Pixel Response Effects on CCD Camera Gain Calibration

Save this PDF as:

Size: px
Start display at page:

Transcription

2 2 of 7 1/21/2014 3:03 PM straightforward way Understanding the mathematics validates the gain calculation technique described in the next section, and it shows why simpler techniques fail to give the correct answer This derivation uses the concepts of "signal" and "noise" CCD performance is usually described in terms of signal to noise ratio, or "S/N", but we shall deal with them separately here The signal is defined as the quantity of information you measure in the image in other words, the signal is the number of electrons recorded by the CCD or the number of counts present in the CCD image The noise is the uncertainty in the signal Since the photons recorded by the CCD arrive in random packets (courtesy of nature), observing the same source many times records a different number of electrons every time This variation is a random error, or "noise" that is added to the true signal You measure the gain of the CCD by comparing the signal level to the amount of variation in the signal This works because the relationship between counts and electrons is different for the signal and the variance There are two ways to make this measurement: Measure the signal and variation within the same region of pixels at many intensity levels Measure the signal and variation in a single pixel at many intensity levels Both of these methods are detailed in section 6 They have the same mathematical foundation To derive the relationship between signal and variance in a CCD image, let us define the following quantities: S C The signal measured in count units in the CCD image S E The signal recorded in electron units by the CCD chip This quantity is unknown N C The total noise measured in count units in the CCD image N E The total noise in terms of recorded electrons This quantity is unknown g The gain, in units of electrons per count This will be calculated R E The readout noise of the CCD, in [electrons] This quantity is unknown s E The photon noise in the signal N E s o An additional noise source in the image This is described below We need an equation to relate the number of electrons, which is unknown, to quantities we measure in the CCD image in units of counts The signals and noises are simply related through the gain factor as and These can be inverted to give and The noise is contributed by various sources We consider these to be readout noise, R E, photon noise attributable to the nature of light,, and some additional noise,, which will be shown to be important in in the following Section

3 3 of 7 1/21/2014 3:03 PM Remembering that the different noise sources are independent of each other, they add in quadrature This means that they add as the square their noise values If we could measure the total noise in units of electrons, the various noise sources would combine in the following way: The random arrival rate of photons controls the photon noise, which makes the square of the noise equal to the signal, or Photon noise obeys the laws of Poissonian statistics, Therefore, we can make the following substitution: Knowing how the gain relates units of electrons and counts, we can modify this equation to read as follows: which then gives We can rearrange this to get the final equation: This is the equation of a line in which is the y axis, is the x axis, and the slope is 1/g The extra terms are grouped together for the time being Below, they will be separated, as the extra noise term has a profound effect on the method we use to measure gain A better way to apply this equation is to plot our measurements with as the y axis and as the x axis, as this gives the gain directly as the slope Theoretically, at least, one could also calculate the readout noise,, from the point where the line hits the y axis at = 0 Knowing the gain then allows this to be converted to a Readout Noise in the standard units of electrons However, finding the intercept of the line is not a good method, because the readout noise is a relatively small quantity and the exact path where the line passes through the y axis is subject to much uncertainty With the mathematics in place, we are now ready to calculate the gain So far, I have ignored the "extra noise term", In the next 2 sections, I will describe the nature of the extra noise term and show how it affects the way we measure the gain of a CCD camera Crude Estimation of the Gain 4 Obtain images at different signal levels and subtract the bias from them This is necessary because the bias level adds to the measured signal but does not contribute noise Measure the signal and noise in each image The mean and standard deviation of a region of pixels give these quantities Square the noise value to get a variance at each signal level For each image, plot Signal on the y axis against Variance on the x axis Find the slope of a line through the points The gain equals the slope Is measuring the gain actually this simple? Well, yes and no If we actually make the measurement over a substantial range of signal, the data points will follow a curve rather than a line Using the present method we will always measure a slope that is too shallow, and with it we will always underestimate the gain Using only low signal levels, this method can give a gain value that is at least "in the ballpark" of the true value At low signal levels, the curvature is not apparent, though present However, the data points have some amount of scatter themselves, and without a long baseline of signal, the slope might not be well determined The curvature in the Signal - Variance plot is caused by the extra noise term which this simple method neglects

4 CD Camera Gain Measurement of 7 1/21/2014 3:03 PM The following factors affect the amount of curvature we obtain: The color of the light source Blue light is worse because CCD s show the greatest surface irregularity at shorter wavelengths These irregularities are described in Section 5 The fabrication technology of the CCD chip These issues determine the relative strength of the effects described in item The uniformity of illumination on the CCD chipif the Illumination is not uniform, then the sloping count level inside the pixel region used to measure it inflates the measured standard deviation Fortunately, we can obtain the proper value by doing just a bit more work We need to change the experiment in a way that makes the data plot as a straight line We have to devise a way to account for the extra noise term, If were a constant value we could combine it with the constant readout noise We have not talked in detail about readout noise, but we have assumed that it merges together all constant noise sources that do not change with the signal level The Extra Noise Term in the Signal-Variance Relationship The mysterious extra noise term,, is attributable to pixel-to-pixel variations in the sensitivity of the CCD, known as the flat field effect The flat field effect produces a pattern of apparently "random" scatter in a CCD image Even an exposure with infinite signal to noise ratio ("S/N") shows the flat field pattern Despite its appearance, the pattern is not actually random because it repeats from one image to another Changing the color of the light source changes the details of the pattern, but the pattern remains the same for all images exposed to light of the same spectral makeup The importance of this effect is that, although the flat field variation is not a true noise, unless it is removed from the image it contributes to the noise you actually measure We need to characterize the noise contributed by the flat field pattern in order to determine its effect on the variance we measure in the image This turns out to be quite simple: Since the flat field pattern is a fixed percentage of the signal, the standard deviation, or "noise" you measure from it is always proportional to the signal For example, a pixel might be 1% less sensitive than its left neighbor, but 3% less sensitive than its right neighbor Therefore, exposing this pixel at the 100 count level produces the following 3 signals: 101, 100, 10 However, exposing at the 10,000 count level gives these results: 10,100, 10,000, 10,300 The standard deviation for these 3 pixels is counts for the low signal case but is counts for the high signal case Thus the standard deviation is 100 times larger when the signal is also 100 times larger We can express this proportionality between the flat field "noise" and the signal level in a simple mathematical way: In the present example, we have k=00233 Substituting this expression for the flat field variation into our master equation, we get the following result: With a simple rearrangement of the terms, this reveals a nice quadratic function of signal: When plotted with the Signal on the x axis, this equation describes a parabola that opens upward Since the Signal - Variance plot is actually plotted with Signal on the y axis, we need to invert this equation to solve for S C : This final equation describes the classic Signal - Variance plot In this form, the equation describes a family of horizontal parabolas that open toward the right The strength of the flat field variation, k, determines the curvature When k = 0, the curvature goes away and it gives the straight line relationship we desire The curvature to the right of the line means that the stronger the flat field pattern, the more the variance is inflated at a given signal level This result shows that it is impossible to accurately determine the gain from a Signal - Variance plot unless we know one of two things: Either 1) we know the value of k, or 2) we setup our measurements to avoid flat field effects Option 2 is the correct strategy Essentially, the weakness of the method described in Section 4 is that it assumes that a straight line relationship exists but ignores flat field effects

5 5 of 7 1/21/2014 3:03 PM To illustrate the effect of flat field variations, mathematical models were constructed using the equation above with parameters typical of commonly available CCD cameras These include readout noise R = 15e- and gain g = 0 e- / E Count Three models were constructed with flat field parameters k = 0, k = 0005, and k = 00 Flat field variations of this order are not uncommon These models are shown in the figure below Increasing values of k correspond to progressively larger flat field irregularities in the CCD chip The amplitude of flat field effects, k, tends to increase with shorter wavelength, particularly with thinned CCD's (this is why Section 4 recommends using a redder light source to illuminate the CCD) The flat field pattern is present in every image exposed to light Clearly, it can be seen from the models that if one simply obtains images at different signal levels and measures the variance in them, then fitting a line through any part of the curve yields a slope lower than its true value Thus the simple method of section 4 always underestimates the gain The best strategy for doing the Signal - Variance method is to find a way to produce a straight line by properly compensating for flat field effects This is important by the "virtue of straightness": Deviation from a straight line is completely unambiguous and easy to detect It avoids the issue of how much curvature is attributable to what cause The electronic design of a CCD camera is quite complex, and problems can occur, such as gain change at different signal levels or unexplained extra noise at high or low signal levels Using a "robust" method for calculating gain, any significant deviation from a line is a diagnostic of possible problems in the camera electronics Two such methods are described in the following section In previous sections, the so-called simple method of estimating the gain was shown to be an oversimplification Specifically, it produces a Signal - Variance plot with a curved relationship resulting from flat field effects This section presents two robust methods that correct the flat field effects in the Signal - Variance relationship to yield the desired straight-line relationship This permits an accurate gain value to be calculated Adjusting the method to remove flat field effects is a better strategy than either to attempt to use a low signal level where flat field effects are believed not to be important or to attempt to measure and compensate for the flat field parameter k When applying the robust methods described below, one must consider some procedural issues that apply to both: Both methods measure sets of 2 or more images at each signal level An image set is defined as 2 or more successive images taken under the same illumination conditions To obtain various signal levels, it is better to change the intensity received by the CCD than to change the exposure time This may be achieved either by varying the light source intensity or by altering the amount of light passing into the camera The illumination received by the CCD should not vary too much within a set of images, but it does not have to be truly constant Cool the CCD camera to reduce the dark current to as low as possible This prevents you from having to subtract dark frames from the images (doing so adds noise, which adversely affects the noise measurements at low signal level) In addition, if the bias varies from one frame to another, be sure to subtract a bias value from every image The CCD should be illuminated the same way for all images within a set Irregularities in illumination within a set are automatically removed by the image processing methods used in the calibration It does not matter if the illumination pattern changes when you change the intensity level for a different image set

6 6 of 7 1/21/2014 3:03 PM Within an image set, variation in the light intensity is corrected by normalizing the images so that they have the same average signal within the same pixel region The process of normalizing multiplies the image by an appropriate constant value so that its mean value within the pixel region matches that of other images in the same set Multiplying by a constant value does not affect the signal to noise ratio or the flat field structure of the image Do not estimate the CCD camera's readout noise by calculating the noise value at zero signal This is the square root of the variance where the gain line intercepts the y axis Especially do not use this value if bias is not subtracted from every frame To calculate the readout noise, use the "Two Bias" method and apply the gain value determined from this test In the Two Bias Method, 2 bias frames are taken in succession and then subtracted from each other Measure the standard deviation inside a region of, say 100x100 pixels and divide by 414 This gives the readout noise in units of counts Multiply this by the gain factor to get the Readout Noise in units of electrons If bias frames are not available, cool the camera and obtain two dark frames of minimum exposure, then apply the Two Bias Method to them Method 1: Correct the flat field effects at each signal level In this strategy, the flat field effects are removed by subtracting one image from another at each signal level Here is the recipe: For each intensity level, do the following: Obtain 2 images in succession at the same light level Call these images A and B Subtract the bias level from both images Keep the exposure short so that the dark current is negligibly small If the dark current is large, you should also remove it from both frames Measure the mean signal level S in a region of pixels on images A and B Call these mean signals S and S It is A B best if the bounds of the region change as little as possible from one image to the next The region might be as small as 50x50 to 100x100 pixels but should not contain obvious defects such as cosmic ray hits, dead pixels, etc 4 Calculate the ratio of the mean signal levels as r = S A / S B Multiply image B by the number r This corrects image B to the same signal level as image A without affecting its noise structure or flat field variation Subtract image B from image A The flat field effects present in both images should be cancelled to within the random errors Measure the standard deviation over the same pixel region you used in step Square this number to get the Variance In addition, divide the resulting variance by 0 to correct for the fact that the variance is doubled when you subtract one similar image from another Use the Signal from step 3 and the Variance from step 7 to add a data point to your Signal - Variance plot Change the light intensity and repeat steps 1 through 8 Method 2: Avoid flat field effects using one pixel in many images This strategy avoids the flat field variation by considering how a single pixel varies among many images Since the variance is calculated from a single pixel many times, rather than from a collection of different pixels, there is no flat field variation To calculate the variance at a given signal level, you obtain many frames, measure the same pixel in each frame, and calculate the variance among this set of values One problem with this method is that the variance itself is subject to random errors and is only an estimate of the true value To obtain a reliable variance, you must use 100 s of images at each intensity level This is completely analogous to measuring the variance over a moderate sized pixel region in Method A; in both methods, using many pixels to compute the variance gives a more statistically sound value Another limitation of this method is that it either requires a perfectly stable light source or you have to compensate for light source variation by adjusting each image to the same average signal level before measuring its pixel Altogether, the method requires a large number of images and a lot of processing For this reason, Method A is preferred In any case, here is the recipe:

7 7 of 7 1/21/2014 3:03 PM Select a pixel to measure at the same location in every image Always measure the same pixel in every image at every signal level For each intensity level, do the following: 4 Obtain at least 100 images in succession at the same light level Call the first image A and the remaining images i Since you are interested in a single pixel, the images may be small, of order 100x100 pixels Subtract the bias level from each image Keep the exposure short so that the dark is negligibly small If the dark current is large, you should also remove it from every frame Measure the mean signal level S in a rectangular region of pixels on image A Measure the same quantity in each of the remaining images The measuring region might be as small as 50x50 to 100x100 pixels and should be centered on the brightest part of the image For each image S other than the first, calculate the ratio of its mean signal level to that of image A This gives a i number for each image, r i = S A / S i 5 Multiply each image i by the number r i This corrects each image to the same average intensity as image A Measure the number of counts in the selected pixel in every one of the images From these numbers, compute a mean count and standard deviation Square the standard deviation to get the variance Use the Signal and Variance from step 6 to add a data point to your Signal - Variance plot Change the light intensity and repeat steps 1 through 7 Summary We have derived the mathematical relationship between Signal and Variance in a CCD image which includes the pixelto-pixel response variations among the image pixels This "flat field" effect must be compensated or the calculated value of the camera gain will be incorrect We have shown how the traditional "simple" method used for gain calculation leads to an erroneous gain value unless flat field effects are not considered We have suggested 2 methods that correctly account for the flat field effect, and these should be implemented in camera testing procedures HO M E P R O D UC T S B R IE F S T E C H NO T E S S UP P O RT P UR C HA S E NE W S W E B T O O L S INF O C O NTA C T Copyright 2012, Mirametrics, Inc All rights reserved

CCD Characteristics Lab

CCD Characteristics Lab Observational Astronomy 6/6/07 1 Introduction In this laboratory exercise, you will be using the Hirsch Observatory s CCD camera, a Santa Barbara Instruments Group (SBIG) ST-8E.

UV/Optical/IR Astronomy Part 2: Spectroscopy

UV/Optical/IR Astronomy Part 2: Spectroscopy Introduction We now turn to spectroscopy. Much of what you need to know about this is the same as for imaging I ll concentrate on the differences. Slicing the

Temperature Dependent Dark Reference Files: Linear Dark and Amplifier Glow Components

Instrument Science Report NICMOS 2009-002 Temperature Dependent Dark Reference Files: Linear Dark and Amplifier Glow Components Tomas Dahlen, Elizabeth Barker, Eddie Bergeron, Denise Smith July 01, 2009

Technical Note CMOS, EMCCD AND CCD CAMERAS FOR LIFE SCIENCES Camera Test Protocol Introduction The detector is one of the most important components of any microscope system. Accurate detector readings

when it s too fast to see, and too important not to. NOTES/ALERTS For the most current version visit www.phantomhighspeed.com Subject to change Rev April 2016 Boosting Sensitivity In this series of articles,

GAMMA-GAMMA CORRELATION Latest Revision: August 21, 2007

C1-1 GAMMA-GAMMA CORRELATION Latest Revision: August 21, 2007 QUESTION TO BE INVESTIGATED: decay event? What is the angular correlation between two gamma rays emitted by a single INTRODUCTION & THEORY:

Physics Laboratory Scattering of Photons from Electrons: Compton Scattering

RR Oct 2001 SS Dec 2001 MJ Oct 2009 Physics 34000 Laboratory Scattering of Photons from Electrons: Compton Scattering Objective: To measure the energy of high energy photons scattered from electrons in

DC Bias. Graphical Analysis. Script

Course: B.Sc. Applied Physical Science (Computer Science) Year & Sem.: Ist Year, Sem - IInd Subject: Electronics Paper No.: V Paper Title: Analog Circuits Lecture No.: 3 Lecture Title: Analog Circuits

INTRODUCTION TO CCD IMAGING

ASTR 1030 Astronomy Lab 85 Intro to CCD Imaging INTRODUCTION TO CCD IMAGING SYNOPSIS: In this lab we will learn about some of the advantages of CCD cameras for use in astronomy and how to process an image.

REAL-TIME X-RAY IMAGE PROCESSING; TECHNIQUES FOR SENSITIVITY

REAL-TIME X-RAY IMAGE PROCESSING; TECHNIQUES FOR SENSITIVITY IMPROVEMENT USING LOW-COST EQUIPMENT R.M. Wallingford and J.N. Gray Center for Aviation Systems Reliability Iowa State University Ames,IA 50011

Experiment 2. Ohm s Law. Become familiar with the use of a digital voltmeter and a digital ammeter to measure DC voltage and current.

Experiment 2 Ohm s Law 2.1 Objectives Become familiar with the use of a digital voltmeter and a digital ammeter to measure DC voltage and current. Construct a circuit using resistors, wires and a breadboard

Computation of dark frames in digital imagers Ralf Widenhorn, a,b Armin Rest, c Morley M. Blouke, d Richard L. Berry, b and Erik Bodegom a,b

Computation of dark frames in digital imagers Ralf Widenhorn, a,b Armin Rest, c Morley M. Blouke, d Richard L. Berry, b and Erik Bodegom a,b a Portland State, Portland, OR 97207, b Digital Clarity Consultants,

DETERMINING CALIBRATION PARAMETERS FOR A HARTMANN- SHACK WAVEFRONT SENSOR

DETERMINING CALIBRATION PARAMETERS FOR A HARTMANN- SHACK WAVEFRONT SENSOR Felipe Tayer Amaral¹, Luciana P. Salles 2 and Davies William de Lima Monteiro 3,2 Graduate Program in Electrical Engineering -

GAIN COMPARISON MEASUREMENTS IN SPHERICAL NEAR-FIELD SCANNING

GAIN COMPARISON MEASUREMENTS IN SPHERICAL NEAR-FIELD SCANNING ABSTRACT by Doren W. Hess and John R. Jones Scientific-Atlanta, Inc. A set of near-field measurements has been performed by combining the methods

Presented by Jerry Hubbell Lake of the Woods Observatory (MPC I24) President, Rappahannock Astronomy Club

Presented by Jerry Hubbell Lake of the Woods Observatory (MPC I24) President, Rappahannock Astronomy Club ENGINEERING A FIBER-FED FED SPECTROMETER FOR ASTRONOMICAL USE Objectives Discuss the engineering

A Study of Slanted-Edge MTF Stability and Repeatability

A Study of Slanted-Edge MTF Stability and Repeatability Jackson K.M. Roland Imatest LLC, 2995 Wilderness Place Suite 103, Boulder, CO, USA ABSTRACT The slanted-edge method of measuring the spatial frequency

Section 4. Ohm s Law: Putting up a Resistance. What Do You See? What Do You Think? Investigate

Section 4 Ohm s Law: Putting up a Resistance Florida Next Generation Sunshine State Standards: Additional Benchmarks met in Section 4 SC.912.N.2.4 Explain that scientific knowledge is both durable and

Satellite TVRO G/T calculations

Satellite TVRO G/T calculations From: http://aa.1asphost.com/tonyart/tonyt/applets/tvro/tvro.html Introduction In order to understand the G/T calculations, we must start with some basics. A good starting

Homework Set 3.5 Sensitive optoelectronic detectors: seeing single photons

Homework Set 3.5 Sensitive optoelectronic detectors: seeing single photons Due by 12:00 noon (in class) on Tuesday, Nov. 7, 2006. This is another hybrid lab/homework; please see Section 3.4 for what you

Low Light Level CCD Performance and Issues

Low Light Level CCD Performance and Issues Nagaraja Bezawada UK Astronomy Technology Centre 04 July 2007 Overview of the Talk Introduction to L3CCD (EM CCD) ULTRASPEC Performance and Issues New L3 CCD

CHARGE-COUPLED DEVICE (CCD)

CHARGE-COUPLED DEVICE (CCD) Definition A charge-coupled device (CCD) is an analog shift register, enabling analog signals, usually light, manipulation - for example, conversion into a digital value that

Specify Gain and Phase Margins on All Your Loops

Keywords Venable, frequency response analyzer, power supply, gain and phase margins, feedback loop, open-loop gain, output capacitance, stability margins, oscillator, power electronics circuits, voltmeter,

FRAUNHOFER AND FRESNEL DIFFRACTION IN ONE DIMENSION

FRAUNHOFER AND FRESNEL DIFFRACTION IN ONE DIMENSION Revised November 15, 2017 INTRODUCTION The simplest and most commonly described examples of diffraction and interference from two-dimensional apertures

E. Slope-Intercept Form and Direct Variation (pp )

and Direct Variation (pp. 32 35) For any two points, there is one and only one line that contains both points. This fact can help you graph a linear equation. Many times, it will be convenient to use the

Results of FE65-P2 Pixel Readout Test Chip for High Luminosity LHC Upgrades

for High Luminosity LHC Upgrades R. Carney, K. Dunne, *, D. Gnani, T. Heim, V. Wallangen Lawrence Berkeley National Lab., Berkeley, USA e-mail: mgarcia-sciveres@lbl.gov A. Mekkaoui Fermilab, Batavia, USA

Luminescent Background Sources and Corrections

Concept Tech Note 1 Luminescent Background Sources and Corrections The background sources of light from luminescent images are inherently very low. This appendix discusses sources of background and how

CCD1600A Full Frame CCD Image Sensor x Element Image Area

- 1 - General Description CCD1600A Full Frame CCD Image Sensor 10560 x 10560 Element Image Area General Description The CCD1600 is a 10560 x 10560 image element solid state Charge Coupled Device (CCD)

Diffraction. Interference with more than 2 beams. Diffraction gratings. Diffraction by an aperture. Diffraction of a laser beam

Diffraction Interference with more than 2 beams 3, 4, 5 beams Large number of beams Diffraction gratings Equation Uses Diffraction by an aperture Huygen s principle again, Fresnel zones, Arago s spot Qualitative

DC and AC Circuits. Objective. Theory. 1. Direct Current (DC) R-C Circuit

[International Campus Lab] Objective Determine the behavior of resistors, capacitors, and inductors in DC and AC circuits. Theory ----------------------------- Reference -------------------------- Young

Image Formation. Light from distant things. Geometrical optics. Pinhole camera. Chapter 36

Light from distant things Chapter 36 We learn about a distant thing from the light it generates or redirects. The lenses in our eyes create images of objects our brains can process. This chapter concerns

I = I 0 cos 2 θ (1.1)

Chapter 1 Faraday Rotation Experiment objectives: Observe the Faraday Effect, the rotation of a light wave s polarization vector in a material with a magnetic field directed along the wave s direction.

Ohm s Law and Electrical Circuits

Ohm s Law and Electrical Circuits INTRODUCTION In this experiment, you will measure the current-voltage characteristics of a resistor and check to see if the resistor satisfies Ohm s law. In the process

2013 LMIC Imaging Workshop. Sidney L. Shaw Technical Director. - Light and the Image - Detectors - Signal and Noise

2013 LMIC Imaging Workshop Sidney L. Shaw Technical Director - Light and the Image - Detectors - Signal and Noise The Anatomy of a Digital Image Representative Intensities Specimen: (molecular distribution)

Seismic Reflection Method

1 of 25 4/16/2009 11:41 AM Seismic Reflection Method Top: Monument unveiled in 1971 at Belle Isle (Oklahoma City) on 50th anniversary of first seismic reflection survey by J. C. Karcher. Middle: Two early

Image analysis. CS/CME/BIOPHYS/BMI 279 Fall 2015 Ron Dror

Image analysis CS/CME/BIOPHYS/BMI 279 Fall 2015 Ron Dror A two- dimensional image can be described as a function of two variables f(x,y). For a grayscale image, the value of f(x,y) specifies the brightness

What applications is a cardioid subwoofer configuration appropriate for?

SETTING UP A CARDIOID SUBWOOFER SYSTEM Joan La Roda DAS Audio, Engineering Department. Introduction In general, we say that a speaker, or a group of speakers, radiates with a cardioid pattern when it radiates

SPEAR BTS Toroid Calibration

SPEAR BTS Toroid Calibration J. Sebek April 3, 2012 Abstract The Booster to SPEAR (BTS) transport line contains several toroids used for measuring the charge that is injected into SPEAR. One of these toroids

REPORT ITU-R SA.2098

Rep. ITU-R SA.2098 1 REPORT ITU-R SA.2098 Mathematical gain models of large-aperture space research service earth station antennas for compatibility analysis involving a large number of distributed interference

Spectral and Polarization Configuration Guide for MS Series 3-CCD Cameras

Spectral and Polarization Configuration Guide for MS Series 3-CCD Cameras Geospatial Systems, Inc (GSI) MS 3100/4100 Series 3-CCD cameras utilize a color-separating prism to split broadband light entering

More on the Mask Error Enhancement Factor

T h e L i t h o g r a p h y E x p e r t (Fall 1999) More on the Mask Error Enhancement Factor Chris A. Mack, FINLE Technologies, Austin, Texas In a previous edition of this column (Winter, 1999) I described

Smith Chart Calculations

The following material was extracted from earlier editions. Figure and Equation sequence references are from the 21st edition of The ARRL Antenna Book Smith Chart Calculations The Smith Chart is a sophisticated

The Design and Construction of an Inexpensive CCD Camera for Astronomical Imaging

The Design and Construction of an Inexpensive CCD Camera for Astronomical Imaging Mr. Ben Teasdel III South Carolina State University Abstract The design, construction and testing results of an inexpensive

EXPERIMENTAL ERROR AND DATA ANALYSIS

EXPERIMENTAL ERROR AND DATA ANALYSIS 1. INTRODUCTION: Laboratory experiments involve taking measurements of physical quantities. No measurement of any physical quantity is ever perfectly accurate, except

T.J.Moir AUT University Auckland. The Ph ase Lock ed Loop.

T.J.Moir AUT University Auckland The Ph ase Lock ed Loop. 1.Introduction The Phase-Locked Loop (PLL) is one of the most commonly used integrated circuits (ICs) in use in modern communications systems.

Photon Count. for Brainies.

Page 1/12 Photon Count ounting for Brainies. 0. Preamble This document gives a general overview on InGaAs/InP, APD-based photon counting at telecom wavelengths. In common language, telecom wavelengths

System Identification and CDMA Communication

System Identification and CDMA Communication A (partial) sample report by Nathan A. Goodman Abstract This (sample) report describes theory and simulations associated with a class project on system identification

CCD User s Guide SBIG ST7E CCD camera and Macintosh ibook control computer with Meade flip mirror assembly mounted on LX200

Massachusetts Institute of Technology Department of Earth, Atmospheric, and Planetary Sciences Handout 8 /week of 2002 March 18 12.409 Hands-On Astronomy, Spring 2002 CCD User s Guide SBIG ST7E CCD camera

IE 361 Module 17. Process Capability Analysis: Part 1. Reading: Sections 5.1, 5.2 Statistical Quality Assurance Methods for Engineers

IE 361 Module 17 Process Capability Analysis: Part 1 Reading: Sections 5.1, 5.2 Statistical Quality Assurance Methods for Engineers Prof. Steve Vardeman and Prof. Max Morris Iowa State University Vardeman

FIBER OPTICS. Prof. R.K. Shevgaonkar. Department of Electrical Engineering. Indian Institute of Technology, Bombay. Lecture: 35. Self-Phase-Modulation

FIBER OPTICS Prof. R.K. Shevgaonkar Department of Electrical Engineering Indian Institute of Technology, Bombay Lecture: 35 Self-Phase-Modulation (SPM) Fiber Optics, Prof. R.K. Shevgaonkar, Dept. of Electrical

Graphs. This tutorial will cover the curves of graphs that you are likely to encounter in physics and chemistry.

Graphs Graphs are made by graphing one variable which is allowed to change value and a second variable that changes in response to the first. The variable that is allowed to change is called the independent

Op-Amp Simulation Part II

Op-Amp Simulation Part II EE/CS 5720/6720 This assignment continues the simulation and characterization of a simple operational amplifier. Turn in a copy of this assignment with answers in the appropriate

Proportional-Integral Controller Performance

Proportional-Integral Controller Performance Silver Team Jonathan Briere ENGR 329 Dr. Henry 4/1/21 Silver Team Members: Jordan Buecker Jonathan Briere John Colvin 1. Introduction Modeling for the response

The 0.84 m Telescope OAN/SPM - BC, Mexico

The 0.84 m Telescope OAN/SPM - BC, Mexico Readout error CCD zero-level (bias) ramping CCD bias frame banding Shutter failure Significant dark current Image malting Focus frame taken during twilight IR

Lesson 7 Slope-Intercept Formula

Lesson 7 Slope-Intercept Formula Terms Two new words that describe what we've been doing in graphing lines are slope and intercept. The slope is referred to as "m" (a mountain has slope and starts with

Photo Scale The photo scale and representative fraction may be calculated as follows: PS = f / H Variables: PS - Photo Scale, f - camera focal

Scale Scale is the ratio of a distance on an aerial photograph to that same distance on the ground in the real world. It can be expressed in unit equivalents like 1 inch = 1,000 feet (or 12,000 inches)

Local Oscillator Phase Noise and its effect on Receiver Performance C. John Grebenkemper

Watkins-Johnson Company Tech-notes Copyright 1981 Watkins-Johnson Company Vol. 8 No. 6 November/December 1981 Local Oscillator Phase Noise and its effect on Receiver Performance C. John Grebenkemper All

Image acquisition. In both cases, the digital sensing element is one of the following: Line array Area array. Single sensor

Image acquisition Digital images are acquired by direct digital acquisition (digital still/video cameras), or scanning material acquired as analog signals (slides, photographs, etc.). In both cases, the

Products of Linear Functions

Math Objectives Students will understand relationships between the horizontal intercepts of two linear functions and the horizontal intercepts of the quadratic function resulting from their product. Students

Determining Dimensional Capabilities From Short-Run Sample Casting Inspection

Determining Dimensional Capabilities From Short-Run Sample Casting Inspection A.A. Karve M.J. Chandra R.C. Voigt Pennsylvania State University University Park, Pennsylvania ABSTRACT A method for determining

Determination of the STIS CCD Gain

Instrument Science Report STIS 2016-01(v1) Determination of the STIS CCD Gain Allyssa Riley 1, TalaWanda Monroe 1, Sean Lockwood 1 1 Space Telescope Science Institute, Baltimore, MD 29 September 2016 ABSTRACT

Dark current behavior in DSLR cameras

Dark current behavior in DSLR cameras Justin C. Dunlap, Oleg Sostin, Ralf Widenhorn, and Erik Bodegom Portland State, Portland, OR 9727 ABSTRACT Digital single-lens reflex (DSLR) cameras are examined and

White paper. Wide dynamic range. WDR solutions for forensic value. October 2017

White paper Wide dynamic range WDR solutions for forensic value October 2017 Table of contents 1. Summary 4 2. Introduction 5 3. Wide dynamic range scenes 5 4. Physical limitations of a camera s dynamic

Controlling Lady Beetles at the Martin Observatory

Controlling Lady Beetles at the Martin Observatory John H. Simonetti, Jane Doe, and Joe Blow Department of Physics, Virginia Tech, Blacksburg VA 24061 January 16, 2012 Abstract We studied the nesting habits

CHEM*3440 Instrumental Analysis Mid-Term Examination Fall Duration: 2 hours

CHEM*344 Instrumental Analysis Mid-Term Examination Fall 4 Duration: hours. ( points) An atomic absorption experiment found the following results for a series of standard solutions for dissolved palladium

The Common Emitter Amplifier Circuit

The Common Emitter Amplifier Circuit In the Bipolar Transistor tutorial, we saw that the most common circuit configuration for an NPN transistor is that of the Common Emitter Amplifier circuit and that

Camera Calibration Certificate No: DMC II

Calibration DMC II 230 015 Camera Calibration Certificate No: DMC II 230 015 For Air Photographics, Inc. 2115 Kelly Island Road MARTINSBURG WV 25405 USA Calib_DMCII230-015_2014.docx Document Version 3.0

An SWR-Feedline-Reactance Primer Part 1. Dipole Samples

An SWR-Feedline-Reactance Primer Part 1. Dipole Samples L. B. Cebik, W4RNL Introduction: The Dipole, SWR, and Reactance Let's take a look at a very common antenna: a 67' AWG #12 copper wire dipole for

Getting the Best Performance from Challenging Control Loops

Getting the Best Performance from Challenging Control Loops Jacques F. Smuts - OptiControls Inc, League City, Texas; jsmuts@opticontrols.com KEYWORDS PID Controls, Oscillations, Disturbances, Tuning, Stiction,

Using Curves and Histograms

Written by Jonathan Sachs Copyright 1996-2003 Digital Light & Color Introduction Although many of the operations, tools, and terms used in digital image manipulation have direct equivalents in conventional

Camera Calibration Certificate No: DMC II

Calibration DMC II 230 027 Camera Calibration Certificate No: DMC II 230 027 For Peregrine Aerial Surveys, Inc. 103-20200 56 th Ave Langley, BC V3A 8S1 Canada Calib_DMCII230-027.docx Document Version 3.0

APPENDIX D: ANALYZING ASTRONOMICAL IMAGES WITH MAXIM DL

APPENDIX D: ANALYZING ASTRONOMICAL IMAGES WITH MAXIM DL Written by T.Jaeger INTRODUCTION Early astronomers relied on handmade sketches to record their observations (see Galileo s sketches of Jupiter s

Camera Calibration Certificate No: DMC II Aero Photo Europe Investigation

Calibration DMC II 250 030 Camera Calibration Certificate No: DMC II 250 030 For Aero Photo Europe Investigation Aerodrome de Moulins Montbeugny Yzeure Cedex 03401 France Calib_DMCII250-030.docx Document

Observation Data. Optical Images

Data Analysis Introduction Optical Imaging Tsuyoshi Terai Subaru Telescope Imaging Observation Measure the light from celestial objects and understand their physics Take images of objects with a specific

Technical Notes. Integrating Sphere Measurement Part II: Calibration. Introduction. Calibration

Technical Notes Integrating Sphere Measurement Part II: Calibration This Technical Note is Part II in a three part series examining the proper maintenance and use of integrating sphere light measurement

Engineering & Design: Geometric Dimensioning

Section Contents NADCA No. Format Page Frequently Asked Questions -2 s e c t i o n 1 Introduction -2 2 What is GD&T? -2 3 Why Should GD&T be Used? -2 4 Datum Reference Frame -4 4.1 Primary, Secondary,

Camera Calibration Certificate No: DMC IIe

Calibration DMC IIe 230 23522 Camera Calibration Certificate No: DMC IIe 230 23522 For Richard Crouse & Associates 467 Aviation Way Frederick, MD 21701 USA Calib_DMCIIe230-23522.docx Document Version 3.0

Camera Calibration Certificate No: DMC II

Calibration DMC II 230 020 Camera Calibration Certificate No: DMC II 230 020 For MGGP Aero Sp. z o.o. ul. Słowackiego 33-37 33-100 Tarnów Poland Calib_DMCII230-020.docx Document Version 3.0 page 1 of 40

Compensation of Dead Time in PID Controllers

2006-12-06 Page 1 of 25 Compensation of Dead Time in PID Controllers Advanced Application Note 2006-12-06 Page 2 of 25 Table of Contents: 1 OVERVIEW...3 2 RECOMMENDATIONS...6 3 CONFIGURATION...7 4 TEST

Novel Approach for LED Luminous Intensity Measurement

Novel Approach for LED Luminous Intensity Measurement Ron Rykowski Hubert Kostal, Ph.D. * Radiant Imaging, Inc., 15321 Main Street NE, Duvall, WA, 98019 ABSTRACT Light emitting diodes (LEDs) are being

Analysis of phase sensitivity for binary computer-generated holograms

Analysis of phase sensitivity for binary computer-generated holograms Yu-Chun Chang, Ping Zhou, and James H. Burge A binary diffraction model is introduced to study the sensitivity of the wavefront phase

Intorduction to light sources, pinhole cameras, and lenses

Intorduction to light sources, pinhole cameras, and lenses Erik G. Learned-Miller Department of Computer Science University of Massachusetts, Amherst Amherst, MA 01003 October 26, 2011 Abstract 1 1 Analyzing

Lab E2: B-field of a Solenoid. In the case that the B-field is uniform and perpendicular to the area, (1) reduces to

E2.1 Lab E2: B-field of a Solenoid In this lab, we will explore the magnetic field created by a solenoid. First, we must review some basic electromagnetic theory. The magnetic flux over some area A is

Lecture 11: Clocking

High Speed CMOS VLSI Design Lecture 11: Clocking (c) 1997 David Harris 1.0 Introduction We have seen that generating and distributing clocks with little skew is essential to high speed circuit design.

Minimizing Input Filter Requirements In Military Power Supply Designs

Keywords Venable, frequency response analyzer, MIL-STD-461, input filter design, open loop gain, voltage feedback loop, AC-DC, transfer function, feedback control loop, maximize attenuation output, impedance,

STATISTICAL DESIGN AND YIELD ENHANCEMENT OF LOW VOLTAGE CMOS ANALOG VLSI CIRCUITS

STATISTICAL DESIGN AND YIELD ENHANCEMENT OF LOW VOLTAGE CMOS ANALOG VLSI CIRCUITS Istanbul Technical University Electronics and Communications Engineering Department Tuna B. Tarim Prof. Dr. Hakan Kuntman

Accuracy of Microwave Cavity Perturbation Measurements

918 IEEE TRANSACTIONS ON MICROWAVE THEORY AND TECHNIQUES, VOL. 49, NO. 5, MAY 2001 Accuracy of Microwave Cavity Perturbation Measurements Richard G. Carter, Member, IEEE Abstract Techniques based on the

Basler ral km. Camera Specification. Measurement protocol using the EMVA Standard 1288 Document Number: BD Version: 01

Basler ral8-8km Camera Specification Measurement protocol using the EMVA Standard 188 Document Number: BD79 Version: 1 For customers in the U.S.A. This equipment has been tested and found to comply with

A Method for Gain over Temperature Measurements Using Two Hot Noise Sources

A Method for Gain over Temperature Measurements Using Two Hot Noise Sources Vince Rodriguez and Charles Osborne MI Technologies: Suwanee, 30024 GA, USA vrodriguez@mitechnologies.com Abstract P Gain over

A PREDICTABLE PERFORMANCE WIDEBAND NOISE GENERATOR

A PREDICTABLE PERFORMANCE WIDEBAND NOISE GENERATOR Submitted by T. M. Napier and R.A. Peloso Aydin Computer and Monitor Division 700 Dresher Road Horsham, PA 19044 ABSTRACT An innovative digital approach

Performance of the HgCdTe Detector for MOSFIRE, an Imager and Multi-Object Spectrometer for Keck Observatory

Performance of the HgCdTe Detector for MOSFIRE, an Imager and Multi-Object Spectrometer for Keck Observatory Kristin R. Kulas a, Ian S. McLean a, and Charles C. Steidel b a University of California, Los

Basic principles of photography. David Capel 346B IST

Basic principles of photography David Capel 346B IST Latin Camera Obscura = Dark Room Light passing through a small hole produces an inverted image on the opposite wall Safely observing the solar eclipse

Detailed Characterisation of a New Large Area CCD Manufactured on High Resistivity Silicon

Detailed Characterisation of a New Large Area CCD Manufactured on High Resistivity Silicon Mark S. Robbins *, Pritesh Mistry, Paul R. Jorden e2v technologies Ltd, 106 Waterhouse Lane, Chelmsford, Essex

CHAPTER 6 INTRODUCTION TO SYSTEM IDENTIFICATION

CHAPTER 6 INTRODUCTION TO SYSTEM IDENTIFICATION Broadly speaking, system identification is the art and science of using measurements obtained from a system to characterize the system. The characterization

Improved Spectra with a Schmidt-Czerny-Turner Spectrograph

Improved Spectra with a Schmidt-Czerny-Turner Spectrograph Abstract For years spectra have been measured using traditional Czerny-Turner (CT) design dispersive spectrographs. Optical aberrations inherent

An Evaluation of Artifact Calibration in the 5700A Multifunction Calibrator

An Evaluation of Artifact Calibration in the 57A Multifunction Calibrator Application Note Artifact Calibration, as implemented in the Fluke Calibration 57A Multifunction Calibrator, was a revolutionary