1 Established 1981 Advanced Test Equipment Rentals ATEC (2832) 1 INA 703 Magnetic field coil User manual A
2 INA 703 Magnetic field coil User manual INA 703 Magnetic field coil
3 contents 1 Explanation of symbols 5 2 Introduction General description Supply frequency magnetic fields Pulsed magnetic fields Oscillatory magnetic fields 9 3 User instructions General Unpacking and installation Positioning Connection to the generator Connecting to Teseq ProfLine system for IEC testing Connecting to Teseq MFO 6501 or MFO 6502 for IEC testing Connecting to any other generator for IEC testing Connecting to Teseq NSG 3040 or NSG 3060 for IEC testing Connecting to Teseq NSG 3040 or NSG 3060 for IEC testing Test execution 17 4 Maintenance and function check General Cleaning Function check Calibration Warranty 20 5 Declaration of conformity (CE) 21 6 Options and accessories INA 703 Stand INA 703 Coil INA 3250 interconnection cable INA 3251 Interconnection cable Pulse wave shape adapter INA MD 300 surge pulse current probe set 24 7 System description 25 8 Addresses 26
4 Lethal danger from high voltages and the risk of radiating illegal electromagnetic interference. This system must be used only for EMC test purposes as specified in these operating instructions. The INA 703 must be installed and used only by authorized and trained EMC specialists. Personnel fitted with a heart pacemaker may not operate the instrument and must not be in the vicinity of the test setup while it is in operation. When the system is used in conjunction with options, accessories or other equipment the safety instructions concerning those devices must also be observed.
5 1 ExplANAtion of symbols 5 Please take note of the following explanations of the symbols used in order to achieve the optimum benefit from this manual and to ensure safety during operation of the equipment. The following symbol draws your attention to a circumstance where nonobservation of the warning could lead to inconvenience or impairment in the performance. Example: This connection must not be confused with the Equipment under Test (EUT) power input. The following symbol draws your attention to a circumstance where nonobservation of the warning could lead to component damage or danger to the operating personnel. Example: Never connect or disconnect the EUT while the test system is performing a test.
6 6 2 Introduction 2.1 General description INA 703 is the top end of a family of magnetic field coils, designed for testing per IEC (supply frequency magnetic fields), IEC (pulsed magnetic fields), IEC (oscillatory magnetic fields). Thanks to a multi wire - 37 turns concept, it allows the generation of fields higher than 1000 A/m while using a current source rated for 30 A, so insures the possibility to meet the IEC standards requirement of a current THD < 8% which can be met only when using a synthetic current source. INA 703 has a connection possibility after wire turn 1 and turn 5, which allows to reach great accuracy when generating low amplitude fields. INA 703 Magnetic field coil
7 The tap off after 1 turn is also used for tests per IEC and IEC , which both require a single turn coil. 7 For tests per IEC , INA 703 can be used as an accessory to the Teseq profline system, together with a NSG kva source, the INA 2141 impedance box and the WIN 2120 software can generate supply frequency fields (50 and 60 Hz) up to 330 A/m continuously and 1100 A/m short term (3 s). It can also be used with the MFO 6501 or 6502 current sources and the NSG 3000 series of generators and generate Supply frequency fields (50 and 60 Hz) up to 120 A/m continuously and 120 A/m short term (3 s). For tests per IEC , INA 703 can be used together with a classic combined wave generator as available in NSG 3040 or NSG 3060 series plus a pulse wave shape adaptor box INA 752. For tests per IEC , INA 703 can be connected to an adequate slow oscillatory waves generator. Thanks to the multi wire concept and to the professional mechanical design, like U shape base on rolls which allows convenient approach and positioning to the test table, INA 703 is the ideal accessory for magnetic field testing. 2.2 Supply frequency magnetic fields Supply frequency magnetic field tests, simulate the magnetic fields typically generated by the current flow in power supply cables as specified in IEC/EN Such magnetic fields can affect the operation of items of equipment that are sensitive to them. For testing to IEC , the INA 703 needs to be supplied by a regulated 50 or 60 Hz current (THD<8%) of maximum 10 A continuously and 35 A short term (<5 s). Single turn, 5 turns, and 37 turns can be selected in order to reach high fields with low currents, but maintaining good resolution for lower fields testing.
8 8 The Teseq ProfLine system comprising an NSG kva source, an INA 2141 impedance box and WIN 2120 software is ideally suited for this application. For this setup the interconnection cable INA 3250 is required. The system can generate supply frequency fields (50 and 60 Hz) up to 330 A/m continuously and 1100 A/m short term (3 s). For more information please consult ProflLne 2100 series user documentation. The INA 703 can also be used with the Teseq MFO 6501 or 6502 current sources and the NSG 3000 series of generators to generate supply frequency fields (50 and 60 Hz) up to 120 A/m continuously and 120 A/m short term (3 s). For this setup the interconnection cable INA 3251 is required. For more information please consult MFO 6501/2 or NSG 3040 series user manuals. 2.3 Pulsed magnetic field Pulsed magnetic fields tests, simulate the type of interference produced by surge pulses as a result of lightning strikes to buildings or other metallic structures such as free standing masts, ground conductors, grounding networks, etc. as specified in IEC/EN Magnetic fields of this type can upset the operation of installations that find themselves within such fields. For testing to IEC , the INA 703 can be used together with a classic combination wave generator such as the NSG 3040 or NSG 3060 series. For this test a single loop ( 1 turn) will be used. In order to meet the pulse waveform required by IEC , the waveshape adapter INA 752 needs to be used. INA 703 Magnetic field coil
9 2.4 Oscillatory magnetic fields Oscillatory magnetic fields tests, simulate the type of interference produced by the switching of medium and high voltage switchgears, as specified in IEC/EN Magnetic fields of this type can upset the operation of installations that find themselves within such fields. For testing to IEC , the INA 703 can be connected to an appropriate slow oscillatory wave generator. For this test a single loop (1 turn) will be used.
10 10 3 User instructions WARNING - Improper or careless operation can be fatal! These operating instructions form an essential part of the equipment and must be available to the operator at all times. The user must obey all safety instructions and warnings. Neither Teseq AG, Luterbach, Switzerland, nor any of its subsidiary sales organisations can accept any liability for personal, material or consequential injury, loss or damage that may result from improper use of equipment and accessories. 3.1 General The INA 703 together with the adequate generator must be operated only by authorized and trained specialists. The generator with the coil is to be used only for the purpose specified by the manufacturer. The user is directly responsible for ensuring that the test setup does not cause excessive radiated interference which could affect other instrumentation or human beings. The test system itself does produce EM radiation up to 1100 A/m. WARNING - Personnel fitted with a heart pacemaker must neither operate the instrument nor approach the test setup while a test is being executed. Only approved accessories, connectors, adapters, etc. are to be used to ensure safe operation. 3.2 Unpacking and installation Check the packaging for signs of damage in transit. Any damage should be reported immediately to the transportation company. INA 703 Magnetic field coil
11 INA 703 is delivered dismantled, safely packed in a solid wooden box. 11 Identify the upper side of the box and remove cover after having removed the screws. The contents of the package will look like: Unpack carefully the various parts. Check the parts for signs of transport damage. Any damage should be reported to the transportation company immediately. Screw the vertical bars to the rolling base.
12 12 Lift (2 persons) the coil (red part) over the base, slide the black fixtures into the vertical bars. Secure the coil by fastening the fixtures to the bar. The INA 703 is now ready to be rolled around to the test setup. NOTE! Do not dispose of packaging materials. All packaging should be retained in the event that the instrument or any of its accessories should need to be returned to a Teseq service center for repair or calibration. INA 703 Magnetic field coil
13 3.3 Positioning The U-shaped base with castors will allow easy positioning of the INA 703 to the EUT, even if this one is placed on a test table. 13 The equipment to be tested shall always be positioned in the geometric center of the red coil part Tests need to be done with the coil in horizontal position and then in vertical position, 2 bubble level indicators as shown in picture below will facilitate the right positioning. Bubble level indicators Securing in horizontal and vertical position goes easy thank to the positioning lock: Position lock NOTE! Standards require only testing for horizontal or vertical position. INA 703 positioning lock has an intermediate position at 45 which may be used for investigation purposes.
14 Connection to the generator The INA 703 features a multipole terminal system which insures comfortable and safe connection of various generators suited for various applications Safety banana plugs: Red and black These 2 plugs are mainly used for pulsed and oscillatory magnetic fields testing. The Black banana is the low entry of the coil, the red banana is connected to the end of Turn Nr 1. connecting to these terminals is equivalent to test with a single turn coil. Round 4 pole connector: The pole marked with PE sign is connected together with the black banana to the low entry of the coil. Pole Nr. 1 is connected together with the red banana to the end of turn Nr. 1 Pole Nr. 2 is connected to the end of turn Nr. 5 Pole Nr. 3 is connected to the end of turn Nr. 37 INA 703 Magnetic field coil
15 3.4.1 Connecting to Teseq ProfLine system for IEC testing An INA 3250 interconnection cable is required. Connect the INA 3250 interconnection cable directly from the 4 pole connector of INA 703 to INA 2141 of ProfLine system. 15 The setting of the various levels and frequencies of supply frequency magnetic fields is supported comfortably by the WIN 2120 software. INA 2141 features 3 ranges (red selector on INA 2141 front: Position 0= Off, position 1 = 1 turn, position 2= 5 turns, position 3 = 37 turns) for better accuracy of low level magnetic fields. For each of the 3 ranges a separate coil factor (the coil factor is available in the INA 703 calibration certificate and is also labelled on INA 703 terminal unit) needs to be used. In the WIN 2120 software configuration file the INA 703 needs to be configured as 3 separate coils. For more information please consult the ProfLine systems user documentation Connecting to Teseq MFO 6501 or MFO 6502 for IEC testing An INA 3251 interconnection cable is required. Connect the INA 3251 interconnection cable directly from the 4 pole connector of INA 703 to the output of MFO 6501.
16 16 The 3 ranges of the INA 703 are selectable by a respective banana plug (Green/ yellow = Low, blue = 1 Turn, Black = 5 Turns, red = 37 Turns) for better accuracy of low level magnetic fields. For each of the 3 ranges a separate coil factor. (The coil factor is available in the INA 703 calibration certificate and is also labelled on INA 703 terminal unit) needs to be used. For more information please consult the NSG 3040 systems or MFO 6501/2 user documentation Connecting to any other generator for IEC testing An INA 3251 interconnection cable is required. Connect the INA 3251 interconnection cable directly from the 4 pole connector of INA 703 to the output of the generator. The 3 ranges of the INA 703 are selectable by a respective banana plug (Green/ yellow = Low, blue = 1 Turn, Black = 5 Turns, red = 37 Turns) for better accuracy of low level magnetic fields. For each of the 3 ranges a separate coil factor. (The coil factor is available in the INA 703 calibration certificate and is also labelled on INA 703 terminal unit) needs to be used. Make sure that the maximum ratings of INA 703 are not exceeded risk of fire. For Standard compliance testing the generator shall have a THD<8%, this is not permanently guaranteed when solutions based on variacs and step down transformers powered from mains supply are used. INA 703 Magnetic field coil
17 3.4.4 Connecting to Teseq NSG 3040 or NSG 3060 for IEC testing The INA 752 pulse wave shape adapter is required. 17 INA 752 has at one end the right plugs fitting into the surge out terminals of NSG 3040 and 3060 series, and at the other end the right safety banana plugs to fit in the INA 703 terminals. For more information please consult the NSG 3040 or NSG 3060 systems user documentation, section about magnetic fields testing Connecting to Teseq NSG 3040 or NSG 3060 for IEC testing The required connection cable is delivered together with the NSG MOW generator series. For more information please consult the NSG 3040 or NSG 3060 systems user documentation, section about magnetic fields testing. 3.5 Test execution The user must observe safety instruction for all the instruments and associated equipment involved in the test setup. Test setup configuration is to be strictly in compliance with the methods described in the relevant standard to ensure that the test is executed in a compliant manner
18 18 WARNING - Users must be aware of the following dangers that can occur during testing: It is recommended for the user to stay away (at least a few meters) from the loop antenna while magnetic fields are generated. Also keep away magnetic field sensitive devices and items such as credit cards magnetic key cards etc which might be influencedby the fields. The field generated in the loop antenna is directly proportional to the current flowing through it: Field strength (A/m) H = Cf x I Where H is the generated field, Cf the coil factor, I the current flowing through the loop. The use and setting of the generator and associated software is given in the generator user documentation. INA 703 Magnetic field coil
19 4 MAINtenance and function check General Maintenance by the user is restricted to visual inspection, cleaning the outer housing, performing a function check and verification of the pulse parameters. 4.2 Cleaning In general a moist cloth is sufficient for cleaning the outer housingl. If necessary add a small amount of a mild, non-foaming household cleanser. No chemicals (acid, etc) should be used for cleaning purposes. Before beginning to clean the test system ensure that it is switched off and the mains power cable is unplugged from the supply. 4.3 Function check The safety measures described previously must be strictly observed while carrying out a function check. Functional check of the whole system (generator plus INA 703) can be made by monitoring the current flowing through the INA 703. This current, multiplied by the coil factor, will give the generated magnetic field. Supply frequency current may be measured by a RMS ampmeter. Pulsed and oscillatory currents can be observed with an oscilloscope.
20 20 Do not connect the oscilloscope directly in order not to exceed its max. input voltage. Teseq recommends the use of an oscilloscope together with MD 300 current probe (see paragraph: accessories). 4.4 Calibration The calibration of a Magnetic field loop is explained in IEC The same calibration is valid for testing per IEC and -10. A magnetic field meter is placed in the center of the loop and a reference supply frequency current is generated through the loop. The ratio between the measured magnetic field and the reference current gives the coil factor. Teseq has a worldwide network of accredited test laboratories specialised in EMC calibrations. Visit 4.5 Warranty Teseq grants a warranty of 2 years on this test system, effective from the date of purchase. During this period, any defective components part will be repaired or replaced free of charge or, if necessary, the test system will be replaced by another of equivalent value. The decision regarding the method of reinstating the functional capability is at the sole discretion of Teseq. Excluded from the warranty is damage or consequential damage caused through negligent operation or use as well as the replacement of parts subject to degradation. The warranty is rendered invalid by any intervention on the part of the customer or a third party. The faulty items have to be returned in their original packaging. Teseq accept no responsibility for damage in transit. INA 703 Magnetic field coil
21 5 declaration of conformity (ce) 21
22 22 6 options and accessories 6.1 INA 703 stand 6.2 INA 703 coil INA 703 Magnetic field coil
23 6.3 INA 3250 Interconnection cable To interconnect INA 703 with Teseq ProfLine system for IEC testing INA 3251 Interconnection cable To interconnect INA 703 with Teseq MFO 6501 or MFO 6502 for IEC testing. 6.5 Pulse wave shape adapter INA 752 The pulse waveshape adapter INA 752 is a standard accessory for the Teseq NSG 3000 series. It provides a convenient means for interconnecting the NSG 3040 or NSG 3060 surge generator with the loop antennas INA 701, 702 or 703 and insures that the generated pulsed magnetic field has the waveshape as specified in the application standard. The combination NSG 3040 with CWM 3450 INA 752 INA 701 (or 702 or 703) is required for magnetic field testing for pulsed fields up to 1200 A/m. It complies to the requirements of IEC
24 MD 300 surge pulse current probe The MD 300 probe has been specially designed to verify surge current pulses as specified in IEC/EN , ANSI C62.41 and their derivates. The main advantage of the MD 300 current probe is, that the measuring system is physically isolated from the circuit under test. The MD 300 current probe is ready to use as coming along with pre-mounted coaxial cable. The BNC-end plug needs to be connected to the high-impedance input or 50 Ω input of an ordinary memory oscilloscope. The conductor carrying the surge current to be measured is passed through the hole in the current probe. The resulting voltage wave shape on the oscilloscope will then be an authentically reproduction of the actual current wave shape within the given accuracy. INA 703 Magnetic field coil
25 7 system description 25 Technical specifications Coil size: 1 x 1 m Homegen field volume: 60 x 60 x 50 cm Base: Strong aluminum with rolls Max vertical positioning Height: 130 cm Turns: 1, 5, 37 Max supply frequency current cont : 10 A Max supply frequency current short (3 s): 35 A Max supply frequency field strength cont: 330 A/m Max supply frequency field strength short (3 s): 1100 A/m Max pulsed current: 1500 A Max pulsed field strength: 1200 A/m Weight: 25 kg Accessories Connection cable for INA 2141: INA 3250 Connection cable for MFO 6501 or 6502: INA 3251 Pulse waveshape adapter for IEC testing: INA 752 Current probe: MD 300 Manual current source (4 A): MFO 6501 Automatic current source (4 A): MFO 6502 Automatic current source (37 A): NSG kva (ProfLine 2105) Other coil models INA 701 single turn coil INA 702 multi turn (1 turn, 11 turns) coil
26 Headquarters Teseq AG 4542 Luterbach, Switzerland T F teseq.com China Teseq Company Limited T F teseq.com Germany Teseq GmbH T F teseq.com Singapore Teseq Pte Ltd. T F teseq.com Taiwan Teseq Ltd. T F teseq.com Manufacturer Teseq AG 4542 Luterbach, Switzerland T F teseq.com France Teseq Sarl T F teseq.com Japan Teseq K.K. T F eseq.com Switzerland Teseq AG T F teseq.com UK Teseq Ltd. T F teseq.com USA Teseq Inc. T F Toll free teseq.com To find your local partner within Teseq s global network, please go to March 2012 Teseq Specifications subject to change without notice. Teseq is an ISOregistered company. Its products are designed and manufactured under the strict quality and environmental requirements of the ISO This document has been carefully checked. However, Teseq does not assume any liability for errors or inaccuracies.
1 MD 200A High voltage differential probe User Manual 601-265B MD 200A High voltage differential probe User Manual MD 200A high voltage differential probe contents 1 Safety terms and symbols 5 2 Safety
1 MD 103 ESD Target Set User Manual 601-266B MD 103 ESD Target Set User Manual MD 103 ESD target set contents 1 Safety terms and symbols 5 2 The ESD target set 6 3 Handling of the target set 7 4 Requirements
1 CDN 117 / 117-M CDN 118-T4 / 118-EXT-1 SIGNAL & DATA LINE COUPLING NETWORKS USER MANUAL 601-142M CDN 117 / 117-M CDN 118-T4 / 118-EXT-1 SIGNAL & DATA LINE COUPLING NETWORKS USER MANUAL CDN 117 / 117-M
User s Guide RP5600 Passive Probe July 2010 RIGOL Technologies, Inc. Guaranty and Declaration Copyright 2010 RIGOL Technologies, Inc. All Rights Reserved. Trademark Information RIGOL is a registered trademark
Table of Contents 3 80i-110s AC/DC CURRENT PROBE Users Manual November 1996 1994, 1996 Fluke Corporation. All rights reserved. All product names are trademarks of their respective companies. Printed in
PHV 1000-3-RO High Voltage Passive Probe Instruction Manual Copyright 2012 PMK GmbH All rights reserved. Information in this publication supersedes that in all previously published material. Specifications
AC 3 Active Antenna Combiner Instruction manual Contents Contents Important safety instructions... 2 The AC 3 active transmitter combiner... 4 Delivery includes... 4 Operating controls... 5 Block diagram...
TETRIS 1500 High Impedance Active Probe User's Guide DO177-1 TETRIS 1500 Copyright 2010 Ltd. All rights reserved. Information in this publication supersedes that in all previously published material. Specifications
Emerging Standards for EMC Emissions & Immunity Requirements for Industrial, Scientific, Medical & Information Technology Equipment CE Marking requirements are the path to increased market access Powerful
Radio System Strobe Wizard Plus Freemask User manual Translation of the original German user manual Doc. No.: 900.0509.00 Version: 09/2017 Contents Information about this manual and about the manufacturer...
Trantec ANTENNA DISTRIBUTOR INSTRUCTION MANUAL S5-ADU Thank you for purchasing TRANTEC Antenna Distributor. Please carefully follow the instructions in this manual to ensure long, trouble-free use of your
V1.3 TBLC08 The TBLC08 is a Line Impedance Stabilization Network for the measurement of line-conducted interference within the range of 9kHz to 30MHz, according to the CISPR16 standard. The device is designed
Manual For Operation AN 200 Series Voltage Transient Emission Test AN 200 AN 200B The AN 200 is used to evaluate automotive electrical and electronic components for conducted emissions of transients along
USER MANUAL Important Safety Information Before using Zuma R300, please be sure to read all operating instructions carefully. Read, follow, and keep these instructions. Heed all warnings. Do not expose
AC/DC Current Probe GCP-100 QUICK START GUIDE ISO-9001 CERTIFIED MANUFACTURER This manual contains proprietary information, which is protected by copyright. All rights are reserved. No part of this manual
3B SCIENTIFIC PHYSICS Digital Multimeter E 1018832 Instruction sheet 12/16 SD/UD 1 probe 1a Finger guards 2 Measurement socket 10 A for current measurement in 10-A (positive) 3 Measurement socket COM (negative)
Operator s Manual PP016 Passive Probe 2017 Teledyne LeCroy, Inc. All rights reserved. Unauthorized duplication of Teledyne LeCroy documentation materials is strictly prohibited. Customers are permitted
ASX SERIES AC POWER SOURCE OPERATION MANUAL PACIFIC POWER SOURCE ASX-SERIES OPERATION MANUAL FOR THE MODELS 115-ASX, 120-ASX 140-ASX, 160-ASX 315-ASX, 320-ASX 345-ASX, 360-ASX 390-ASX, 3120-ASX PPS PART
INSTRUMENTS, INC. Model 2960AX Disciplined Quartz Frequency Standard 2960AX Section Page Contents 1.0............................. 2......................... Description 2.0.............................
Diagnostics Unit for Electric Charging Stations (Type 2 Connector Socket and Plug) 3-349-878-03 1/3.16 Opening the Instrument / Repairs The instrument may only be opened by authorized, trained personnel
Immunity Testing for the CE Mark Summary The European Union (EU) currently has 25 member countries with 2 additional countries to be added in 2007. The total population at that time will be nearly a half
17035561 001 Seite 2 von 27 Page 2 of 27 TEST SUMMARY 5.1.1 HARMONICS ON AC MAINS RESULT: Passed 5.1.2 VOLTAGE FLUCTUATIONS ON AC MAINS RESULT: Passed 5.1.3 TERMINAL CONTINUOUS DISTURBANCE VOLTAGE AT RESULT:
Technical data Network nalyzer for Low-, Medium- and High-Voltage Networks Model PQ-Box 150 1 Fault detection 1 Evaluation of voltage quality according to EN50160 and IEC61000-2-2 (2-4) 1 FFT nalysis up
Immunity Tests Combination Wave Test System Brief Overview of Phenomena............... 2 Applicable Standards................... 3 Test System Overview.................. 4 Generator Specifications.................
USER'S MANUAL for HIGH VOLTAGE PHASE COMPARATORS PC Series Double Poles Model System Voltage Full Scale MT6k 6.6kV 8kV MT11k 11kV 12kV MT22k 22kV 24kV MT33k 33kV 36kV MT44k 44kV 48kV Exceeds VDE681-5 /
INSTRUCTION MANUAL DM-45 Digital Multimeter Read and understand all of the instructions and safety information in this manual before operating or servicing this tool. Description The Greenlee DM-45 Digital
IEC / EN 61000-4-6 specifies the design and performance of a range of coupling / decoupling networks (CDNs). Each CDN is specific to the type of cable and the intended signal carried on the cable. Teseq
Component Tests Insulation Test System Brief Overview of Phenomena............... 2 Applicable Standards................... 3 Test System Overview.................. 3 Generator Specifications.................
AC-DC power supply module Datasheet production data Features Open frame switch mode power supply European input voltage range Single output 5 V, 8 W peak power, 4 W continuous operating mode EMC compliance
INSTALLATION AND MAINTENANCE MANUAL FOR GROUND MONITOR GM-250 COPYRIGHT 1983 AMERICAN MINE RESEARCH, INC. MANUAL PART NUMBER 180-0036 ORIGINAL: 1-17-83 REVISION: B (8-26-86) NOT TO BE CHANGED WITHOUT MSHA
Original instructions INCA-1 Tina Emergency stop for enclosure installation INCA-1S Tina Safety stop for enclosure installation ABB AB / Jokab Safety Varlabergsvägen 11, SE-434 39 Kungsbacka, Sweden www.abb.com/lowvoltage
Instruction Manual for Digital Grounding Resistance Meter Instruction Manual for Digital Grounding Resistance Meter Table of Contents I. Overview...2 II. Open-case Inspection...3 III. Safety Precautions...4
SIR-WRR1 User's Guide SIRIUS Echo Antenna Signal Repeater System Accessory Desktop SIRIUS Docking Echo Station Antenna FCC NOTICE: This device complies with part 15 of the FCC Rules and with RSS-210 of
Operating Instructions for Pressure Sensor with Ceramic Sensor Element Model: SEN-9601 We don t accept warranty and liability claims neither upon this publication nor in case of improper treatment of the
TA043 100 MHz x10/x100 Differential Probe User s Manual This probe complies with IEC-1010.1, IEC-1010.2 031 CAT III, Pollution Degree 2. 1. Safety terms and symbols Terms appearing in this manual: WARNING
Tornado KT2 - KT2-HV KT2C - KT2C-HV KTL2 - KTL2-HV KTL2C - KTL2C-HV USER GUIDE English KT2 - KT2-HV KTL2 - KTL2-HV KT2C - KT2C-HV KTL2C - KTL2C-HV NEW All Tornado models are also available in a 70V version!
Section : 0 Document : DQM-01 Topic : FORMAT 6 OF PROCUREMENT PROCEDURE OF CPRI (NON-WORKS) Section I(B)-Technical Specification Tender Enquery No PUR/UARC-02/14-15 Name of the Equipment: Damped Oscillatory
Technical Criteria for the Accreditation Of Electromagnetic Compatibility (EMC) And Radio Testing Laboratories ACIL - American Council of Independent Laboratories 1629 K Street, NW, Washington, DC 20006-1633
Lightning Surge Simulator LSS-F03 series Compliant with IEC61000-4-5 Edition 3 requirements Lightning Surge Simulator LSS-F03 series Feature "Output voltage 15kV, current 7500A" which can conduct breakdown
M30 SPOT WELDER INSTRUCTION MANUAL 230V 1PH. We have the right to improve and update the machine. The picture and the content are just for your reference. IN20808 2017 Chief Automotive Technologies CO9910.4
TA 450 MK-X TA 600 MK-X TA 1050 MK-X TA 1400 MK-X TA 2400 MK-X power amplifier user manual Musikhaus Thomann e. K. Treppendorf 30 96138 Burgebrach Germany Telephone: +49 (0) 9546 9223-0 E-mail: email@example.com
DECLARATION OF CONFORMITY Manufacturer s Name: DigiTech Manufacturer s Address: 8760 S. Sandy Parkway Sandy, Utah 84070, USA declares that the product: Product name: Synth Wah Product option: all (requires
CSM-S-11056 TRIGGER DISTRIBUTION MODULE USER S MANUAL 82-0051-000 Release April 7, 2003 VXI Technology, Inc. 2031 Main Street Irvine, CA 92614-6509 (949) 955-1894 bus 2 VXI Technology, Inc. www.vxitech.com
Bulk Current Injection Probe Test Procedure 1 TABLE OF CONTENTS INTRODUCTION 3 GENERAL INFORMATION 4 TEST METHODS 6 SAFETY 8 FIGURES 9 FORMULAS 12 MAINTENANCE 13 WARRANTY 14 2 INTRODUCTION CURRENT PROBE
ELECTROSURGICAL UNIT ANALYZER ESU-2000A USER MANUAL BC BIOMEDICAL ESU-2000A TABLE OF CONTENTS WARNINGS, CAUTIONS, NOTICES... ii DESCRIPTION... 1 OVERVIEW... 2 OPERATING INSTRUCTIONS... 3 MANUAL REVISIONS...
IDEAL INDUSTRIES, INC. TECHNICAL MANUAL MODEL: 61-340 MODEL: 61-342 Multimeter Service Information The Service Information provides the following information: Precautions and safety information Specifications
Smart-UPS On-Line Isolation and Step-Down Transformer SRT5KTF Safety Messages Read the instructions carefully to become familiar with the equipment before attempting to install, operate, service or maintain
M I C R O O H M M E T E R Reference Manual IBEKO POWER AB Sweden 2 IBEKO POWER AB Manual Version:. MV.01 IBEKO POWER AB 2003 This Reference Manual is a publication of IBEKO POWER AB, 18 125 Lidingö, Sweden.
Grunge Distortion Owner s Manual DECLARATION OF CONFORMITY Manufacturer s Name: DigiTech Manufacturer s Address: 8760 S. Sandy Parkway Sandy, Utah 84070, USA declares that the product: Product name: DigiTech
INSTALLATION MANUAL Model: Smart Analyzer Manufacturer: Smart Impulse Power meter with consumption breakdown by use 03/12/13 Table of contents Table of contents... 2 1. Introduction... 3 2. Installation
European Union (EU) Council Directive 89/336/EEC Electromagnetic Compatibility (EMC) Test Report 2620 Modular Measurement and Control System Sensoray March 31, 2006 April 4, 2006 Tests Conducted by: ElectroMagnetic
Established 1981 Advanced Test Equipment Rentals www.atecorp.com 800-404-ATEC (2832) A.H. Systems Model TDS-536 Tuned Dipole Set TDS-536 TV Dipole Set Operation Manual 1 TABLE OF CONTENTS WARRANTY 2 INTRODUCTION
SPLIT.CONVERTER User's Manual Version 2.0 Copyright All rights reserved. Permission to reprint or electronically reproduce any document or graphic in whole or in part for any reason is expressly prohibited,
Active Transmitter Combiner 8:1 AC 3000 Bedienungsanleitung Instructions for use Notice d emploi Istruzioni per l uso Instrucciones para el uso Gebruiksaanwijzing 8 Bedienungsanleitung...3 Instructions
A V T E C H E L E C T R O S Y S T E M S L T D. N A N O S E C O N D W A V E F O R M E L E C T R O N I C S S I N C E 1 9 7 5 P.O. BOX 265 OGDENSBURG, NY U.S.A. 13669-0265 TEL: 888-670-8729 (USA & Canada)
MegaPulse Defib-5 Impulse Tester Instruction Manual COMPLIANCE WEST USA Dear Customer: Congratulations! Compliance West USA is proud to present you with your MegaPulse Defib-5 Impulse Tester. Your instrument
User manual Probe set set ESD generator (IEC 61000-4-2) Copyright January 2017 LANGER GmbH 2017.01.09 User manual Table of contents: Page 1 ESD generator (IEC 61000-4-2) 3 1.1 Design and function of the
EMC Seminar Series All about EMC Testing and Measurement Seminar 1 Introduction to EMC Conducted Immunity Jeffrey Tsang Organized by : Department of Electronic Engineering 1 Basic Immunity Standards: IEC
HX-3 Headphone Distribution Amplifier User Guide and Technical Information Sound Devices, LLC 300 Wengel Drive Reedsburg, WI USA +1 (608) 524-0625 fax: +1 (608) 524-0655 Toll-Free: (800) 505-0625 www.sounddevices.com
Instruction Manual P7350SMA 5 GHz Differential Probe 071-1264-01 Warning The servicing instructions are for use by qualified personnel only. To avoid personal injury, do not perform any servicing unless
NSPL-500 AIS/VHF antenna splitter User Manual ENGLISH www.bandg.com www.simrad-yachting.com www.lowrance.com Preface As Navico is continuously improving this product, we retain the right to make changes
AT-1000 Advanced Wire Tracer Users Manual AT-1000 Advanced Wire Tracer English Users Manual AT1000_Rev001 2008 Amprobe Test Tools. All rights reserved. Limited Warranty and Limitation of Liability Your
99 Washington Street Melrose, MA 02176 Fax 781-665-0780 TestEquipmentDepot.com #61-732 #61-736 400 AAC Clamp Meter Instruction Manual AC HOLD APO DC KMΩ mva WARNING Read First: Safety Information Understand
MZ2 HEADPHONE AMPLIFIER, PREAMP, & STEREO AMPLIFIER USER GUIDE Linear Tube Audio Takoma Park, MD, USA WARNING: For safety, the cover of this amplifier should be secured at all times. DC voltages as high
Model 3101, 3102 and 3103 Conical Log-Spiral Antennas MANUAL EMC TEST SYSTEMS, L.P. SEPTEMBER 2002 EMC Test Systems, L.P. reserves the right to make changes to any products herein to improve functioning,
UT232 Operating Manual Digital Power Clamp Meter Model UT232 OPERATING MANUAL TABLE OF CONTENTS TITLE PAGE Overview Unpacking Inspection Safety Information Rules For Safe Operation International Electrical
Model 79 VHF Radio User's Manual ALL WEATHER INC 65 NATIONAL DRIVE SACRAMENTO, CA 95834 WWW.ALWEATHERINC.COM 79 VHF RADIO USER'S MANUAL CONTENTS INTRODUCTION... Description... Transmitter Module... Power