolated
More informationDS600ID. Specification highlights Symbol Unit Min Typ Max. Features. Applications: Linearity error maximum 1 ppm. MPS for particles accelerators
Ultrastable, high precision (ppm class) fluxgate technology DS Series current transducer for nonintrusive, isolated DC and AC current measurement up to 1000A Features Linearity error maximum 1 ppm Fluxgate,
More informationDS200ID. Specification highlights Symbol Unit Min Typ Max. Features. Applications: Linearity error maximum 2 ppm. MPS for particles accelerators
Ultrastable, high precision (ppm class) fluxgate technology DS Series current transducer for nonintrusive, isolated DC and AC current measurement up to 370A Features Linearity error maximum 2 ppm Fluxgate,
More informationDS400ID. Specification highlights Symbol Unit Min Typ Max. Features. Applications: Linearity error maximum 1.5 ppm. MPS for particles accelerators
Ultrastable, high precision (ppm class) fluxgate technology DS Series current transducer for nonintrusive, isolated DC and AC current measurement up to 600A Features Linearity error maximum 1.5 ppm Fluxgate,
More informationDQ600ID. Specification highlights Symbol Unit Min Typ Max. Features. Applications: Linearity error maximum 1 ppm. MPS for particles accelerators
Ultrastable, high precision (ppm class) fluxgate technology DS Series current transducer for nonintrusive, isolated DC and AC current measurement up to 1000A Features Linearity error maximum 1 ppm Fluxgate,
More informationRef: HLSR 10SM, HLSR 16SM, HLSR 20SM, HLSR 32SM, HLSR 40SM, HLSR 50SM
Current Transducer HLSRSM series N = 1... 5 A Ref: HLSR 1SM, HLSR 16SM, HLSR 2SM, HLSR 32SM, HLSR 4SM, HLSR 5SM For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation
More informationRef: HLSR 10SM/SP33, HLSR 20SM/SP33, HLSR 32SM/SP33, HLSR 40P/SP33, HLSR 50SM/SP33
Current Transducer HLSRSM/SP33 series I PN = 10... 50 A Ref: HLSR 10SM/SP33, HLSR 20SM/SP33, HLSR 32SM/SP33, HLSR 40P/SP33, HLSR 50SM/SP33 For the electronic measurement of current: DC, AC, pulsed...,
More informationRef: HO 50S/SP33, HO 100S/SP33, HO 150S/SP33, HO 200S/SP33, HO 250S/SP33
Current Transducer HOS/SP33 series I PN = 50, 100, 150, 200, 250 A Ref: HO 50S/SP33, HO 100S/SP33, HO 150S/SP33, HO 200S/SP33, HO 250S/SP33 For the electronic measurement of current: DC, AC, pulsed...,
More informationCustomer: Standard type Page 1 of 7 Description Fluxgate current sensor with toroidal core PCB mounting
Customer: Standard type Page 1 of 7 Description Fluxgate current sensor with toroidal core PCB mounting Electrical data Ratings min. typ. max. Unit I P Primary nominal RMS current (1phase / 3phase) 80
More informationDL2000ID. Specification highlights Symbol Unit Min Typ Max. Features. Applications: 1 ppm linearity. MPS for particles accelerators.
DL000ID Ultrastable, high precision (ppm class) fluxgate technology DS Series current transducer for nonintrusive, isolated DC and AC current measurement up to 3000A Features ppm linearity 6 ppm offset
More informationDM1200ID. Specification highlights Symbol Unit Min Typ Max. Applications: Features. Power measurement and power analysis
Ultrastable, high precision (ppm class) fluxgate technology DM Series current transducer for nonintrusive, isolated DC and AC current measurement up to 1500A Features Ø45mm aperture enabling large isolated
More informationCustomer: Standard type Page 1 of 7 Description Fluxgate current sensor with toroidal core PCB mounting
Customer: Standard type Page 1 of 7 Description Fluxgate current sensor with toroidal core PCB mounting Electrical data Ratings min. typ. max. Unit I P Primary nominal RMS current (1phase / 3phase) 80
More informationDS200IDCD100. Specification highlights Symbol Unit Min Typ Max. Features. Applications: Linearity error maximum 2 ppm. MPS for particles accelerators
Ultrastable, high precision (ppm class) fluxgate technology DS Series current transducer for nonintrusive, isolated DC and AC current measurement up to 370A Features Linearity error maximum 2 ppm 100
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current Transducer HONSM/SP33 series N = 8, 15, 25 A Ref: HO 8NSM/SP33, HO 15NSM/SP33, HO 25NSM/SP33 For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the
More informationFor ultrahigh precision measurement of current: DC, AC, pulsed..., with galvanic separation between primary and secondary. Applications.
Current Transducer IT 200S ULTRASTAB I PM = 200 A For ultrahigh precision measurement of current: DC, AC, pulsed..., with galvanic separation between primary and secondary. Features Closed loop (compensated)
More informationDS2000ICLA. Specification highlights Symbol Unit Min Typ Max. Features. Applications: 1 ppm linearity. MPS for particles accelerators.
Ultrastable, high precision (ppm class) fluxgate technology DS Series current transducer for nonintrusive, isolated DC and AC current measurement up to 3000A Features ppm linearity 0 ppm offset Current
More informationDS2000UBLA10V. Features. Applications: 20 ppm linearity. MPS for particles accelerators. 15 ppm offset. Gradient amplifiers for MRI devices
Ultrastable, high precision (ppm class) fluxgate technology DS Series current transducer for nonintrusive, isolated DC and AC current measurement up to 2000A Features 20 ppm linearity 15 ppm offset 10V
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current Transducer HONSM SERIES I PN = 8, 15, 25 A Ref: HO 8NSM, HO 15NSM, HO 25NSM For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and
More informationI PN. Ref: HO 50S, HO 100S, HO 150S, HO 200S, HO 240S, HO 250S
Current Transducer HOS series I PN = 50, 100, 150, 200, 240, 250 A Ref: HO 50S, HO 100S, HO 150S, HO 200S, HO 240S, HO 250S For the electronic measurement of current: DC, AC, pulsed..., with galvanic
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current Transducer LF 21S/SPA2 I P N = 2 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Bipolar and insulated
More informationHO 50S/SP30, HO 100S/SP30, HO 150S/SP30, HO 200S/SP30, HO 250S/SP30 and HO 200S/SP31
Current Transducer HOS/SP3 series Ref: I PN = 5,, 15, 2, 25 A HO 5S/SP3, HO S/SP3, HO 15S/SP3, HO 2S/SP3, HO 25S/SP3 For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current Transducer ITL 4000S I PN = 4000 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. I > 0 P Features Closed
More informationCustomer: Standard type Page 1 of 7 Description Fluxgate current sensor with toroidal core PCB mounting
Customer: Standard type Page 1 of 7 Description Fluxgate current sensor with toroidal core PCB mounting Characteristics Excellent accuracy AECQ qualified components Switching opencollector outputs Compact
More information= 1000 V. Voltage transducer DVC 1000P V P N
Voltage transducer DVC 1P V P N = 1 V For the electronic measurement of voltage: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Bipolar and insulated
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current Transducer LF 1010S/SPA5 I P N = 1000 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Bipolar and
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current Transducer HONP series I PN = 4, 6, 12, 15 Ref: HO 4NP, HO 6NP, HO 12NP, HO 15NP For the electronic measurement of current: DC, C, pulsed..., with galvanic separation between the primary and
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current Transducer N = 6, 15, 25, 5 A Ref: LESR 6NP, LESR 15NP, LESR 25NP, LESR 5NP For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the
More informationI P. /dt. di p V S Applications. Standards 1) IEC : 2007; IEC : ) IEC : 2016; IEC : 2017
Ref: ARTB22D70, ARTB22D125, ARTB22D175, ARTB22D300 Flexible cliparound Rogowski coil for the electronic measurement of AC current with galvanic separation between the primary circuit (power) and
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current Transducer LESR series N = 6, 15, 25, 5 A Ref: LESR 6NP, LESR 15NP, LESR 25NP, LESR 5NP For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current transducer LF 1010S/SPA2 I PN = 1000 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Bipolar and
More informationDC CURRENT SENSOR DC VOLTAGE SENSOR
VC T JH H 0 5 tn o L /S P 0 0 5 5m 0 C 0 DC P : 5 V J 0 o : ±5 5 H T DC CURRENT SENSOR DC VOLTGE SENSOR J&D C/DC CTs provide noncontact detection of C/DC current. The C/DC CTs have two types of splitcore
More informationDifferential Current Sensor acc. to the standard UL2231 Ed.2.0. Customer: Standard type Page 1 of 7. Electrical data Ratings min. typ. max.
Customer: Standard type Page 1 of 7 Electrical data Ratings min. typ. max. Unit I P Primary nominal RMS current (1phase / 3phase) 32 40 A I N1 Rated residual operating current 1 5 ma rms I N2 Rated residual
More informationDS10000ULFS10V/10kA
Ultrastable, high precision (ppm class) fluxgate technology DS Series current transducer for nonintrusive, isolated DC and AC current measurement up to 11kA Features Linearity error maximum +/ 7ppm
More informationI P. /dt. di p V S+ Applications. Standards. 1) IEC ed1.0: 2007; IEC : ed1.0: 2012
Ref: ARTB22D70, ARTB22D125, ARTB22D175 Flexible cliparound Rogowski coil for the electronic measurement of AC current with galvanic separation between the primary circuit (power) and the secondary
More informationCMS2005 MagnetoResistive Current Sensor (I PN
The CMS2000 current sensor family is designed for highly dynamic electronic measurement of DC, AC, pulsed and mixed currents with integrated galvanic isolation. The MagnetoResistive technology enables
More informationCMS2050 MagnetoResistive Current Sensor (I PN
The CMS2000 current sensor family is designed for highly dynamic electronic measurement of DC, AC, pulsed and mixed currents with integrated galvanic isolation. The MagnetoResistivetechnology enables an
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current Transducer ITL 4000S I P N = 4000 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. I > 0 P Features Closed
More informationAUTOMOTIVE CURRENT TRANSDUCER HC2F100SN CLIPS
AUTOMOTIVE CURRENT TRANSDUCER HCFSN CLIPS Introduction The HCF CLIPS Family is for the electronic measurement of DC, AC or pulsed currents in high power and low voltage automotive applications with galvanic
More informationFor the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Current transducer ITC 2000S/SP2 N = 2000 A For the electronic measurement of current: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Bipolar and insulated
More informationFor the electronic measurement of voltage: DC, AC, pulsed..., with galvanic isolation between the primary and the secondary circuit.
Voltage transducer V PN = 1 V Ref: DV 1/SP For the electronic measurement of voltage: DC, AC, pulsed..., with galvanic isolation between the primary and the secondary circuit. Features Bipolar and isolated
More informationFor the electronic measurement of voltage: DC, AC, pulsed..., with galvanic isolation between the primary and the secondary circuit.
Voltage transducer DVL 15 V PN = 15 V For the electronic measurement of voltage: DC, AC, pulsed..., with galvanic isolation between the primary and the secondary circuit. Features Bipolar and isolated
More informationFor the electronic measurement of voltage: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit.
Voltage transducer DV V PN = V For the electronic measurement of voltage: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Applications Bipolar and insulated
More informationAUTOMOTIVE CURRENT SENSOR HC6H300S
AUTOMOTIVE CURRENT SENSOR HC6H300S Page 1/ 5 Introduction Principle of HC6H Family The HC6H Family is for use on the electronic measurement of DC, AC or pulsed currents in high power and low voltage automotive
More informationAUTOMOTIVE CURRENT SENSOR HC6H500S. Datasheet
AUTOMOTIVE CURRENT SENSOR HC6H500S Datasheet Page 1/ 5 Introduction Principle of HC6H Family The HC6H Family is for use on the electronic measurement of DC, AC or pulsed currents in high power and low
More informationMSA Hall effect transducer
 Hall effect transducer Datasheet Features Specially designed for railway applications losed loop (compensated) High dielectric strength Precise linearity Precise accuracy High dynamic response No foucault
More informationAUTOMOTIVE CURRENT TRANSDUCER HC6F600S
AUTOMOTIVE CURRENT TRANSDUCER HC6F600S 18198102215 Page 1/5 Introduction Principle of HC6F Family The HC6F Family is for use on the electronic measurement of DC, AC or pulsed currents in high power and
More informationActive. Advanced Materials The Key to Progress
Active Current SEnsors FOR PCB MOUNTING Advanced Materials The Key to Progress The Company VACUUMSCHMELZE VACUUMSCHMELZE GmbH & Co. KG (VAC) is one of the world s leading producers of special metallic
More informationLow profile safety relay with forcibly guided double contacts FEATURES. Relay height, 14.5mm. 4a2b
Low profile safety relay with forcibly guided double contacts SFN4D RELAY 3.3 33.0 FEATURES Relay complies with EN 020, Type B. Polarized magnet system with snap action function Tolerance ±0.3 mm Weight
More information= 600 V. Voltage transducer DVM 600 V PN
Voltage transducer DVM 6 V PN = 6 V For the electronic measurement of voltage: DC, AC, pulsed..., with galvanic separation between the primary and the secondary circuit. Features Bipolar and insulated
More informationVoltage transducer DVL 50
Voltage transducer DVL 5 V PN = 5 V For the electronic measurement of voltage: DC, AC, pulsed..., with galvanic isolation between the primary and the secondary circuit. Features Bipolar and isolated measurement
More informationMSA500  Hall effect transducer
 Hall effect transducer Datasheet Features Specially designed for railway applications losed loop (compensated) High dielectric strength Precise linearity Precise accuracy High dynamic response No foucault
More information