Advanced Camera and Image Sensor Technology. Steve Kinney Imaging Professional Camera Link Chairman

Save this PDF as:

Size: px
Start display at page:

Transcription

1 Advanced Camera and Image Sensor Technology Steve Kinney Imaging Professional Camera Link Chairman

2 Content Physical model of a camera Definition of various parameters for EMVA1288 EMVA1288 and image quality CCD and CMOS sensors, image artifacts Noise in cameras Spectral response and penetration depth Area scan and line scan Spatial trigger Monochrome and color Single-, dual-, tri-linear sensors, TDI Mechanical and optical pixel size limitations Camera mounting standards

3 Physical Model of a Camera

4 What are the Important Things? Quantum Efficiency When light is emitted from a light source, then passes through the optics and hits the Silicon, electrons will be generated. The probability of how many electrons will be generated per 100 photons is called Quantum Efficiency (QE). QE = number of electrons / number of photons Typical values are 30 to 60 %.

5 What are the Important Things? Full Well and Saturation Capacity Each pixel has a maximum capacity to collect electrons. But we will limit that to a lower maximum, because non-linear effects occur as the maximum capacity is reached. A pixel may have a saturation capactity of electrons. A glass of wheat beer contains approx drops of beer. A glass of wine or champagne might have drops. Full Well Capacity Saturation Capacity

6 What are the Important Things? Signal to Noise Ratio When refilling the pixel to the same previous value, there is a slight jitter in the amount of electrons created. This noise is equivalent to the square root of the number of electrons. noise = e Refilled Level So the maximum Signal to Noise Ratio (SSS mmm ) is: SSS mmm = e mmm e mmm = sss.ccc. sss.ccc. = sss. ccc. 5000e = approx. 224 Refilled beer glasses do not have the same content.

7 What are the Important Things? Dark noise There is always some residual noise, even when no light hits the sensor. Depending on the sensor, this is between 8 to 110 electrons. Compare this to the remaining drops of beer in an empty glass. Full Well Capacity Saturation Capacity Heat is a contributor. Dark Noise

8 What are the Important Things? Detection Limit If the signal is the same as the dark noise, we call this detection limit. CCD sensors usually have a detection limit of 8 to 25 electrons. This corresponds to photons. CMOS sensors will start at 10 to 110 electrons, or 's of photons. Detection Limit Dark Noise

9 What are the Important Things? Dynamic Range Dynamic range is the ratio between a full glass of beer and an empty glass. Saturation Capacity So, a pixel's dynamic range is: DDD = sss.ccc. dddd nnnnn = 50000e 25e = 200 Dark Noise A glass of beer will always have dynamics!

10 The Diagram db 50 Signal to Noise Ratio bits Typical CCD 30 5 Typical CMOS Dynamic Range Number of Photons

11 How to Understand the Diagram? The x-axis We use a logarithmic scale to display the large range of values we are concerned with. A linear scale will not represent this properly. example: Imagine a scale that can display both kilometers and millimeters at the same time with the same precision. Set Quantum Efficiency to 50%, 100 photons will generate 50 electrons Number of Photons Number of Electrons

12 How to Understand the Diagram? The y-axis Signal to Noise Ratio db bits Actual SNR is also displayed on a logarithmic scale. A linear scale always adds the same unit. A logarithmic scale always multiplies by the same unit. Bits (2 n ) and decibel (dd) are logarithmic scales. Multiplying SNR by 2 = 1 bit = 6.02 db increasing by 1 bit = increasing by 6 dd.

13 How to Understand the Diagram? Image Quality Signal to Noise Ratio db bits An excellent image is SNR = 40 or better. A good image quality is SNR =

14 Examples: Sony ICX285 Saturation capacity: electrons Dark noise: 9 electrons 545 nm: 56 % (electrons per photons) Saturation reached with: photons Detection limit: photons Max SNR: 2 7 ) bits 43 db ( = Dynamic range: saturation cap./dark noise =

15 Examples: Sony ICX274 Dynamic range: saturation cap./dark noise = 16000/16 = bits 60 db ( = 2 10 ) Saturation capacity: 8000 electrons Dark noise: 8 electrons 545 nm: 50 % (electrons per photons) Saturation with photons Detection limit photons Max SNR: bits 39 db (89 is between 64 = 2 6 and 128 = 2 7 )

16 Limitations in the Diagram Image quality depends on the signal to noise ratio (SNR). Total noise consists of: Temporal Noise (~ number of photons) Dark Noise (~ number of photons) SNR ~ # Photons 100 ~# Photons Number of Photons

17 Today's CMOS Sensors Today's CMOS sensors are able to achieve the same low read noise as CCD sensors (about 7 to 10 electrons). In sensors with a rolling shutter, it can be even less, due to less transistors being used.

18 Progressive Scan vs. Interlaced Scan Why is progressive scan important? Progressive scan gives you a full image for all lines (or pixels). Interlaced scan uses the odd & even lines separately. This can result in a fringe effect like you see in a TV's news ticker. Progressive Scan: Interlaced Scan:

19 Standard Resolution, Progressive Scan

20 Standard Resolution, Interlaced

21 Interlaced Artifacts

22 Global Shutter vs. Rolling Shutter Why is global shutter important? Global shutter has the start and stop of exposure time for all pixels at the same time. A rolling shutter starts and stops exposure one line after the other (sequentially). This is like a flatbed scanner. Global Shutter: Rolling Shutter:

23 Flash Window and Hybrid Mode Rolling shutter cameras can offer a offer a flash window. When the last line starts exposure, the flash window opens until the first line stops exposure. Using a strobe light during this time can freeze the image, effectively becoming a global shutter. Flash Window: Some "hybrid" sensors exist that start exposure like a global shutter, and end exposure like a rolling shutter. Downside is uneven exposure times.

24 Standard Resolution, Global Shutter

25 Standard Resolution, Rolling Shutter

26 Global Shutter Artifacts Source: Wikipedia

27 Resolution What is the best or required resolution? Most people think, the higher the resolution, the better. This is not true for machine vision, because all data have to be calculated and displayed. Often the higher sensor resolution results in a smaller pixel size. Smaller pixels are often below optical resolution and have a lower saturation capacity. A lower saturation capacity results in a worse signal-to-noise ratio and a lower dynamic range. Saturation Capacity Refilled Level Dark Noise

28 Standard Resolution

29 High Resolution

30 Column Noise Usually a CMOS sensor has one ADC (analog to digital converter) per column, a CCD has only one (per tap), meaning typically 1 to 4 ADCs for a CCD. In the case of a 4 Mpixel CMOS camera, there are 2048 ADCs. All of them will be slightly different. A correction done in the dark and without power to the sensor can calibrate all ADCs.

31 FPN or DSNU Fixed pattern noise (FPN) or Dark Signal Non-Uniformity (DSNU) occurs, when all voltages are applied to the sensor, but no light is collected. This can be corrected by adding a certain offset per pixel. CMOS s FPN is up to 10x worse than a CCD s, but latest designs are rather equal!! CMOS CCD

32 PRNU Photo Response Non-Uniformity (PRNU) describes the different gain per pixel. With an illuminated sensor, the different (conversion-) gains can be corrected to create an homogeneous image unter bright light conditions. CMOS and CCD behave the same. Raw data Through a low pass shading corrected

33 Spectral Response The spectral response depends on wavelength and sensor type. CMOS sensors often have a wavy curve due to interferometric issues. 60% Total Quantum Efficiency [%] 50% 40% 30% 20% 10% 0% CMOS sensors Wavelength [nm]

34 Penetration Depth Cameras cover a spectral range of the visible (VIS) and near infrared (NIR), wavelengths from 400 to 1100 nm λ Wavelength [nm] Wavelength : 1/100 of a human hair Penetration depth: 1/10000 of a human hair

35 Area Scan and Line Scan Scan types can be separated by area scan and line scan. Area scan is analogous to a digital still camera. 1 shot and the 2D image is taken. As an example: resolution 1300 x 1000 pixels. After an exposure time of 10 ms, everything is captured. All pixels have an exposure time of 10 ms. Line scan is analogous to a Xerox machine. One line after the other is taken to get the total image. As an example: resolution 1000 pixels, 1300 lines. With a total exposure time of 10 ms the image is taken, BUT: every single line (or each pixel) has only an exposure time of 7.7 µs! This is a very short exposure time You need much more light!

36 The Need of Spatial Trigger Camera speed independent of object speed can distort. (e.g. acceleration after a traffic light). Only a spatial trigger gives the right information. It synchronizes the camera with the object (e.g., an encoder).

37 Color: Area Scan Color on area scan can be taken either with a: 3 CCD setup: a beamsplitter separates the colors to three different CCDs. Advantage: Every pixel has the full color information. Disadvantages: Expensive, special lenses, alignment, color shades. In most cameras a Bayer pattern is used.

38 Color: Line Scan I Single Line Dual Line

39 Color: Line Scan II Triple Line 3 CCD Line

40 How to Match RGB to One Image I The object on the conveyer belt is moved beneath the camera.

41 How to Match RGB to One Image II Take 1, 8, and 15, but there is a color shade within one group RGB. Lines 1, 8, 15: 100% 100% 100% Lines 2, 9, 16: 100% 80% 20% Lines 3, 10, 17: 0% 0% 10% Lines 4, 11, 18: 60% 100% 100%

42 What is Possible with the Camera? Raw Image Raw Corrected Corrected Image

43 TDI Line Scan Sensor TDI stands for Time Delay and Integration. The object is exposed several times across many lines, charges are accumulated and shifted simultaneously with the trigger. Signal t 1 is taken t 2 N times, t 3 Noise t 4 reduces t N by Read N. Out Time

44 Depth of Focus (DOF) The depth of focus (DOF) depends on the pixel size, the diameter of the iris and the focal length of the lens.

45 Depth of Focus (DOF) The depth of focus (DOF) depends on the pixel size, the diameter of the iris and the focal length of the lens. d: Diameter Depth of Focus Pixelsize f: Focal Length DOF = 2 * Pixelsize * f / d = 2 * Pixelsize * F/#

46 Tilt of the Sensor The sensor has to be aligned perpendicular to the optical axis. Angle Given max Depth of Focus The whole sensor has to be taken into account

47 Examples of Geometrical DOF Real numbers: Pixel size = 5 µm F/# = 4 Depth of Focus: ±20 µm (40µm) Real numbers: Pixel size = 4 µm F/# = 2 d: Diameter f: Focal Length Depth of Focus Pixelsize Depth of Focus: ±8 µm (16 µm)

48 Diffraction Due to the physical structure of light as an electromagnetic wave, the rays are blurred by diffraction. Diffraction depends on the F-number and wavelength λ. Ø Airy = 2.44 * λ * F/# As a rule of thumb, the diameter of the Airy disc is F/# in microns, like F/# is 4, the diameter of the Airy disc is approx. 4 µm. Barely resolved resolved No longer

49 Real Spot-size and DOF The real spot-size and DOF is the geometrics folded with the diffraction. To make life a little easier we will treat the diameter as independent errors (deviations): Ø total = Ø geom 2 + Ø diff 2 Total DOF Geometric DOF

50 Errors of Higher Order In case of an 8k line scan sensor with 10 µm pitch we might have a further issue: The sensor might be bent by Δz = 50 to 80 µm (acceptable) Optical Axis Sensor Focal Plane The lens (e.g. 5.6/90mm) will cause a field curvature of Δz = 50 to 100 µm, whereas the astigmatism might occur and the meridional and sagittal focal plane might run the opposite direction. Δz = 50 to 80 µm

51 Field Curvature and Astigmatism In case of an 8k line scan sensor with 10 µm pitch we might have a difference of about 100 µm between both focal planes (meridional and sagittal).

52 Pixel Size Pixels are mainly between 10 to 3.5 µm. Trend is toward smaller pixels, because they allow higher resolution with less silicon More sensors from a single wafer lower cost. "Reasonable" limit: 5 µm for monochrome, 2.5 µm for color. Full-Well capacity (saturation capacity, resp.) is lower for smaller pixels. Therefore the max SNR is not as good as on a larger pixel. \$ System Optics Sensor µm

53 CCD and CMOS Sensors Sony: Diagonal: 1 Tube Deflection yoke Image Format The image size in inches relates back to the tube camera. The image format of a tube which could be placed in a 1 deflection coil was called 1 format. The active image size is 16 mm in diagonal. It is the same as for a 16 mm film format. BIPM (Bureau International des Poids & Mesures) recommend to use metric designation. Below is the most common image formats shown. 1 format 2/3 format 1/2 format 1/3 format Type 1 Type 2/3 Type 1/2 Type 1/3 16 mm 11 mm 8 mm 6 mm Image size 12.8 mm 8.8 mm 6.4 mm 4.4 mm

54 Optical Sizes In case of C-mount lenses are usually restricted to 2/3 optical format imagers. Longer focal length or higher f-stop lenses may work up to 1 format The CMOSIS sensor 2MP is 2048 x 1088 pixels with a mm in diagonal and fits 2/3 optical format. The 4MP sensor is 2048 x 2048 pixels, with a diagonal of mm requiring a 1 format lens. The 4 MP sensor must use proper lenses! Otherwise there is a shading or vignetteing seen in the image. CCD 1 ^ = 25.4 mm CMOS CMOS 16 mm sensor with 1 format 16 mm sensor with 2/3 format will cause vignetting

55 Recommended Mechanical Interfaces (Mounts) Class Min. Image Size [mm] Max. Image Size [mm] Optical Size [type] Mechanical Interface (1st Choice) I 0 4 ¼ C-, CS-, NF-, S-Mount II C-, CS-, NF-, S-Mount III F-Mount, 48 mm Ring, M42 x 1, M48 x 0.75 IV M58 x 0.75 (and F- Mount if possible) V M72 x 0.75 VI M95 x 1 VII M105 x 1 Please check with JIIA LER for more details.

56 Questions?

57 Contact Information Steve Kinney Imaging Professional AIA Camera Link Chairman Phone: +1 (408)

e2v Launches New Onyx 1.3M for Premium Performance in Low Light Conditions

e2v Launches New Onyx 1.3M for Premium Performance in Low Light Conditions e2v s Onyx family of image sensors is designed for the most demanding outdoor camera and industrial machine vision applications,

Basler. Line Scan Cameras

Basler Line Scan Cameras Next generation CMOS dual line scan technology Up to 140 khz at 2k or 4k resolution, up to 70 khz at 8k resolution Color line scan with 70 khz at 4k resolution High sensitivity

Basler aca gm. Camera Specification. Measurement protocol using the EMVA Standard 1288 Document Number: BD Version: 01

Basler aca5-14gm Camera Specification Measurement protocol using the EMVA Standard 188 Document Number: BD563 Version: 1 For customers in the U.S.A. This equipment has been tested and found to comply with

Applications for cameras with CMOS-, CCD- and InGaAssensors. Jürgen Bretschneider AVT, 2014

Applications for cameras with CMOS-, CCD- and InGaAssensors Jürgen Bretschneider AVT, 2014 Allied Vision Technologies Profile Foundation: 1989,Headquarters: Stadtroda (Thüringen), Employees: aprox. 265

Basler ral km. Camera Specification. Measurement protocol using the EMVA Standard 1288 Document Number: BD Version: 01

Basler ral8-8km Camera Specification Measurement protocol using the EMVA Standard 188 Document Number: BD79 Version: 1 For customers in the U.S.A. This equipment has been tested and found to comply with

IMAGE SENSOR SOLUTIONS. KAC-96-1/5" Lens Kit. KODAK KAC-96-1/5" Lens Kit. for use with the KODAK CMOS Image Sensors. November 2004 Revision 2

KODAK for use with the KODAK CMOS Image Sensors November 2004 Revision 2 1.1 Introduction Choosing the right lens is a critical aspect of designing an imaging system. Typically the trade off between image

CMOS Today & Tomorrow

CMOS Today & Tomorrow Uwe Pulsfort TDALSA Product & Application Support Overview Image Sensor Technology Today Typical Architectures Pixel, ADCs & Data Path Image Quality Image Sensor Technology Tomorrow

BIG PIXELS VS. SMALL PIXELS THE OPTICAL BOTTLENECK. Gregory Hollows Edmund Optics

BIG PIXELS VS. SMALL PIXELS THE OPTICAL BOTTLENECK Gregory Hollows Edmund Optics 1 IT ALL STARTS WITH THE SENSOR We have to begin with sensor technology to understand the road map Resolution will continue

Data Sheet SMX-160 Series USB2.0 Cameras

Data Sheet SMX-160 Series USB2.0 Cameras SMX-160 Series USB2.0 Cameras Data Sheet Revision 3.0 Copyright 2001-2010 Sumix Corporation 4005 Avenida de la Plata, Suite 201 Oceanside, CA, 92056 Tel.: (877)233-3385;

Ideal for display mura (nonuniformity) evaluation and inspection on smartphones and tablet PCs.

2D Color Analyzer 8 Ideal for display mura (nonuniformity) evaluation and inspection on smartphones and tablet PCs. Accurately and easily measures the distribution of luminance and chromaticity. Advanced

Digital camera. Sensor. Memory card. Circuit board

Digital camera Circuit board Memory card Sensor Detector element (pixel). Typical size: 2-5 m square Typical number: 5-20M Pixel = Photogate Photon + Thin film electrode (semi-transparent) Depletion volume

Sensors and Sensing Cameras and Camera Calibration

Sensors and Sensing Cameras and Camera Calibration Todor Stoyanov Mobile Robotics and Olfaction Lab Center for Applied Autonomous Sensor Systems Örebro University, Sweden todor.stoyanov@oru.se 20.11.2014

Ideal for display mura (nonuniformity) evaluation and inspection on smartphones and tablet PCs.

2D Color Analyzer Ideal for display mura (nonuniformity) evaluation and inspection on smartphones and tablet PCs. Accurately and easily measures the distribution of luminance and chromaticity. The included

brief history of photography foveon X3 imager technology description

brief history of photography foveon X3 imager technology description imaging technology 30,000 BC chauvet-pont-d arc pinhole camera principle first described by Aristotle fourth century B.C. oldest known

Digital Photographs, Image Sensors and Matrices

Digital Photographs, Image Sensors and Matrices Digital Camera Image Sensors Electron Counts Checkerboard Analogy Bryce Bayer s Color Filter Array Mosaic. Image Sensor Data to Matrix Data Visualization

Camera Image Processing Pipeline

Lecture 13: Camera Image Processing Pipeline Visual Computing Systems Today (actually all week) Operations that take photons hitting a sensor to a high-quality image Processing systems used to efficiently

Technical Note CMOS, EMCCD AND CCD CAMERAS FOR LIFE SCIENCES Camera Test Protocol Introduction The detector is one of the most important components of any microscope system. Accurate detector readings

CMOS Image Sensors in Cell Phones, Cars and Beyond. Patrick Feng General manager BYD Microelectronics October 8, 2013

CMOS Image Sensors in Cell Phones, Cars and Beyond Patrick Feng General manager BYD Microelectronics October 8, 2013 BYD Microelectronics (BME) is a subsidiary of BYD Company Limited, Shenzhen, China.

Evaluating Commercial Scanners for Astronomical Images. The underlying technology of the scanners: Pixel sizes:

Evaluating Commercial Scanners for Astronomical Images Robert J. Simcoe Associate Harvard College Observatory rjsimcoe@cfa.harvard.edu Introduction: Many organizations have expressed interest in using

The future of the broadloom inspection

Contact image sensors realize efficient and economic on-line analysis The future of the broadloom inspection In the printing industry the demands regarding the product quality are constantly increasing.

when it s too fast to see, and too important not to. NOTES/ALERTS For the most current version visit www.phantomhighspeed.com Subject to change Rev April 2016 Boosting Sensitivity In this series of articles,

2013 LMIC Imaging Workshop. Sidney L. Shaw Technical Director. - Light and the Image - Detectors - Signal and Noise

2013 LMIC Imaging Workshop Sidney L. Shaw Technical Director - Light and the Image - Detectors - Signal and Noise The Anatomy of a Digital Image Representative Intensities Specimen: (molecular distribution)

CS 443: Imaging and Multimedia Cameras and Lenses

CS 443: Imaging and Multimedia Cameras and Lenses Spring 2008 Ahmed Elgammal Dept of Computer Science Rutgers University Outlines Cameras and lenses! 1 They are formed by the projection of 3D objects.

TDI Imaging: An Efficient AOI and AXI Tool

TDI Imaging: An Efficient AOI and AXI Tool Yakov Bulayev Hamamatsu Corporation Bridgewater, New Jersey Abstract As a result of heightened requirements for quality, integrity and reliability of electronic

IMAGE FORMATION. Light source properties. Sensor characteristics Surface. Surface reflectance properties. Optics

IMAGE FORMATION Light source properties Sensor characteristics Surface Exposure shape Optics Surface reflectance properties ANALOG IMAGES An image can be understood as a 2D light intensity function f(x,y)

Cameras As Computing Systems

Cameras As Computing Systems Prof. Hank Dietz In Search Of Sensors University of Kentucky Electrical & Computer Engineering Things You Already Know The sensor is some kind of chip Most can't distinguish

Astronomical Cameras

Astronomical Cameras I. The Pinhole Camera Pinhole Camera (or Camera Obscura) Whenever light passes through a small hole or aperture it creates an image opposite the hole This is an effect wherever apertures

Measurement of the Modulation Transfer Function (MTF) of a camera lens. Laboratoire d Enseignement Expérimental (LEnsE)

Measurement of the Modulation Transfer Function (MTF) of a camera lens Aline Vernier, Baptiste Perrin, Thierry Avignon, Jean Augereau, Lionel Jacubowiez Institut d Optique Graduate School Laboratoire d

Signal-to-Noise Ratio (SNR) discussion

Signal-to-Noise Ratio (SNR) discussion The signal-to-noise ratio (SNR) is a commonly requested parameter for hyperspectral imagers. This note is written to provide a description of the factors that affect

Time Delay Integration (TDI), The Answer to Demands for Increasing Frame Rate/Sensitivity? Craige Palmer Assistant Sales Manager

Time Delay Integration (TDI), The Answer to Demands for Increasing Frame Rate/Sensitivity? Craige Palmer Assistant Sales Manager Laser Scanning Microscope High Speed Gated PMT Module High Speed Gating

INTRODUCTION TO CCD IMAGING

ASTR 1030 Astronomy Lab 85 Intro to CCD Imaging INTRODUCTION TO CCD IMAGING SYNOPSIS: In this lab we will learn about some of the advantages of CCD cameras for use in astronomy and how to process an image.

Using molded chalcogenide glass technology to reduce cost in a compact wide-angle thermal imaging lens

Using molded chalcogenide glass technology to reduce cost in a compact wide-angle thermal imaging lens George Curatu a, Brent Binkley a, David Tinch a, and Costin Curatu b a LightPath Technologies, 2603

Image acquisition. In both cases, the digital sensing element is one of the following: Line array Area array. Single sensor

Image acquisition Digital images are acquired by direct digital acquisition (digital still/video cameras), or scanning material acquired as analog signals (slides, photographs, etc.). In both cases, the

Lecture 4: Geometrical Optics 2. Optical Systems. Images and Pupils. Rays. Wavefronts. Aberrations. Outline

Lecture 4: Geometrical Optics 2 Outline 1 Optical Systems 2 Images and Pupils 3 Rays 4 Wavefronts 5 Aberrations Christoph U. Keller, Leiden University, keller@strw.leidenuniv.nl Lecture 4: Geometrical

Application Note. Digital Low-Light CMOS Camera. NOCTURN Camera: Optimized for Long-Range Observation in Low Light Conditions

Digital Low-Light CMOS Camera Application Note NOCTURN Camera: Optimized for Long-Range Observation in Low Light Conditions PHOTONIS Digital Imaging, LLC. 6170 Research Road Suite 208 Frisco, TX USA 75033

ROAD TO THE BEST ALPR IMAGES

ROAD TO THE BEST ALPR IMAGES INTRODUCTION Since automatic license plate recognition (ALPR) or automatic number plate recognition (ANPR) relies on optical character recognition (OCR) of images, it makes

Pixel CCD RASNIK. Kevan S Hashemi and James R Bensinger Brandeis University May 1997

ATLAS Internal Note MUON-No-180 Pixel CCD RASNIK Kevan S Hashemi and James R Bensinger Brandeis University May 1997 Introduction This note compares the performance of the established Video CCD version

INTRODUCTION THIN LENSES. Introduction. given by the paraxial refraction equation derived last lecture: Thin lenses (19.1) = 1. Double-lens systems

Chapter 9 OPTICAL INSTRUMENTS Introduction Thin lenses Double-lens systems Aberrations Camera Human eye Compound microscope Summary INTRODUCTION Knowledge of geometrical optics, diffraction and interference,

ME 6406 MACHINE VISION. Georgia Institute of Technology

ME 6406 MACHINE VISION Georgia Institute of Technology Class Information Instructor Professor Kok-Meng Lee MARC 474 Office hours: Tues/Thurs 1:00-2:00 pm kokmeng.lee@me.gatech.edu (404)-894-7402 Class

FRAUNHOFER AND FRESNEL DIFFRACTION IN ONE DIMENSION

FRAUNHOFER AND FRESNEL DIFFRACTION IN ONE DIMENSION Revised November 15, 2017 INTRODUCTION The simplest and most commonly described examples of diffraction and interference from two-dimensional apertures

MSE 595T Transmission Electron Microscopy. Laboratory III TEM Imaging - I

MSE 595T Basic Transmission Electron Microscopy TEM Imaging - I Purpose The purpose of this lab is to: 1. Make fine adjustments to the microscope alignment 2. Obtain a diffraction pattern 3. Obtain an

Digital Cameras The Imaging Capture Path

Manchester Group Royal Photographic Society Imaging Science Group Digital Cameras The Imaging Capture Path by Dr. Tony Kaye ASIS FRPS Silver Halide Systems Exposure (film) Processing Digital Capture Imaging

Lecture PowerPoint. Chapter 25 Physics: Principles with Applications, 6 th edition Giancoli

Lecture PowerPoint Chapter 25 Physics: Principles with Applications, 6 th edition Giancoli 2005 Pearson Prentice Hall This work is protected by United States copyright laws and is provided solely for the

Experiment 1: Fraunhofer Diffraction of Light by a Single Slit

Experiment 1: Fraunhofer Diffraction of Light by a Single Slit Purpose 1. To understand the theory of Fraunhofer diffraction of light at a single slit and at a circular aperture; 2. To learn how to measure

Resolution test with line patterns

Resolution test with line patterns OBJECT IMAGE 1 line pair Resolution limit is usually given in line pairs per mm in sensor plane. Visual evaluation usually. Test of optics alone Magnifying glass Test

Figure 1 HDR image fusion example

TN-0903 Date: 10/06/09 Using image fusion to capture high-dynamic range (hdr) scenes High dynamic range (HDR) refers to the ability to distinguish details in scenes containing both very bright and relatively

Reflectors vs. Refractors

1 Telescope Types - Telescopes collect and concentrate light (which can then be magnified, dispersed as a spectrum, etc). - In the end it is the collecting area that counts. - There are two primary telescope

Beam Profiling. Introduction. What is Beam Profiling? by Michael Scaggs. Haas Laser Technologies, Inc.

Beam Profiling by Michael Scaggs Haas Laser Technologies, Inc. Introduction Lasers are ubiquitous in industry today. Carbon Dioxide, Nd:YAG, Excimer and Fiber lasers are used in many industries and a myriad

OFFSET AND NOISE COMPENSATION

OFFSET AND NOISE COMPENSATION AO 10V 8.1 Offset and fixed pattern noise reduction Offset variation - shading AO 10V 8.2 Row Noise AO 10V 8.3 Offset compensation Global offset calibration Dark level is

IPD3. Imaging Photon Detector APPLICATIONS KEY ATTRIBUTES

Imaging Photon Detector The Photek IPD3 is based on a true single photon counting sensor that uniquely provides simultaneous position and timing information for each detected photon. The camera outputs

Phase One 190MP Aerial System

White Paper Phase One 190MP Aerial System Introduction Phase One Industrial s 100MP medium format aerial camera systems have earned a worldwide reputation for its high performance. They are commonly used

Presented by Jerry Hubbell Lake of the Woods Observatory (MPC I24) President, Rappahannock Astronomy Club

Presented by Jerry Hubbell Lake of the Woods Observatory (MPC I24) President, Rappahannock Astronomy Club ENGINEERING A FIBER-FED FED SPECTROMETER FOR ASTRONOMICAL USE Objectives Discuss the engineering

F400. Detects subtle color differences. Color-graying vision sensor. Features

Color-graying vision sensor Detects subtle color differences Features In addition to regular color extraction, the color-graying sensor features the world's first color-graying filter. This is a completely

sensicam em electron multiplication digital 12bit CCD camera system

sensicam em electron multiplication digital 12bit CCD camera system electron multiplication gain of up to 1000 superior resolution (1004 1002 pixel) for EMCCD extremely low noise < 1e excellent quantum

Noise and ISO. CS 178, Spring Marc Levoy Computer Science Department Stanford University

Noise and ISO CS 178, Spring 2014 Marc Levoy Computer Science Department Stanford University Outline examples of camera sensor noise don t confuse it with JPEG compression artifacts probability, mean,

Pregius CMOS Global Shutter

SENSOR INSIGHTS: SONY Pregius CMOS Global Shutter The state of the art in machine vision image sensors Introduction to the SONY Pregius Series THE LAUNCH OF THE SONY PREGIUS SERIES IN LATE 2013 MARKED

PHY 431 Homework Set #5 Due Nov. 20 at the start of class

PHY 431 Homework Set #5 Due Nov. 0 at the start of class 1) Newton s rings (10%) The radius of curvature of the convex surface of a plano-convex lens is 30 cm. The lens is placed with its convex side down

A 120dB dynamic range image sensor with single readout using in pixel HDR

A 120dB dynamic range image sensor with single readout using in pixel HDR CMOS Image Sensors for High Performance Applications Workshop November 19, 2015 J. Caranana, P. Monsinjon, J. Michelot, C. Bouvier,

White paper. Wide dynamic range. WDR solutions for forensic value. October 2017

White paper Wide dynamic range WDR solutions for forensic value October 2017 Table of contents 1. Summary 4 2. Introduction 5 3. Wide dynamic range scenes 5 4. Physical limitations of a camera s dynamic

HR2000+ Spectrometer. User-Configured for Flexibility. now with. Spectrometers

Spectrometers HR2000+ Spectrometer User-Configured for Flexibility HR2000+ One of our most popular items, the HR2000+ Spectrometer features a high-resolution optical bench, a powerful 2-MHz analog-to-digital

Improved Spectra with a Schmidt-Czerny-Turner Spectrograph

Improved Spectra with a Schmidt-Czerny-Turner Spectrograph Abstract For years spectra have been measured using traditional Czerny-Turner (CT) design dispersive spectrographs. Optical aberrations inherent

Εισαγωγική στην Οπτική Απεικόνιση

Εισαγωγική στην Οπτική Απεικόνιση Δημήτριος Τζεράνης, Ph.D. Εμβιομηχανική και Βιοϊατρική Τεχνολογία Τμήμα Μηχανολόγων Μηχανικών Ε.Μ.Π. Χειμερινό Εξάμηνο 2015 Light: A type of EM Radiation EM radiation:

Confocal Imaging Through Scattering Media with a Volume Holographic Filter

Confocal Imaging Through Scattering Media with a Volume Holographic Filter Michal Balberg +, George Barbastathis*, Sergio Fantini % and David J. Brady University of Illinois at Urbana-Champaign, Urbana,

Optical and mechanical parameters. 100 mm N. of elements 20.5 mm Dimensions 11.7 degrees Weight F/N = 4 (fixed) N.A.

OB SWIR 100 LENS OB-SWIR100/4 P/N C0416 General Description This family of high resolution SWIR lenses image from 0.9 2.3 µmm making them especially well-suited for PCB inspection, special laser applications,

pco.1600 cooled digital 14bit CCD camera system

pco.1600 cooled digital 14bit CCD camera system n excellent resolution (1600 1200 pixel) n 14 bit dynamic range n frame rate of 30 fps at full resolution n image memory in camera (camram up to 4 GB) n

Image Formation: Camera Model

Image Formation: Camera Model Ruigang Yang COMP 684 Fall 2005, CS684-IBMR Outline Camera Models Pinhole Perspective Projection Affine Projection Camera with Lenses Digital Image Formation The Human Eye

Digital Cameras vs Film: the Collapse of Film Photography Can Your Digital Camera reach Film Photography Performance? Film photography started in

Digital Cameras vs Film: the Collapse of Film Photography Can Your Digital Camera reach Film Photography Performance? Film photography started in early 1800 s almost 200 years Commercial Digital Cameras

SNR IMPROVEMENT FOR MONOCHROME DETECTOR USING BINNING

SNR IMPROVEMENT FOR MONOCHROME DETECTOR USING BINNING Dhaval Patel 1, Savitanandan Patidar 2, Pranav Parmar 3 1 PG Student, Electronics and Communication Department, VGEC Chandkheda, Gujarat, India 2 PG

Device design for global shutter operation in a 1.1-um pixel image sensor and its application to nearinfrared

Device design for global shutter operation in a 1.1-um pixel image sensor and its application to nearinfrared sensing Zach M. Beiley Robin Cheung Erin F. Hanelt Emanuele Mandelli Jet Meitzner Jae Park

Selecting an image sensor for the EJSM VIS/NIR camera systems

Selecting an image sensor for the EJSM VIS/NIR camera systems presented by Harald Michaelis (DLR-PF) Folie 1 EJSM- Jan. 18th 2010; ESTEC What for a detector/sensor we shall chose for EJSM? Vortragstitel

Computer Generated Holograms for Optical Testing

Computer Generated Holograms for Optical Testing Dr. Jim Burge Associate Professor Optical Sciences and Astronomy University of Arizona jburge@optics.arizona.edu 520-621-8182 Computer Generated Holograms

Backside illuminated CMOS-TDI line scan sensor for space applications

Backside illuminated CMOS-TDI line scan sensor for space applications Omer COHEN, Oren OFER, Gil ABRAMOVICH, Nimrod BEN-ARI, Gal GERSHON, Maya BRUMER, Adi SHAY, Yaron SHAMAY SemiConductor Devices (SCD)

Beamscope-P8 Wavelength Range. Resolution ¼ - 45 ¼ - 45

Scanning Slit System Beamscope-P8 Typical Applications: Laser / diode laser characterisation Laser assembly development, alignment, characterisation, production test & QA. Lasers and laser assemblies for

X-RAY COMPUTED TOMOGRAPHY

X-RAY COMPUTED TOMOGRAPHY Bc. Jan Kratochvíla Czech Technical University in Prague Faculty of Nuclear Sciences and Physical Engineering Abstract Computed tomography is a powerful tool for imaging the inner

Performance Comparison of Spectrometers Featuring On-Axis and Off-Axis Grating Rotation

Performance Comparison of Spectrometers Featuring On-Axis and Off-Axis Rotation By: Michael Case and Roy Grayzel, Acton Research Corporation Introduction The majority of modern spectrographs and scanning

Camera Image Processing Pipeline: Part II

Lecture 14: Camera Image Processing Pipeline: Part II Visual Computing Systems Today Finish image processing pipeline Auto-focus / auto-exposure Camera processing elements Smart phone processing elements

pco.1300 solar cooled digital 12bit CCD camera system

pco.1300 solar cooled digital 12bit CCD camera system designed for electroluminescence (EL) applications quantum efficiency of up to 13 % @ 880 nm superior low noise of typ. 6 e - rms @ 10 MHz resolution

Double Aperture Camera for High Resolution Measurement

Double Aperture Camera for High Resolution Measurement Venkatesh Bagaria, Nagesh AS and Varun AV* Siemens Corporate Technology, India *e-mail: varun.av@siemens.com Abstract In the domain of machine vision,

Jewelry. Condos. Store. Gas station Parking lot. Nightclub. Surveillance view planes (17) Private areas (3)

Wi4Net White Paper: Outdoor Camera Surveillance Design Outdoor surveillance requires great flexibility to be efficient. Surveillance has to be done over distances from few feet to over 300 feet at all

Horiba LabRAM ARAMIS Raman Spectrometer Revision /28/2016 Page 1 of 11. Horiba Jobin-Yvon LabRAM Aramis - Raman Spectrometer

Page 1 of 11 Horiba Jobin-Yvon LabRAM Aramis - Raman Spectrometer The Aramis Raman system is a software selectable multi-wavelength Raman system with mapping capabilities with a 400mm monochromator and

DESIGNING AND IMPLEMENTING AN ADAPTIVE OPTICS SYSTEM FOR THE UH HOKU KE`A OBSERVATORY ABSTRACT

DESIGNING AND IMPLEMENTING AN ADAPTIVE OPTICS SYSTEM FOR THE UH HOKU KE`A OBSERVATORY University of Hawai`i at Hilo Alex Hedglen ABSTRACT The presented project is to implement a small adaptive optics system

Determining MTF with a Slant Edge Target ABSTRACT AND INTRODUCTION

Determining MTF with a Slant Edge Target Douglas A. Kerr Issue 2 October 13, 2010 ABSTRACT AND INTRODUCTION The modulation transfer function (MTF) of a photographic lens tells us how effectively the lens

Camera Overview. Digital Microscope Cameras for Material Science: Clear Images, Precise Analysis. Digital Cameras for Microscopy

Digital Cameras for Microscopy Camera Overview For Materials Science Microscopes Digital Microscope Cameras for Material Science: Clear Images, Precise Analysis Passionate about Imaging: Olympus Digital

A tutorial for designing. fundamental imaging systems

A tutorial for designing fundamental imaging systems OPTI 521 College of Optical Science University of Arizona November 2009 Abstract This tutorial shows what to do when we design opto-mechanical system

The Science Seeing of process Digital Media. The Science of Digital Media Introduction

The Human Science eye of and Digital Displays Media Human Visual System Eye Perception of colour types terminology Human Visual System Eye Brains Camera and HVS HVS and displays Introduction 2 The Science

Optoliner NV. Calibration Standard for Sighting & Imaging Devices West San Bernardino Road West Covina, California 91790

Calibration Standard for Sighting & Imaging Devices 2223 West San Bernardino Road West Covina, California 91790 Phone: (626) 962-5181 Fax: (626) 962-5188 www.davidsonoptronics.com sales@davidsonoptronics.com

Residual Bulk Image Characterization using Photon Transfer Techniques

https://doi.org/10.2352/issn.2470-1173.2017.11.imse-189 2017, Society for Imaging Science and Technology Residual Bulk Image Characterization using Photon Transfer Techniques Richard Crisp Etron Technology

PROCEEDINGS OF SPIE. Automated asphere centration testing with AspheroCheck UP

PROCEEDINGS OF SPIE SPIEDigitalLibrary.org/conference-proceedings-of-spie Automated asphere centration testing with AspheroCheck UP F. Hahne, P. Langehanenberg F. Hahne, P. Langehanenberg, "Automated asphere

Hyperspectral Sensor

Hyperspectral Sensor Detlev Even 733 Bishop Street, Suite 2800 Honolulu, HI 96813 phone: (808) 441-3610 fax: (808) 441-3601 email: detlev@nova-sol.com Arleen Velasco 15150 Avenue of Science San Diego,

Basic principles of photography. David Capel 346B IST

Basic principles of photography David Capel 346B IST Latin Camera Obscura = Dark Room Light passing through a small hole produces an inverted image on the opposite wall Safely observing the solar eclipse

Sony PXW-FS7 Guide. October 2016 v4

Sony PXW-FS7 Guide 1 Contents Page 3 Layout and Buttons (Left) Page 4 Layout back and lens Page 5 Layout and Buttons (Viewfinder, grip remote control and eye piece) Page 6 Attaching the Eye Piece Page

PROCEEDINGS OF SPIE. Measurement of the modulation transfer function (MTF) of a camera lens

PROCEEDINGS OF SPIE SPIEDigitalLibrary.org/conference-proceedings-of-spie Measurement of the modulation transfer function (MTF) of a camera lens Aline Vernier, Baptiste Perrin, Thierry Avignon, Jean Augereau,

DOING PHYSICS WITH MATLAB COMPUTATIONAL OPTICS. GUI Simulation Diffraction: Focused Beams and Resolution for a lens system

DOING PHYSICS WITH MATLAB COMPUTATIONAL OPTICS GUI Simulation Diffraction: Focused Beams and Resolution for a lens system Ian Cooper School of Physics University of Sydney ian.cooper@sydney.edu.au DOWNLOAD

EBU - Tech 3335 : Methods of measuring the imaging performance of television cameras for the purposes of characterisation and setting

EBU - Tech 3335 : Methods of measuring the imaging performance of television cameras for the purposes of characterisation and setting Alan Roberts, May 18 2012 SUPPLEMENT 002: Assessment of a Nikon D4

A Light Source for Testing CMOS Imagers

A Light Source for Testing CMOS Imagers Jed 1. Hancock Department of Electrical and Computer Engineering Utah State University Logan, Utah 84322 Abstract Testing the optical properties of complementaiy